Adjustable gastric band
|Adjustable gastric band|
Diagram of an adjustable gastric band
A laparoscopic adjustable gastric band, commonly called a lap-band, A band, or LAGB, is an inflatable silicone device placed around the top portion of the stomach to treat obesity, intended to slow consumption of food and thus reduce the amount of food consumed.
Adjustable gastric band surgery is an example of bariatric surgery designed for obese patients with a body mass index (BMI) of 40 or greater — or between 35 and 40 in cases of patients with certain comorbidities that are known to improve with weight loss, such as sleep apnea, diabetes, osteoarthritis, GERD, Hypertension (high blood pressure), or metabolic syndrome, among others.
In February 2011 the United States Food and Drug Administration (FDA) expanded approval of adjustable gastric bands to patients with a BMI between 30 to 40 and one weight-related medical condition, such as diabetes or high blood pressure. However, an adjustable gastric band may only be used after other methods such as diet and exercise have previously been tried.
- 1 Working principle
- 2 Surgical indications
- 3 Comparison with other bariatric surgeries
- 4 Losing weight after surgery
- 5 History and development of gastric banding
- 6 Controversy
- 7 In popular culture
- 8 References
- 9 External links
The inflatable band is placed around the upper part of the stomach to create a smaller stomach pouch. This slows and limits the amount of food that can be consumed at one time, thus giving the opportunity for the sense of satiety to be met with the release of peptide YY (PYY). It does not decrease gastric emptying time. The individual achieves sustained weight loss by choosing healthy food options, limiting food intake and volume, reducing appetite, and progress of food from the top portion of the stomach to the lower portion digestion.
According to the American Society for Metabolic Bariatric Surgery, bariatric surgery is not an easy option for obesity sufferers. It is a drastic step, and carries the usual pain and risks of any major gastrointestinal surgical operation.
However, gastric banding is the least invasive surgery of its kind and is completely reversible, with another "keyhole" operation. Gastric banding is performed using laparoscopic surgery and usually results in a shorter hospital stay, faster recovery, smaller scars, and less pain than open surgical procedures. Because no part of the stomach is stapled or removed, and the patient's intestines are not re-routed, he or she can continue to absorb nutrients from food normally. Gastric bands are made entirely of biocompatible materials, so they are able to stay in the patient’s body without causing harm.
However, not all patients are suitable for laparoscopy. Patients who are extremely obese, who have had previous abdominal surgery, or have complicating medical problems may require the open approach.
Placement via laparoscopic surgery
The surgical insertion of an adjustable gastric band is often referred to as a lap band procedure or band placement. First, a small incision (typically less than 1.25 cm or 0.5 in.) is made near the belly button. Carbon dioxide (a gas that occurs naturally in the body) is introduced into the abdomen to create a work space for the surgeon. Then a small laparoscopic camera is placed through the incision into the abdomen. The camera sends a picture of the stomach and abdominal cavity to a video monitor. It gives the surgeon a good view of the key structures in the abdominal cavity. A few additional, small incisions are made in the abdomen. The surgeon watches the video monitor and works through these small incisions using instruments with long handles to complete the procedure. The surgeon creates a small, circular tunnel behind the stomach, inserts the gastric band through the tunnel, and locks the band around the stomach.
Clinical studies of laparoscopic (minimally invasive) bariatric surgery patients found that they felt better, spent more time doing recreational and physical activities, benefited from enhanced productivity and economic opportunities, and had more self-confidence than they did prior to surgery.
The placement of the band creates a small pouch at the top of the stomach. This pouch holds approximately ½ cup of food, whereas the typical stomach holds about 6 cups of food. The pouch fills with food quickly, and the band slows the passage of food from the pouch to the lower part of the stomach. As the upper part of the stomach registers as full, the message to the brain is that the entire stomach is full, and this sensation helps the person to be hungry less often, feel full more quickly and for a longer period of time, eat smaller portions, and lose weight over time.
As patients lose weight, their bands will need adjustments, or “fills", to ensure comfort and effectiveness. The gastric band is adjusted by introducing a saline solution into a small access port placed just under the skin. A specialized non-coring needle is used to avoid damage to the port membrane and prevent leakage. There are many port designs (such as high profile and low profile), and they may be placed in varying positions based on the surgeon’s preference, but are always attached (through sutures, staples, or another method) to the muscle wall in and around the diaphragm.
