An electromagnetic diaphragm is a form of capacitive sensor used on an electronic stethoscope. The diaphragm is coated with a conductive material. A conductive plate is positioned behind and parallel to the diaphragm, so that the two conductive elements form a capacitor. Capacitance is a function of plate area, dielectric properties of the space between the conductors, and the distance between the conductors. It is this latter parameter which is modulated by vibration such that the capacitance varies with the distance between the electromagnetic diaphragm and the plate, forming an acoustic sensor.
The electromagnetic diaphragm is a form of capacitive sensor, and shares properties with condenser microphones. The difference between the electromagnetic diaphragm and a condenser microphone is that a condenser microphone operates on the basis of air transmission, has a very thin membrane instead of a diaphragm, and cannot be placed against a body to sense sounds within the body.
See also 
- "Electromagnetic Diaphragm (EmD)". Thinklabs. thinklabsmedical.com. Retrieved 4 June 2011.
- "Transducer for sensing body sounds". patft.uspto.gov. 9 December 2003. Retrieved 4 June 2011.
- "Transducer for sensing body sounds". United States Patent and Trademark Office. patft.uspto.gov. 10 May 2011. Retrieved 4 June 2011.