An example of a drug-eluting stent. This is the TAXUS Express2 Paclitaxel-Eluting Coronary Stent System, which releases paclitaxel.
A drug-eluting stent (DES) is a peripheral or coronary stent (a scaffold) placed into narrowed, diseased peripheral or coronary arteries that slowly releases a drug to block cell proliferation. This prevents fibrosis that, together with clots (thrombus), could otherwise block the stented artery, a process called restenosis. The stent is usually placed within the peripheral or coronary artery by an Interventional cardiologist or Interventional Radiologist during an angioplasty procedure.
Drug-eluting stents in current clinical use were approved by the FDA after clinical trials showed they were statistically superior to bare-metal stents (BMS) for the treatment of native coronary artery narrowings, having lower rates of major adverse cardiac events (MACE) (usually defined as a composite clinical endpoint of death + myocardial infarction + repeat intervention because of restenosis).
The first procedure to treat blocked coronary arteries was coronary artery bypass graft surgery (CABG), wherein a section of vein or artery from elsewhere in the body is used to bypass the diseased segment of coronary artery. In 1977, Andreas Grüntzig introduced percutaneous transluminal coronary angioplasty (PTCA), also called balloon angioplasty, in which a catheter was introduced through a peripheral artery and a balloon expanded to dilate the narrowed segment of artery.
As equipment and techniques improved, the use of PTCA rapidly increased, and by the mid-1980s, PTCA and CABG were being performed at equivalent rates. Balloon angioplasty was generally effective and safe, but restenosis was frequent, occurring in ~30–40% of cases, usually within the first year after dilation. In ~3% of balloon angioplasty cases, failure of the dilation and acute or threatened closure of the coronary artery (often because of dissection) prompted emergency CABG.
Dotter and Melvin Judkins had suggested using prosthetic devices inside arteries (in the leg) to maintain blood flow after dilation as early as 1964. In 1986, Puel and Sigwart implanted the first coronary stent in a human patient. Several trials in the 1990s showed the superiority of stent placement over balloon angioplasty. Restenosis was reduced because the stent acted as a scaffold to hold open the dilated segment of artery; acute closure of the coronary artery (and the requirement for emergency CABG) was reduced, because the stent repaired dissections of the arterial wall. By 1999, stents were used in 84% of percutaneous coronary interventions (i.e., those done via a catheter, and not by open-chest surgery.)
Early difficulties with coronary stents included a risk of early thrombosis (clotting) resulting in occlusion of the stent. Coating stainless steel stents with other substances such as platinum or gold did not eliminate this problem. High-pressure balloon expansion of the stent to ensure its full apposition to the arterial wall, combined with drug-therapy using aspirin and another inhibitor of platelet aggregation (usually ticlopidine or clopidogrel) nearly eliminated this risk of early stent thrombosis.
Though it occurred less frequently than with balloon angioplasty or other techniques, stents nonetheless remained vulnerable to restenosis, caused almost exclusively by neointimal tissue growth. To address this issue, developers of drug-eluting stents used the devices themselves as a tool for delivering medication directly to the arterial wall. While initial efforts were unsuccessful, it was shown in 2001 that the release (elution) of drugs with certain specific physicochemical properties from the stent can achieve high concentrations of the drug locally, directly at the target lesion, with minimal systemic side effects. As currently used in clinical practice, "drug-eluting" stents refers to metal stents which elute a drug designed to limit the growth of neointimal scar tissue, thus reducing the likelihood of stent restenosis.
The first successful trials were of sirolimus-eluting stents. A clinical trial in 2002 led to approval of the sirolimus-eluting Cypher stent in Europe in 2002. After a larger pivotal trial (one designed for the purpose of achieving FDA approval), published in 2003, the device received FDA approval and was released in the U.S. in 2003. Soon thereafter, a series of trials of paclitaxel-eluting stents led to FDA approval of the Taxus stent in 2004. The Xience V everolimus eluting stent was approved by the FDA in July 2008 and has been available in Europe and other international markets since late 2006. It is an investigational device in Japan.
Clinical trials have shown the benefits of coronary stenting with BMS over other methods of angioplasty, including balloon angioplasty and atherectomy. Drug-eluting stents (DES) have also been extensively studied, and are generally superior to bare-metal stents as regards Major Adverse Cardiac Events (MACE, generally defined as death, myocardial infarction, or the need for a repeat revascularization procedure.) Stents are indicated to improve the diameter of the coronary artery lumen, when narrowing (generally because of atherosclerosis) causes ischemia (reduced oxygen delivery to the muscle supplied by that artery.)