Adjustable gastric bands hold between 4 to 12 cc of saline solution, depending on their design. When the band is inflated with saline solution, it places pressure around the outside of the stomach. This decreases the size of the passage between the pouch created from the upper part of the stomach and the lower stomach and further restricts the movement of food. Over the course of several visits to the doctor, the band is filled until the optimal restriction has been achieved – neither so loose that hunger is not controlled, nor so tight that food cannot move through the digestive system. The number of adjustments required is an individual experience and cannot be accurately predicted.
Types of adjustable bands
In the U.S. market, two types of adjustable gastric bands have been approved by the FDA: Realize Band and Lap-Band. The Lap-Band System obtained FDA approval in 2001. The device comes in five different sizes and has undergone modification over the years. The latest models, the Lap-Band AP-L and Lap-Band AP-S, feature a standardized injection port sutured into the skin and fill volumes of 14 mL and 10 mL respectively.
The Realize Adjustable Gastric Band obtained FDA approval in 2007. Realize Band-C has a 14% greater adjustment range than the Realize Band. Both the Realize Band and Realize Band-C, however, are one-size-fits-all. The device differentiates itself from the Lap-Band AP series through its sutureless injection port installation and larger range of isostatic adjustments. The maximum fill volume for the Realize Band is 9 ml, while the newer Realize Band-C has a fill volume of 11 ml. Both fill volumes fall within a low pressure range to prevent discomfort or strain to the band.
Two other adjustable gastric bands are in use outside of the United States — Heliogast and Midband. Neither band has been approved by the FDA. The Midband was the first to market in 2000. In order to preserve the gastric wall in event of rubbing, the device contains no sharp edges or irregularities. It is also opaque to x-rays, making it easy to locate and adjust. The Heliogast band entered the market in 2003. The device features a streamlined band to ease insertion during the operation.
In general, gastric banding is indicated for people for whom all of the following apply:
- Body Mass Index above 40, or those who are 100 pounds (7 stone/45 kilograms) or more over their estimated ideal weight, according to the National Institutes of Health, or those between 30 to 40 with co-morbidities that may improve with weight loss (type 2 diabetes, hypertension, high cholesterol, non-alcoholic fatty liver disease and obstructive sleep apnea.)
- Age between 18 and 55 years (although there are doctors who will work outside these ages, some as young as 12).
- Failure of medically supervised dietary therapy (for about 6 months).
- History of obesity (up to 5 years).
- Comprehension of the risks and benefits of the procedure and willingness to comply with the substantial lifelong dietary restrictions required for long term success.
Gastric banding is usually not recommended for people with any of the following:
- If the surgery or treatment represents an unreasonable risk to the patient
- Untreated endocrine diseases such as hypothyroidism
- Inflammatory diseases of the gastrointestinal tract such as ulcers, esophagitis or Crohn’s disease.
- Severe cardiopulmonary diseases or other conditions which may make them poor surgical candidates in general.
- An allergic reaction to materials contained in the band or who have exhibited a pain intolerance to implanted devices
- Dependency on alcohol or drugs
- People with severe learning or cognitive disabilities or emotionally unstable people
Special considerations for pregnancy
If considering pregnancy, ideally the patient should be in optimum nutritional condition prior to, or immediately following, conception; deflation of the band may be required prior to a planned conception. Deflation should also be considered should the patient experience morning sickness. The band may remain deflated during pregnancy and once breast feeding is completed, or if bottle feeding, the band may be gradually re-inflated to aid postpartum weight loss as needed.
It is highly advised to take extra precautions during intercourse after surgery as rapid weight loss increases fertility. Effective birth control methods must be used at all times to avoid unwanted pregnancies. Two factors have been pointed out by experts that may help explain this increase in fertility: PCOS, or polycystic ovary syndrome and excess estrogen.
