|This article does not cite any references or sources. (October 2009)|
A digital microscope is a variation of a traditional optical microscope that uses optics and a charge-coupled device (CCD) camera to output a digital image to a monitor, sometimes by means of software running on a computer. A digital microscope usually has its own in-built LED light source, and differs from a stereo microscope in that there is no provision to observe the sample directly through an eyepiece. Since the optical image is projected directly on the CCD camera, the entire system is designed for the monitor image. The optics for the human eye are omitted.
Digital microscope are commonly low priced commercial microscopes designed for use with a computer, although advanced systems that do not require a separate computer and can cost tens of thousands of dollars. The low price commercial microscopes normally omit the optics for illumination (for example Köhler illumination and phase contrast illumination) and are more akin to webcams with a macro lens. For information about stereo microscopes with a digital camera in research and development see optical microscope.
An early digital microscope was made by a lens company in Tokyo, Japan in 1986, which is now known as Hirox Co Ltd. It included a control box and a lens connected to a computer. The original connection to the computer was analog through an S-video connection. Over time that connection was changed to Firewire 800 to handle the large amount of digital information coming from the digital camera. But some of their current advanced versions are all-in-one and do not require a computer. The most advanced all-in-one systems now have HD monitors built-in.
Invention of the USB port resulted in a multitude of USB microscopes ranging in quality and magnification. They continue to fall in price, especially compared with traditional optical microscopes. They offer high resolution images which are normally recorded directly to a computer, and which also use the computer for its in-built LED light source. The optical resolution is directly related to the number of megapixels available on a specific model, from 1.3 MP, 2 MP, 5 MP and upwards.
Digital Micro-imaging Adaptor for Microscopes
Recently indian inventor Sachin G Lokapure has granted a patent for Digital microimaging adaptor built to meet the documentation demands of routine life science, industrial and educational applications at very competitive price-points. Designed around the CMOS color sensor, they share a broad dynamic range, progressive scan, accurate color replication and optimal image quality. The Software included SGL-11 software for Windows provides for excellent color rendering and fast image processing especially when capturing multiple images. Simple, intuitive and easy-to-use controls offer image capture, notation, geometric measuring and storage in JPG, BMP, TIFF, RAW format. The software also includes auto white balance, color enhancement, anti-flicker and other image-enhancing benefits.
Stereo and digital microscopes
A primary difference between a stereo microscope and a digital microscope is the magnification. With a stereo microscope the magnification is found by multiplying the lens magnification by the eyepiece magnification. Since the digital microscope does not have an eyepiece, the magnification cannot be found using this method. Instead the magnification for a digital microscope is found by how many times larger the sample is reproduced on the monitor. Therefore the magnification will depend on the size of the monitor. The average digital microscope system has a 15" monitor, would result in an average difference in magnification between an optical microscope and a digital microscope of about 40%. Thus the magnification number of a stereo microscope is usually 40% less than the magnification number of a digital microscope.
Since the digital microscope has the image projected directly on to the CCD camera, it is possible to have higher quality recorded images than with an stereo microscope. With the stereo microscope, the lenses are made for the optics of the eye. Attaching a CCD camera to an stereo microscope will result in an image that has compromises made for the eyepiece. Although the monitor image and recorded image may be of higher quality with the digital microscope, the application for the microscope may dictate which microscope is preferred.
With a typical 2 megapixel CCD, a 1600×1200 pixels image is generated. The resolution of the image depends on the field of view of the lens used with the camera. The approximate pixel resolution can be determined by dividing the horizontal field of view (FOV) by 1600.
Increased resolution can be accomplished by creating a sub-pixel image. The Pixel Shift Method uses an actuator to physically move the CCD in order to take multiple overlapping images. By combining the images within the microscope, sub-pixel resolution can be generated. This method provides sub-pixel information, averaging a standard image is also a proven method to provide sub-pixel information.
Most of the high-end digital microscope systems have the ability to measure samples in 2D. The measurements are done onscreen by measuring the distance from pixel to pixel. This allows for length, width, diagonal, and circle measurements as well as much more. Some systems are even capable of counting particles.
3D measurement is achieved with a digital microscope by image stacking. Using a step motor, the system takes images from the lowest focal plane in the field of view to the highest focal plane. Then it reconstructs theses images into a 3D model based on contrast to give a 3D color image of the sample. From these 3D model measurements can be made, but their accuracy is based on the step motor and depth of field of the lens.
2D and 3D tiling
2D and 3D tiling, also known as stitching or creating a panoramic, can now be done with the more advanced digital microscope systems. In 2D tiling the image is automatically tiled together seamlessly in real-time by moving the XY stage. 3D tiling combines the XY stage movement of 2D tiling with the Z-axis movement of 3D measurement to create a 3D panoramic.
Digital microscopes range from inexpensive units costing from perhaps US$20 which connect to a computer via USB connector, to units costing tens of thousands of dollars. These advanced digital microscope systems usually are self-contained, and do not require a computer.
Most of the cheaper microscopes which connect via USB have no stand, or a simple stand with clampable joints. They are essentially very simple webcams with small lenses and sensors—and can be used to view subjects which are not very close to the lens— mechanically arranged to allow focus at very close distances. Magnification is typically claimed to be user-adjustable from 10× to 200-400×.
Devices which connect to a computer require software to operate. Basic operation includes viewing the microscope image and recording "snapshots". More advanced functionality, possible even with simpler devices, includes recording moving images, time-lapse photography, measurement, image enhancement, annotation, etc. Many of the simpler units which connect to a computer use standard operating system facilities, and do not require device-specific drivers. A consequence of this is that many different microscope software packages can be used interchangeably with different microscopes, although such software may not support features unique to the more advanced devices. Basic operation may be possible with software included as part of computer operating systems—in Windows XP, images from microscopes which do not require special drivers can be viewed and recorded from "Scanners and Cameras" in Control Panel.
The more advanced digital microscope units have stands that hold the microscope and allow it to be racked up and down, similarly to standard optical microscopes. Calibrated movement in all three dimensions may be available. Resolution, image quality, and dynamic range vary with price. Systems with a lower number of megapixels have a higher frame rate (30fps) and faster processing. The faster processing can be seen when using functions like HDR (high dynamic range). In addition to general-purpose microscopes, instruments specialised for specific applications are produced. These units can have a magnification range up to 0-7000x, are usually all-in-one (don't require a computer), and can be portable. They also differ from the cheaper USB microscopes in not only the quality of the image but the quality of the system's construction giving these types of systems a longer lifetime.