Off-label use 
Drug-eluting stents also have been shown to be superior to bare-metal stents in reducing short-term complications of stenting in saphenous vein grafts; however, use in these bypass grafts is an example of an "off-label" use of drug-eluting stents. That is, this application has not been sufficiently examined by the Food and Drug Administration for that agency to recommend the use. For "on-label" applications, the FDA "believes that coronary drug-eluting stents remain safe and effective when used for the FDA-approved indications. These devices have significantly reduced the need for a second surgery to treat restenosis for thousands of patients each year."
There is some concern about overzealous use of stents in general. Two studies found that about half of patients received stents for unapproved reasons, with worse outcomes for the patients in both studies. More recent data suggests that off label use of both bare metal stents as well as drug eluting stents have increased risks. However, DES seemed to have similar or improved rates of death or MI compared with BMS, and consistently reduced need for target vessel revascularization (TVR). Overall, the data supports the use of DES for off-label indications.
Alternatives (to stents in general) 
Medical therapy for coronary artery disease has also improved since the 1970s, and for many kinds of patients may be as successful as stenting or surgery. For those requiring PCI or surgery, medical therapy and revascularization should be viewed as complementary rather than opposing strategies.
Coronary artery bypass graft surgery is the best treatment for some patients. Differences between outcomes with stenting and with coronary bypass grafting (CABG) are a point of controversy. A recent study comparing the outcomes of all patients in New York state treated with coronary artery bypass surgery (CABG) or percutaneous coronary intervention (PCI) demonstrated CABG was superior to PCI with DES in multivessel (two or more diseased arteries) coronary artery disease (CAD). Patients treated with CABG had lower rates of death and of death or myocardial infarction than treatment with a drug-eluting stent. Patients undergoing CABG also had lower rates of repeat revascularization. The New York State registry included all patients undergoing revascularization for coronary artery disease, but was not a randomized trial, and so may have reflected other factors besides the method of coronary revascularization.
No randomized trial comparing CABG and DES has been completed, although two trials of DES versus CABG are currently enrolling patients - SYNTAX (Synergy Between Percutaneous Coronary Intervention With Taxus and Cardiac Surgery) and FREEDOM (Future Revascularization Evaluation in Patients With Diabetes Mellitus—Optimal Management of Multivessel Disease). The registries of the nonrandomized patients screened for these trials may provide as much robust data regarding revascularization outcomes as the randomized analysis.
Other studies, including the ARTS II registry, suggest that drug-eluting stenting is not inferior to coronary bypass for treatment of multivessel coronary disease. The ARTS II registry compared a cohort of patients treated with multi-vessel stenting with DES, to the historical CABG cohort in the ARTS I trial (itself a randomized comparison between multivessel bare metal stenting vs. CABG.) At three-year follow-up, major adverse cardiac events were comparable between the ARTS II DES group and the ARTS I CABG group. Re-intervention was lower in the ARTS I CABG group. In all comparison studies of stenting vs. bypass surgery, it is worth noting that only a small minority of patients with multivessel coronary disease have been eligible for inclusion in the studies, and that for most patients, clinical judgement by experienced operators suggests that one or the other approach is preferred.
Like all invasive medical procedures, implanting stents in the coronary arteries carries risk. For the newer drug-eluting stents, very-long-term results are not yet available; however, five-years after implantation sirolimus-eluting stents remained superior to bare-metal stents.
Risks associated with cardiac catheterization procedures include bleeding, allergic reaction to the X-ray contrast agents used to visualize the coronary arteries, and myocardial infarction. With PCI, the requirement for emergency CABG has markedly decreased since the days of balloon angioplasty, such that in some communities, coronary stenting is permitted in hospitals without on-site cardiac surgery facilities, though this remains highly controversial in the United States, not least because of the rare but largely unpredictable risk of coronary artery perforation. Rarely, a type of allergic reaction to the drug may occur; episodes of fatality have been reported.