Comparison with other bariatric surgeries
Unlike more open forms of weight loss surgery (e.g. Roux-en-Y gastric bypass surgery (RNY), Biliopancreatic Diversion (BPD) and Duodenal Switch (DS)), gastric banding does not require cutting or removing any part of the digestive system. It is removable, requiring only a laparoscopic procedure to remove the band, after which the stomach usually returns to its normal pre-banded size so it is not unusual for a person to gain weight after having a band removed. However, it is not entirely reversible as adhesions and tissue scarring are inevitable. Unlike those who have procedures such as RNY, DS, or BPD, it is unusual for gastric band patients to experience any nutritional deficiencies or malabsorption of micro-nutrients. Calcium supplements and Vitamin B12 injections are not routinely required following gastric banding (as is often the case with RNY, for example). Gastric dumping syndrome issues also do not occur since intestines are not removed or re-routed.
Typically, patients who undergo adjustable gastric banding procedures lose less weight over the first 3.5 years than those who have RNY gastric bypass, BPD, or DS surgeries. Although other procedures appear to result in greater weight loss than adjustable gastric banding in the short term, results from the study by Maggard suggest that this difference decreases significantly over time. Gastric banding patients lose an average[clarification needed] of 47.5% of their excess weight, according to a meta analysis by Buchwald.
It is important to note that, in order to maintain their weight reduction, patients must carefully follow post-operative guidelines relating to diet, exercise, and band maintenance. Weight regain is possible with any weight loss procedure, including the more radical procedures that initially result in rapid weight loss. The National Institutes of Health recommendation for weight loss is 1 to 2 pounds (½ to 1 kilogram) per week, and an average banded patient may lose this amount. This is variable based on the individual and their personal circumstances, motivation, and mobility.
Benefits of gastric banding when compared to other bariatric surgeries
- Lower mortality rate: only 1 in 1000 versus 1 in 250 for Roux-en-Y gastric bypass surgery
- No cutting or stapling of the stomach
- Short hospital stay
- Quick recovery
- Adjustable without additional surgery
- No malabsorption issues (because no intestines are bypassed)
- Fewer life-threatening complications (see complications table for details)
A commonly reported occurrence for banded patients is regurgitation of non-acidic swallowed food from the upper pouch, commonly known as Productive Burping (PBing). Productive Burping is not to be considered normal. The patient should consider eating less, eating more slowly and chewing their food more thoroughly. Occasionally, the narrow passage into the larger, lower part of the stomach might also become blocked by a large portion of unchewed or unsuitable food.
Other complications include:
- Gastritis (irritated stomach tissue)
- Erosion -The band may slowly migrate through the stomach wall. This will result in the band moving from the outside of the stomach to the inside. This may occur silently but can cause severe problems. Urgent treatment may be required if there is any internal leak of gastric contents or bleeding.
- Slippage - An unusual occurrence in which the lower part of the stomach may prolapse through the band causing an enlarged upper pouch. In severe instances this can cause an obstruction and require an urgent operation to fix.
- Malposition of the band (rare with experienced practitioners):
- If the band does not surround the vertical (esophagus-to-duodenum) axis of the stomach but instead surrounds only a non-vertical (horizontal or diagonal) axis of the stomach, as by encircling only a side or portion of a side of the stomach, then the restriction on food passage will be less, weight loss will be reduced, and a painful kink in the stomach may result.
- In two reported cases, prompted by patients' reporting an absence of appetite suppression and of weight loss, investigation revealed that the band had not enclosed the stomach at all, instead enclosing only perigastric fat, with the effect that the passage of food was not restricted at all. In these cases, the patients experienced no additional adverse symptoms.
- Problems with the port and/or the tube connecting port and band - The port can "flip over" so that the membrane can no longer be accessed with a needle from the outside (this often goes hand in hand with a tube kink, and may require repositioning as a minor surgical procedure under local anaesthesia); the port may get disconnected from the tube or the tube may be perforated in the course of a port access attempt (both would result in loss of fill fluid and restriction, and likewise require a minor operation).
- Internal bleeding
Documented adverse effects
The following are the adverse effects of gastric banding as documented by the FDA.
