|Industry||Tritium Navigation; Tactical Weaponry|
|Founded||1992, Holland, Michigan, U.S.|
|Headquarters||Holland, MI, USA|
|Key people||Michael D. Alexander
(President) & (CEO)
|Products||Lensatic Compass; EasyLoader; EasyMag|
|Employees||40 - July 2009|
Cammenga is an outdoors products company, known chiefly for producing navigation equipment under contract for U.S. military forces. They are the official supplier of the M-1950 lensatic field compass issued to U.S. Army and Marine Corps infantry and artillery units.
Cammenga was founded in 1992 by a group of investors to produce the M-1950 3H lensatic compass to the U.S. Military. Since 1992, Cammenga has held the U.S. military contract for the M-1950. The firm develops, manufactures, and distributes a wide variety of products to military, law enforcement, and commercial markets around the world, including land navigation instruments, firearm sighting and loading systems, firearm magazines, and tritium light source integration services. The company provides development assistance in tritium (self-luminous) lighting products, NRC approval processing, tritium capsule assembly and installation procedures, and product testing. The company is ISO 9001:2000 certified.
Cammenga's primary production plant includes a clean room environment for the assembly and testing of tritium products and critical devices. The company provides Cammenga assists with tritium product design, NRC approval processing, tritium installation, and product testing.
U.S. M-1950 3H Lensatic Compass
The M-1950 U.S. military lensatic field compass with self-luminous lighting (designated the Model 3H by Cammenga) is fitted with self-luminous tritium vial lighting. Under U.S. milspec performance criteria, ten pre-production samples of the M-1950 are required to meet a battery of performance, durability, and accuracy standards. This includes an overall inherent accuracy standard of + or - 40 mils (2.25 degrees) from actual magnetic azimuth; a tritium illumination test; and a shock or impact test, in which samples of the M-1950 are tested dropping the compass from a height of 90 cm (35 in.) on a solid surface covered with 10 cm (4 in.) of plastic-covered sand, followed by testing to ensure performance within specified accuracy standards. A water leakage test is also performed.
A rifle magazine loading accessory, the EasyLoader is used by fighting forces and sport shooters.
Cammenga firearm magazines:
MODEL EM3-556 - 30 ROUND - NSN: 1005-01-569-1863
MODEL EM2-556 - 20 ROUND - NSN: 1005-01-573-6771
- 'Article MIL-PRF-10436N, rev. 31 October 2003, Washington, D.C.: U.S. Dept. of Defense
- U.S. MIL-PRF-10436N:Performance Specification, Compass, Magnetic, Unmounted, 31 OCT 2003: 22.214.171.124. Compass error. The error in magnetic azimuth, including that caused by pivot friction, shall not be greater than 40 mils (see 126.96.36.199). 188.8.131.52 Friction error. The error caused by friction between the pivot supporting the dial and its bearing shall not be greater than 20 mils (see 184.108.40.206.3).
- U.S. MIL-PRF-10436N:Performance Specification, 3.4.10: Illumination. The compass shall have internal, self-exciting light sources of constant luminosity for sighting and reading.
- U.S. MIL-PRF-10436N: Performance Specification 220.127.116.11, Durability. The compass in the open configuration, if applicable, shall be dropped from a height of 90 cm, alternately on its face, side and bottom, onto a solid surface covered with 10 cm of 40 grit kiln-dried sand. The sand may be covered with a sheet of plastic not greater than 2 mils thick.
- MIL-PRF-10436N, Performance Specification, 18.104.22.168 Durability. e. After each drop, alternating targets, sight the compass on one of three targets of known magnetic azimuths approximately 120º apart and record each reading. f. After every tenth drop test the compass for friction error as defined in 22.214.171.124.3
- MIL-PRF-10436N, Performance Specification, 126.96.36.199: Water leakage. The complete compass shall be submerged in 300 ml of distilled or deionized water for 24 hours at 23°±5°C. The compass shall be removed from the water and examined for water leakage. If there is evidence of water in the compass bowl at the completion of the test, it shall constitute failure of this test.