Stent thrombosis 
Although drug-eluting stents continue to represent a major medical advance for angioplasty, evidence has always shown new clot thrombosis formation with stents to be a problem, thus clotting suppressant agents are routinely given during placement and anti-clotting agents should be continued; the question is for how long. Coronary arterial healing occurs after the placement of a drug eluting stent, however complete healing of the vessel takes time. For drug eluting stents, the time course of complete healing in human is unknown.
A stent is a foreign object in the body, and the body responds to the stent’s presence in a variety of ways. Macrophages accumulate around the stent, and nearby smooth muscle cells proliferate. These physiological changes, which can cause restenosis, are limited by the drugs released by the stent, but these drugs also limit formation of a new endothelial layer over the new stent to inhibit clot formation. Endothelialization is a hallmark of vascular healing and is important for the prevention of thrombus formation. Lack of healing caused by antiproliferative drugs can make the stent an exposed surface on which a clot, sometimes life-threatening, can form.
For drug-eluting stents (which, by design, delay formation of a new endothelium cover over the stent), the incidence of clot formation within the stent may persist for a longer period of time, perhaps as long as five years after treatment. Drug eluting stents have been associated with delayed arterial healing and the prevalence of latent thrombus after five years, suggesting that patients may continue to be risk for stent thrombosis for an extended period of time.
Though less frequent with drug-eluting stents, neointimal proliferation can still occur in DES and cause restenosis. Stent occlusion because of thrombosis may occur during the procedure, in the following days, or later. The presence of thrombus around the stent may in turn affect the drug-eluting performance of the stent. Treatment with the antiplatelet drugs aspirin and clopidogrel appears to be the most important factor reducing this risk of thrombosis, and early cessation of one or both of these drugs after drug-eluting stenting markedly increases the risk of stent thrombosis and myocardial infarction. A recent histopathology study showed that very late DES thrombosis is associated with histopathological signs of inflammation and intravascular ultrasound evidence of vessel remodeling. Compared with other causes of myocardial infarction, eosinophilic infiltrates are more common in thrombi harvested from very late DES thrombosis and correlate with the extent of stent malapposition.
Whether drug-eluting stents are at higher risk than bare-metal stents for late thrombosis is intensely debated. In meta-analyses of the sirolimus and paclitaxel-eluting stent trials, there was a small but statistically higher risk of thrombosis after the first year, compared to bare metal stents. Late stent thrombosis often causes myocardial infarction and sometimes death. In other analyses, the late thrombosis risk is offset by drug-eluting stents' markedly reduced risk of restenosis and its complications including myocardial infarction. A meta-analysis concluded that the mortality risk associated with drug-eluting and bare-metal stents is similar.
Comparing different drug-eluting stents Whether sirolimus or paclitaxel-eluting stents are measurably different in their outcomes is a topic of great interest, including to the marketing departments of the manufacturers themselves. Analyses favoring one or the other stent have been advanced. The differences, if any, between the two devices are small.
The SPIRIT II study showed the Xience V DES was better than the Taxus.
Drug-eluting stents with a biodegradable coating In order to address the risk of late stent thrombosis, drug-eluting stents with a biodegradable coating have been developed. One such stent is the BioMatrix stent of Biosensors International, which has been approved by European authorities in January 2008. Two year results of a large all-comers trial suggest a reduced risk of late stent thrombosis, especially after discontinuation of dual antiplatelet therapy. The Nevo stent of Cordis/J&J, also uses a biodegradable coating and is currently undergoing trials.
Drug-eluting stents consist of three parts. Stent platform, coating, and drug.
The stent itself is an expandable metal alloy framework. Many DES are based on a bare-metal stent (BMS). The stents have elaborate mesh-like designs to allow expansion, flexibility and in some cases the ability to make/enlarge side openings for side vessels. Cobalt chrome alloy is stronger (and more radio-opaque) than the usual 316L stainless steel so the struts can be thinner which seems to reduce the degree of restenosis. (The L605 CoCr alloy has less nickel than 316L stainless steel and so may cause less allergy.)
A coating, typically of a polymer, holds and elutes (releases) the drug into the arterial wall by contact transfer. The first few DES licenced used durable coatings, but some newer coating are designed to biodegrade after or as the drug is eluted. Coatings are typically spray coated or dip coated. There can be one to three or more layers in the coating e.g. a base layer for adhesion, a main layer for holding the drug, and sometimes a top coat to slow down the release of the drug and extend its effect.
