In case of an arc-auxanometer (see picture), there is a wire fixed with the plant apex on one end and a dead-weight on the other. It passes over a pulley which has a pointer attached to it. When the plant's height increases, the pulley rotates and the pointer moves on a circular scale to directly give the magnitude of growth.
Sensitive auxanometers allow measurement of growth as small as a micrometer, which allows measurement of growth in response to short-term changes in atmospheric composition. Auxanometers are used in the laboratory, the field, and the classroom.
- Chisholm, Hugh, ed. (1911). "Auxanometer". Encyclopædia Britannica 3 (11th ed.). Cambridge University Press. p. 50
- Beach, Chandler B., ed. (1914). "Auxanometer". The New Student's Reference Work. Chicago: F. E. Compton and Co.
- Bovie, W. T. (1912), "A Precision Auxanometer", Botanical Gazette 53 (6): 504, doi:10.1086/330848
- Ranson, S. L.; Harrison, A. (1955), "Experiments on Growth in Length of Plant Organs", Journal of Experimental Botany 6: 75, doi:10.1093/jxb/6.1.75
- Gallagher, J. N.; Biscoe, P. V.; Saffell, R. A. (1976), "A Sensitive Auxanometer for Field Use", Journal of Experimental Botany 27 (4): 704, doi:10.1093/jxb/27.4.704
- W. T. Bovie; W. T. Bovie (1915), "A Simplified Precision Auxanometer", American Journal of Botany 2 (2): 95–99, doi:10.2307/2435215, JSTOR 2435215
|This botany article is a stub. You can help Wikipedia by expanding it.|