Cosmic Origins Spectrograph
The Cosmic Origins Spectrograph (COS) is a science instrument that was installed on the Hubble Space Telescope during Servicing Mission 4 (STS-125) in May 2009. It is designed for ultraviolet (115–320 nm) spectroscopy of faint point sources with a resolving power of ≈1,550–24,000. Science goals include the study of the origins of large scale structure in the universe, the formation and evolution of galaxies, and the origin of stellar and planetary systems and the cold interstellar medium. COS was developed and built by the Center for Astrophysics and Space Astronomy (CASA-ARL) at the University of Colorado at Boulder and the Ball Aerospace and Technologies Corporation in Boulder, Colorado.
COS is installed into the axial instrument bay previously occupied by the Corrective Optics Space Telescope Axial Replacement (COSTAR) instrument, and is intended to complement the Space Telescope Imaging Spectrograph (STIS) that was repaired during the same mission. While STIS operates across a wider wavelength range, COS is many times more sensitive in the UV.
The Cosmic Origins Spectrograph is an ultraviolet spectrograph that is optimized for high sensitivity and moderate spectral resolution of compact (point like) objects (stars, quasars, etc.). COS has two principal channels, one for Far Ultraviolet (FUV) spectroscopy covering 115–205 nm and one for Near Ultraviolet (NUV) spectroscopy spanning 170–320 nm. The FUV channel can work with one of three diffraction gratings, the NUV with one of four, providing both low and medium resolution spectra (table 1). In addition, COS has a narrow field of view NUV imaging mode intended for target acquisition.
One key technique for achieving high sensitivity in the FUV is minimizing the number of optics. This is done because FUV reflection and transmission efficiencies are typically quite low compared to what is common at visible wavelengths. In accomplishing this, the COS FUV channel uses a single (selectable) optic to diffract the light from HST, correct for the Hubble spherical aberration, focus the diffracted light onto the FUV detector and correct for astigmatism typical of this sort of instrument. Since aberration correction is performed after the light passes into the instrument, the entrance to the spectrograph must be an extended aperture, rather than the traditional narrow entrance slit, in order to allow the entire aberrated HST image from a point source to enter the instrument. The 2.5 arc second diameter entrance aperture allows ≈ 95% of the light from compact sources to enter COS, yielding high sensitivity at the design resolution for compact sources.
|Grating (Channel)||Approximate Useful Wavelength Range||Resolving power (λ/Δλ)|
|G130M (FUV)||115–145 nm||16,000–21,000|
|G160M (FUV)||141–178 nm||16,000–21,000|
|G140L (FUV)||<90–205 nm||1,500–4,000|
|G185M (NUV)||170–210 nm||22,000–28,000|
|G225M (NUV)||210–250 nm||28,000–38,000|
|G285M (NUV)||250–320 nm||30,000–41,000|
|G230L (NUV)||170–320 nm||2,100–3,900|
|TA1 (target acquisition imager)||170–320 nm||~0.05 arc sec. angular resolution|
Post launch performance closely matched expectations. Instrument sensitivity is close to pre-launch calibration values, and detector background is exceptionally low (0.16 counts per resolution element per 1000 seconds for the FUV detector, and 1.7 counts per resolution element per 100 seconds for the NUV detector). FUV resolution is slightly lower than pre-launch predictions due to mid-frequency polishing errors on the HST primary mirror, while NUV resolution exceeds pre-launch values in all modes. In addition, thanks to the minimal number of reflections, the G140L mode can observe light at wavelengths shorter than 90 nm despite the very low reflectivity of the MgF2 coated optics at these wavelengths.
The Cosmic Origins Spectrograph is designed to enable the observation of faint, point-like UV targets at moderate spectral resolution, allowing COS to observe hot stars (OB stars, white dwarfs, cataclysmic variables and binary stars) in the Milky Way and to observe the absorption features in the spectra of active galactic nuclei. Observations are also planned of extended objects. Spectroscopy provides a wealth of information about distant astronomical objects that is unobtainable through imaging:
Spectroscopy lies at the heart of astrophysical inference. Our understanding of the origin and evolution of the cosmos critically depends on our ability to make quantitative measurements of physical parameters such as the total mass, distribution, motions, temperatures, and composition of matter in the Universe. Detailed information on all of these properties can be gleaned from high-quality spectroscopic data. For distant objects, some of these properties (e.g., motions and composition) can only be measured through spectroscopy.