The drug is mainly to inhibit neointimal growth (due to proliferation of smooth muscle cells) which would cause restenosis. Much of the neointimal hyperplasia seems to be caused by inflammation. Hence immunosuppressive and antiproliferative drugs are used. Both sirolimus and paclitaxel were previously used for other medical applications; new drugs are being evaluated for coronary stents.  
Examples (approved for clinical use in the U.S.) :
- Cypher (J&J, Cordis ) uses a 316L stainless steel BxVelocity stent (140 µm struts) and adds a 12.6 µm 3 layer coating (2 µm Parylene C base coat, 10 µm main coat of PEVA, PBMA and sirolimus, and a 0.6 µm top coat of PBMA). The sirolimus elutes over a period of about 30 days.
- Taxus (Boston Scientific) uses a 316L stainless steel Express2 stent (132 µm struts) and adds a 16 µm single layer Translute SIBS copolymer coating containing paclitaxel which elutes over a period of about 90 days.
- Endeavour (Medtronic) uses a cobalt chrome Driver stent (91 µm struts) and adds a 4.3 µm phosphorylcholine coating that includes zotarolimus, on a 1 µm base coat.
- Xience V (Guidant, Abbott) uses an L605 cobalt chrome ML Vision stent (81 µm struts) and adds a 7.6 µm fluropolymer multilayer coating with drug everolimus.
Examples approved outside the US :
- Zilver PTX (Cook Medical) Nitinol stent, no polymer, coated with paclitaxel, for use in peripheral arteries 
- TaxCor (EuroCor GmbH) Highly Flexible Cobalt Chromium Stent Platform coated with fully biodegradable polymer as a carrier for Paciltaxel 
- Axxion (Biosensors Int) Stainless steel stent, Synthetic Glycocalix coating with paclitaxel.
- BioMatrix (Biosensors Int) S stent platform, bioabsorbable PLA coating with Biolimus A9 drug.
- ARTAX (Aachen Resonance) double helix stainless steel platform, without polymer, metal coated with paclitaxel drug.
Investigation and Alternative drugs 
There are also several other anti-proliferative drugs under investigation in human clinical trials. In general, these are analogues of sirolimus. Like sirolimus, these block the action of mTOR. Medtronic has developed zotarolimus; unlike sirolimus and paclitaxel, this sirolimus analogue designed for use in stents with phosphorylcholine as a carrier. Their ZoMaxx stent is a zotarolimus-eluting, stainless steel and tantalum–based stent; a modified phosphorylcholine slowly releases the zotarolimus . Zotarolimus has been licensed to Medtronic which is researching the effectiveness in a drug-eluting stent of their own. Their Endeavor stent, which is a cobalt alloy, also uses phosphorylcholine to carry the zotarolimus was approved for use in Europe in 2005 is now close to U.S. FDA approval .
Clinical trials are currently examining two stents carrying everolimus, an analog of sirolimus. Guidant, which has the exclusive license to use everolimus in drug-eluting stents, is the manufacturer of both stents. The Guidant vascular business was subsequently sold to Abbott . The Champion stent uses a bioabsorbable polylactic acid carrier on a stainless steel stent.   In contrast, its Xience stent uses a durable (non-bioabsorbable) polymer on a cobalt alloy stent.
One alternative to drug-eluting stents is a stent surface designed to reduce the neointimal proliferation. One such is the Genous bioengineered stent.
In place of the stainless steel (and now cobalt chrome) currently used in stents, various biodegradable frameworks are under early phases of investigation. Since metal, as a foreign substance, provokes inflammation, scarring, and thrombosis (clotting), it is hoped that biodegradable or bioabsorbable stents may prevent some of these effects. A magnesium alloy–based stent has been tested in animals, though there is currently no carrier for drug elution.  A promising biodegradable framework is made from poly-L-lactide, a polymer of a derivative of L-lactic acid. One of these stents, the Igaki-Tamai stent, has been studied in pigs; tranilast  and paclitaxel have been used as eluted drugs.
- Moses, JW; Leon, MB; Popma, JJ; Fitzgerald, PJ; Holmes, DR; O'Shaughnessy, C; Caputo, RP; Kereiakes, DJ et al. (2003). "Sirolimus-eluting stents versus standard stents in patients with stenosis in a native coronary artery.". New England Journal of Medicine 349 (14): 1315–23. doi:10.1056/NEJMoa035071. PMID 14523139.
