Atmospheric-pressure laser ionization
Atmospheric pressure laser ionization is an atmospheric pressure ionization method for mass spectrometry (MS). Laser light in the UV range is used to ionize molecules in a resonance-enhanced multiphoton ionization (REMPI) process. It is a selective and sensitive ionization method for aromatic and polyaromatic compounds.
The excitation of electrons in atoms and molecules by the absorption of one ore more photons can be sufficient for the spatial separation of the electron and the atom or molecule. In the gas phase, this process is called Photoionization. The combined energy of the absorbed photons in this process must be above the ionization potential of the atom or molecule.
In the simplest case, a single photon has sufficient energy to overcome the ionization potential. This process is therefore called single photon ionization, it is the basic principle of Atmospheric pressure photo ionization (APPI). For sufficiently high power densities of the incident light also nonlinear absorption processes like the absorption of at least two photons in a rapid sequence via virtual or real states can occur. If the combined energy of the absorbed photons is higher than the ionization potential, this multiphoton absorption process can also lead to ionization of the atom or molecule. This process is called multi photon ionization (MPI).
The laser light sources used in APLI have power densities which allow multiphoton ionization via stable electronic states of the molecule or atom. The required power density has to be sufficiently high, so that in the lifetime of the first reached electronic state, which is in the range of a few nanoseconds, a second photon can be absorbed with a reasonable probability. Then a radical cation is formed:
This process is called "resonance enhanced multi photon ionization" (REMPI). In the case of APLI both absorbed photons have the same wavelength, which is called "1+1 REMPI".
Most of the organic molecules which are favorable for a photoionization method have ionization potentials smaller than approximately 10 eV. Thus APLI utilizes light with a photon energy of around 5 eV which corresponds to a wavelength of about 250 nm, which is in the ultraviolet (UV) part of the electromagnetic spectrum.
APLI has some special characteristics because of the ionization with UV-laser-light:
Easy to couple
It is relatively simple to couple an existing atmospheric pressure (AP) ion source with APLI. In principle only the ionizing laser light has to be coupled into the existing ion source trough UV transparent windows.
APLI is s selective ionization method, because the 1+1 REMPI ionization requires an adequate existing electronic intermediate state and both electronic transitions have to be quantum mechanically allowed.
In particular polynuclear aromatic compounds fulfil the spectroscopic requirements for 1+1 REMPI, thus APLI is an ideal ionization method for the detection of polyaromatic hydro carbons (PAH).
The selectivity is also a disadvantage, if the direct ionization of an analyte molecule is not possible with APLI. In this case, the analyte molecule could be chemically coupled with a label molecule which is sensitive to APLI. If such a derivatization reaction is available, the selectivity of APLI can be broadened to other molecule classes.
In comparison to the single photon ionization (APPI) with vacuum ultraviolet light (λ = 128 nm) APLI is much more sensitive, in particular in applications with liquid chromatography (LC-MS). The selectivity of APLI is one cause of this effect, but under these conditions, APPI suffers from another effect. The VUV light utilized by APPI does not penetrate deep into the ion source geometry, because the solvents used by LC, which are present as vapor in the ion source, strongly absorb the VUV light. For the UV light of APLI the LC solvents are virtually transparent, thus APLI allows the generation of ions in the entire ion source volume.
Independence of ion formation from electrical fields
In contrast to other ionization methods like (Electrospray-Ionization (ESI) and Atmospheric-pressure chemical ionization (APCI), APLI allows the generation of ions independent from electrical fields, because the zone of ion formation is only governed by the laser light. This allows some special methods, like the measurement of the spatial resolved ion signal (Distribution of Ion Acceptance - DIA) with APLI for example, which is applied in the development of new ion sources.
- M. Constapel, M. Schellenträger, O. J Schmitz, S. Gäb, K. J Brockmann, R. Giese, Th Benter (2005). "Atmospheric‐pressure laser ionization: a novel ionization method for liquid chromatography/mass spectrometry". Rapid Communications in Mass Spectrometry 19 (3): 326–336. doi:10.1002/rcm.1789.
- Stefan Droste, Marc Schellenträger, Marc Constapel, Siegmar Gäb, Matthias Lorenz, Klaus J Brockmann, Thorsten Benter, Dieter Lubda, Oliver J Schmitz (2005). "A silica‐based monolithic column in capillary HPLC and CEC coupled with ESI‐MS or electrospray‐atmospheric‐pressure laser ionization‐MS". ELECTROPHORESIS 26 (21): 4098–4103. doi:10.1002/elps.200500326. PMID 16252331.
- R. Schiewek, M. Schellenträger, R. Mönnikes, M. Lorenz, R. Giese, K. J. Brockmann, S. Gäb, Th. Benter, O. J. Schmitz (2007). "Ultrasensitive Determination of Polycyclic Aromatic Compounds with Atmospheric-Pressure Laser Ionization as an Interface for GC/MS". Analytical Chemistry 79 (11): 4135–4140. doi:10.1021/ac0700631.
- Matthias Lorenz, Ralf Schiewek, Klaus J. Brockmann, Oliver J. Schmitz, Siegmar Gäb, Thorsten Benter (2008). "The Distribution of Ion Acceptance in Atmospheric Pressure Ion Sources: Spatially Resolved APLI Measurements". Journal of the American Society for Mass Spectrometry 19 (3): 400–410. doi:10.1016/j.jasms.2007.11.021.