Ultraviolet (UV) spectroscopy provides some of the most fundamental diagnostic data necessary for discerning the physical characteristics of planets, stars, galaxies, and interstellar and intergalactic matter. The UV offers access to spectral features that provide key diagnostic information that cannot be obtained at other wavelengths.
Obtaining absorption spectra of interstellar and intergalactic gas forms the basis of many of the COS science programs. These spectra will address questions such as how was the Cosmic Web formed, how much mass can be found in interstellar and intergalactic gas, and what is the composition, distribution and temperature of this gas. In general, COS will address questions such as:
- What is the large-scale structure of matter in the Universe?
- How did galaxies form out of the intergalactic medium?
- What types of galactic halos and outflowing winds do star-forming galaxies produce?
- How were the chemical elements for life created in massive stars and supernovae?
- How do stars and planetary systems form from dust grains in molecular clouds?
- What is the composition of planetary atmospheres and comets in our Solar System (and beyond)?
Some specific programs include the following:
Large-Scale Structure of Baryonic Matter: With its high FUV spectroscopic sensitivity, COS uniquely suited for exploring the Lyman-alpha forest. This is the ‘forest’ of absorption spectra seen in the spectra of distant galaxies and quasars caused by intergalactic gas clouds, which may contain the majority of baryonic matter in the universe. Because the most useful absorption lines for these observations are in the far ultraviolet and the sources are faint, a high sensitivity FUV spectrograph with wide wavelength coverage is needed to perform these observations. By determining the redshift and line width of the intervening absorbers, COS will be able to map out the temperature, density and composition of dark baryonic matter in the Cosmic Web.
Great Wall Structure: Background active galactic nuclei will be used to study intergalactic absorbers to estimate their transverse size and physical density and determine how the distribution of material correlates with nearby galaxy distributions in the CFA2 Great Wall.
Additional instrument design details
COS has two channels, the Far Ultraviolet (FUV) covering 115–205 nm and the Near Ultraviolet (NUV) covering 170–320 nm. All COS optics are reflective (except for the bright object aperture filter and NUV order sorters) to maximize efficiency and to avoid chromatic aberration. Principal COS observing modes are summarized in table 1.
Light from the Hubble Space Telescope enters the instrument through either the Primary Science Aperture (PSA) or the Bright Object Aperture (BOA). The BOA introduces a neutral density filter to the optical path that attenuates the light by approximately a factor of one hundred (five astronomical magnitudes). Both apertures are oversized (2.5 arc second clear aperture) permitting more than 95% of the light from a point source to enter the spectrograph.
After passing through the PSA or BOA the light travels to one of the optics on the first of two optic select wheels, either one of the three FUV diffraction gratings or the first of the NUV collimation mirrors (table 1), depending on whether an FUV, NUV, or target acquisition channel is selected. All optics on the first wheel have an aspheric profile to correct for the Hubble spherical aberration.
The FUV channel has two medium and one low resolution spectroscopy modes. The FUV channels are modified Rowland Circle spectrographs in which the single holographically ruled aspheric concave diffraction grating simultaneously focuses and diffracts the incoming light and corrects both for the HST spherical aberration and for aberrations introduced by the extreme off-Rowland layout. The diffracted light is focused onto a 170x10 mm cross delay line microchannel plate detector. The FUV detector active area is curved to match the spectrograph’s focal surface and is divided into two physically distinct segments separated by a small gap.
The NUV channel has three medium and one low resolution spectroscopy modes as well as an imaging mode with an approximately 1.0 arc second unvignetted field of view. The NUV channels utilize a modified Czerny-Turner design in which collimated light is fed to the selected grating, followed by three camera mirrors that direct the diffracted light onto three separate stripes on a 25×25 mm Multi Anode Microchannel Array (MAMA) detector. The imaging mode is primarily intended for target acquisition.
- Wide Field and Planetary Camera
- Wide Field and Planetary Camera 2
- Wide Field Camera 3
- Advanced Camera for Surveys
- Near Infrared Camera and Multi-Object Spectrometer
- Faint Object Camera
- The Cosmic Origins Spectrograph Web Site at the University of Colorado
- the COS Web site at the Space Telescope Science Institute (STScI)