- Stone, GW; Ellis, SG; Cox, DA; Hermiller, J; O'Shaughnessy, C; Mann, JT; Turco, M; Caputo, R et al. (2004). "One-year clinical results with the slow-release, polymer-based, paclitaxel-eluting TAXUS stent: the TAXUS-IV trial.". Circulation 109 (16): 1942–7. doi:10.1161/01.CIR.0000127110.49192.72. PMID 15078803.
- Comparison of DES, BMS and CABG over 12 months
- Grüntzig, AR; A Senning, & WE Siegenthaler (1979-07-12). "Nonoperative dilatation of coronary-artery stenosis: percutaneous transluminal coronary angioplasty". New England Journal of Medicine 301 (2): 61–68. doi:10.1056/NEJM197907123010201. PMID 449946. Retrieved 2006-07-22.
- Baim, Donald S. (2005) . "Percutaneous Coronary Revascularization". In Dennis L. Kasper, Anthony S. Fauci, Dan L. Longo, Eugene Braunwald, Stephen L. Hauser, & J. Larry Jameson. Harrison's Principles of Internal Medicine (16th ed.). New York: McGraw-Hill. pp. 1459–1462.
- Dotter, Charles T.; & Melvin P. Judkins (November 1, 1964). "Transluminal Treatment of Arteriosclerotic Obstruction". Circulation 30 (5): 654–670. doi:10.1161/01.CIR.30.5.654. PMID 14226164. Retrieved 2006-07-22. (abstract)
- Serruys PW, Kutryk MJ, Ong AT (2006). "Coronary-artery stents". N. Engl. J. Med. 354 (5): 483–95. doi:10.1056/NEJMra051091. PMID 16452560.
- Hwang, CW; Wu D, Edelman ER (2001). "Physiological transport forces govern drug distribution for stent-based delivery". Circulation 104 (5): 600–605. doi:10.1161/hc3101.092214. PMID 11479260.
- "New Device Approval - P030025 - TAXUS Express2 Paclitaxel-Eluting Coronary Stent System". Archived from the original on 2008-02-03. Retrieved 2008-02-25.
- Lee MS, Shah AP, Aragon J, Jamali A, Dohad S, Kar S, Makkar RR (2005). "Drug-eluting stenting is superior to bare metal stenting in saphenous vein grafts". Catheter Cardiovasc Interv 66 (4): 507–11. doi:10.1002/ccd.20498. PMID 16270361.
- "US FDA/CDRH: FDA Statement on Coronary Drug-Eluting Stents". Retrieved 2008-02-25.[dead link]
- Win HK, Caldera AE, Maresh K, et al. (2007). "Clinical outcomes and stent thrombosis following off-label use of drug-eluting stents". JAMA 297 (18): 2001–9. doi:10.1001/jama.297.18.2001. PMID 17488965.
- Beohar N, Davidson CJ, Kip KE, et al. (2007). "Outcomes and complications associated with off-label and untested use of drug-eluting stents". JAMA 297 (18): 1992–2000. doi:10.1001/jama.297.18.1992. PMID 17488964.
- Dixon et a. Year in Interventional Cardiology JACC Vol. 53, No. 22, 2009
- Kumar, R; Lee, TT; Jeremias, A; Ruisi, CP; Sylvia, B; Magallon, J; Kirtane, AJ; Bigelow, B et al. (2000). "Medical therapy versus coronary angioplasty in stable coronary artery disease: a critical review of the literature.". J Am Coll Cardiol 100 (8): 1187–91. doi:10.1016/j.amjcard.2007.05.038. PMID 17920355.
- Hannan EL, Wu C, Walford G, et al. (2008). "Drug-eluting stents vs. coronary-artery bypass grafting in multivessel coronary disease". N. Engl. J. Med. 358 (4): 331–41. doi:10.1056/NEJMoa071804. PMID 18216353.
- Desai ND (2008). "Pitfalls assessing the role of drug-eluting stents in multivessel coronary disease". Ann. Thorac. Surg. 85 (1): 25–7. doi:10.1016/j.athoracsur.2007.08.063. PMID 18154771.
- Serruys PW, et al. (2007). "Three-year follow-up of the ARTS-II - sirolimus-eluting stents for the treatment of patients with multivessel coronary artery disease". Eurointervention (12): 450–9.
- Morice, MC; Serruys, PW; Barragan, P; Bode, C; Van Es, GA; Stoll, HP; Snead, D; Mauri, L et al. (2007). "Long-term clinical outcomes with sirolimus-eluting coronary stents: five-year results of the Ravel trial.". JACC 50 (14): 1299–304. doi:10.1016/j.jacc.2007.06.029. PMID 17903626.
- Peels JO, Hautvast RW, de Swart JB, et al. (2008). "Percutaneous coronary intervention without on site surgical back-up; two-years registry of a large dutch community hospital". Int. J. Cardiol. 132 (1): 59–65. doi:10.1016/j.ijcard.2007.10.037. PMID 18241941.
- Virmani R, Guagliumi G, Farb A, Musumeci G, Grieco N, Motta T, Mihalcsik L, Tespili M, Valsecchi O, Kolodgie FD (2004). "Localized hypersensitivity and late coronary thrombosis secondary to a sirolimus-eluting stent: should we be cautious?". Circulation 109 (6): 701–5. doi:10.1161/01.CIR.0000116202.41966.D4. PMID 14744976.
- Finn AV, Nakazawa G, Joner M, Kolodgie FD, Mont EK, Gold HK, Virmani R (2007). "Vascular responses to drug eluting stents: importance of delayed healing". Arterioscler Thromb Vasc Biol. 27 (7): 1500–10. PMID 17510464.
- Yamamoto M, Takano M, Murakami D, Inami T, Kobayashi N, Inami S, Okamatsu K, Ohba T, Ibuki C, Hata N, Seino Y, Jang IK, Mizuno K (2011). "The possibility of delayed arterial healing 5 years after implantation of sirolimus-eluting stents: serial observations by coronary angioscopy". Am Heart J. 161 (6): 1200–6. doi:10.1016/j.ahj.2011.03.006. PMID 21641369.
- Hwang CW, Levin AD, Jonas M, Li PH, Edelman ER (2005). "Thrombosis modulates arterial drug distribution for drug-eluting stents". Circulation 111 (13): 1619–26. doi:10.1161/01.CIR.0000160363.30639.37. PMID 15795325.
- Iakovou I, Schmidt T, Bonizzoni E, et al. (2005). "Incidence, predictors, and outcome of thrombosis after successful implantation of drug-eluting stents". JAMA 293 (17): 2126–30. doi:10.1001/jama.293.17.2126. PMID 15870416.
- Cook S, Ladich E, Nakazawa G, Eshtehardi P, Neidhart M, Vogel R, Togni M, Wenaweser P, Billinger M, Seiler C, Gay S, Meier B, Pichler WJ, Jüni P, Virmani R, Windecker S. (2009). "Correlation of intravascular ultrasound findings with histopathological analysis of thrombus aspirates in patients with very late drug-eluting stent thrombosis". Circulation 120 (5): 391–9. doi:10.1161/CIRCULATIONAHA.109.854398. PMID 19620501.
- Daemen J, Serruys PW (2007). "Drug-eluting stent update 2007: part II: Unsettled issues". Circulation 116 (8): 961–8. doi:10.1161/CIRCULATIONAHA.107.691451. PMID 17709651.
- Bavry AA, Kumbhani DJ, Helton TJ, Borek PP, Mood GR, Bhatt DL (2006). "Late thrombosis of drug-eluting stents: a meta-analysis of randomized clinical trials". Am. J. Med. 119 (12): 1056–61. doi:10.1016/j.amjmed.2006.01.023. PMID 17145250.
- Stettler C, Wandel S, Allemann S, Kastrati A, Morice MC, Schömig A, Pfisterer ME, Stone GW, Leon MB, de Lezo JS, Goy JJ, Park SJ, Sabaté M, Suttorp MJ, Kelbaek H, Spaulding C, Menichelli M, Vermeersch P, Dirksen MT, Cervinka P, Petronio AS, Nordmann AJ, Diem P, Meier B, Zwahlen M, Reichenbach S, Trelle S, Windecker S, Jüni P (2007). "Outcomes associated with drug-eluting and bare-metal stents: a collaborative network meta-analysis". Lancet 370 (9591): 937–48. doi:10.1016/S0140-6736(07)61444-5. PMID 17869634.
- SPIRIT II tests cobalt chrome DES
- Klauss, Volker. "Leaders: Biolimus A9 Stent Noninferior to SES". Retrieved 25 October 2009.
- DES drugs. Brown U.
- DES inc Structures of PEVA, PBMA and Parylene C.
- Sheiban I, Villata G, Bollati M, Sillano D, Lotrionte M, Biondi-Zoccai G (2008). "Next-generation drug-eluting stents in coronary artery disease: focus on everolimus-eluting stent (Xience V®)". Vasc Health Risk Manag 4 (1): 31–8. doi:10.2147/vhrm.2008.04.01.31. PMC 2464756. PMID 18629361.
- Axxion website BioSensorsIntl.com
- Leaders trial Biolimus vs Sirolimus
- "Vascular Devices". Abbott. Archived from the original on 2006-06-15. Retrieved 2006-07-23.
- "FDA advisers OK Medtronic stent". Star Tribune. Retrieved 2007-10-10.
- "Abbott Completes Acquisition of Guidant Vascular Business". Archived from the original on 2006-12-06. Retrieved 2007-01-12.
- Grube, Eberhard; Shinjo Sonoda, Fumiaki Ikeno, Yasuhiro Honda, Saibal Kar, Charles Chan, Ulrich Gerckens, Alexandra J. Lansky, & Peter J. Fitzgerald (2004). "Six- and Twelve-Month Results From First Human Experience Using Everolimus-Eluting Stents With Bioabsorbable Polymer". Circulation 109 (18): 2168–2171. doi:10.1161/01.CIR.0000128850.84227.FD. PMID 15123533.
- "Guidant News Release — April 5, 2004". Guidant. 2004-04-05. Retrieved 2007-07-23.
- "Guidant News Release — June 22, 2005". Guidant. 2005-06-22. Retrieved 2006-07-23.
- Genous bioengineered R stent
- Heublein, B.; R. Rhode, V. Kaese, N. Niemeyer, W. Hartung, & A. Haverich (2003). "Biocorrosion of magnesium alloys: a new principle in cardiovascular implant technology?". Heart 89 (6): 651–6. doi:10.1136/heart.89.6.651. PMC 1767674. PMID 12748224. Retrieved 2006-07-23.
- Tsuji, T.; H. Tamai, K. Igaki, E. Kyo, K. Kosuga, T. Hata, T. Nakamura, S. Fujita, S. Takeda, S. Motohara, & H. Uehata (2003). "Biodegradable stents as a platform to drug loading". International Journal of Cardiovascular Interventions 5 (1): 13–6. doi:10.1080/14628840304609. PMID 12623560.
- Vogt, Felix; Andreas Steina, Gösta Rettemeier, Nicole Krott, Rainer Hoffmann, Jürgen vom Dahl, Anja-Katrin Bosserhoff, Walter Michaeli, Peter Hanrath, Christian Weber, & Rüdiger Blindt (2004). "Long-term assessment of a novel biodegradable paclitaxel-eluting coronary polylactide stent". European Heart Journal 25 (15): 1330–40. doi:10.1016/j.ehj.2004.06.010. PMID 15288161. Retrieved 2006-07-22.
Further reading 
- Fischetti, Mark (July 2006). "Vascular Stents: Expanding Use". Scientific American: 94. (layperson overview, subscription required)
- Serruys, Patrick W.; Michael J.B. Kutryk, and Andrew T.L. Ong (2006-02-02). "Coronary-Artery Stents". New England Journal of Medicine 354 (5): 483–95. doi:10.1056/NEJMra051091. PMID 16452560. (journal review article, subscription required)
|Wikimedia Commons has media related to: Stent|
- Drug-Eluting Stents — Angioplasty.Org Good overview and detail
- CIMIT Center For Integration of Medicine and Innovative Technology
- Cypher DES
- Image of the experimental CoStar Cobalt chrome stent
- Manu Sondhi, Amitha Jagannath & John B. Wong: A Meta-Analysis Of Randomized Controlled Trials With Coronary Drug-Eluting Stents Compared With Bare-Metal Stents: The Internet Journal of Cardiology. 2006; Volume 3, Number 2.
- Safety Profile of Drug-eluting Stents Similar to Bare-metal Stents (re G.W.Stone's presentation at TCT2006)
- TCT: Transcatheter Cardiovascular Therapeutics Meeting Coverage peer-reviewed articles from Medpage Today