Band- and port-specific
- Band slippage/pouch dilation
- Esophageal dilatation/dysmotility
- Erosion of the band into the gastric lumen
- Mechanical malfunctions - port leakage, cracking of the kink-resistant tubing or disruption of the tubing connection from the port to the band
- Port site pain
- Port displacement
- Infection of the fluid within the band
- Bulging of the port through the skin
- Nausea and/or vomiting
- Gastroesophageal reflux
- Stoma obstruction
- Abnormal stools
Body as a whole
- Abdominal pain
- Chest pain
- Incisional infection
- Incision pain
- Blood clots
- Hair loss
- Gall stones
- Abnormal healing
- Band intolerance
- An inability to maintain proper restriction
Losing weight after surgery
The average gastric banding patient loses 500 grams to a kilogram (1–2 pounds) per week consistently, but heavier patients often lose faster in the beginning. This comes to roughly 22 to 45 kilograms (48 to 99 pounds) the first year for most band patients. It is important to keep in mind that while most of the RNY patients drop the weight faster in the beginning, some studies have found that LAGB patients will have the same percentage of excess weight loss and comparable ability to keep it off after only a couple of years. The procedure tends to encourage better eating habits which, in turn, helps in producing long term weight stability. However, with greater experience and longer patient follow up, several studies have found suboptimal weight loss and high complication rates for the gastric band.
A systematic review concluded "LAGB has been shown to produce a significant loss of excess weight while maintaining low rates of short-term complications and reducing obesity-related comorbidities. LAGB may not result in the most weight loss but it may be an option for bariatric patients who prefer or who are better suited to undergo less invasive and reversible surgery with lower perioperative complication rates. One caution with LAGB is the uncertainty about whether the low complication rate extends past three years, given a possibility of increased band-related complications (e.g., erosion, slippage) requiring re-operation".
Band adjustments and weight loss
Correct and sensitive adjustment of the band is imperative for weight loss and the long term success of the procedure. Adjustments (also called "fills") may be performed using an X-ray fluoroscope so that the radiologist can assess the placement of the band, the port, and the tubing that runs between the port and the band. The patient is given a small cup of liquid containing a clear or white radio-opaque fluid similar to barium—clear or white. When swallowed, the fluid is clearly shown on X–ray and is watched as it travels down the esophagus and through the restriction caused by the band. The radiologist is then able to see the level of restriction in the band and to assess if there are potential or developing issues of concern. These may include dilation of the esophagus, an enlarged pouch, prolapsed stomach (when part of the stomach moves into the band where it does not belong), erosion or migration. Reflux type symptoms may indicate too great a restriction and further assessment may be required.
Under some circumstances fluid is removed from the band prior to further investigation and re-evaluation. In other cases further surgery may be required (e.g. removal of the band), should gastric erosion or a similar complication be detected. Some health practitioners adjust the band without the use of X-ray control (fluoroscopy). In these cases, the doctor assesses the patient’s weight loss and potential reflex symptoms described by the patient, such as heartburn, regurgitation, or chest pain. From this information, the doctor decides whether a band adjustment is necessary. Adjustments are often indicated if a patient has regained weight, if their weight loss has leveled off, or if the patient has a distinct feeling that food is difficult to move through the stomach.
During a clinical visit, patients visiting for a regular fill adjustment will typically find they will spend more time talking about the adjustment and their progress than the actual fill itself, which generally will only take about one to two minutes. However, for some patients this type of fill is not possible, due to issues such as partial rotation of the port, or excess tissue above the port making it difficult to determine its precise location. In these cases, a fluoroscope will generally be used. It is more common practice for the band not to be filled at surgery—although some surgeons choose to place a small amount in the band at the time of surgery. The stomach tends to swell following surgery and it is possible that too great a restriction would be achieved if filled at that time. Clearly, this is undesirable.
Many health practitioners make the first adjustment between 4 and 6 weeks post operatively to allow the stomach time to heal. After that, fills are performed as needed.
No accurate number of adjustments required can be given. The amount of saline/isotonic solution needed in the band varies from patient to patient. There are a small number of people who find they do not need a fill at all and have sufficient restriction immediately following surgery. Others may need significant adjustments to the maximum the band is able to hold.
Post-surgical diet and care
The patient may be prescribed a liquid-only diet, followed by mushy foods and then solids. This is prescribed for a varied length of time and each surgeon and manufacturer varies. Some may find that before their first fill that they are still able to eat fairly large portions. This is not surprising since before the fill there is little or no restriction in the band. This is why a proper post-op diet and a good after-care plan is essential to success. A recent study found that patients who did not change their eating habits were 2.2 times more likely to be unsuccessful than those who did, and that patients who had not increased their physical activity were 2.3 times more likely to be unsuccessful than those who did.
In principle, a diet long term post gastric band surgery should consist of normal healthy food, that is solid in nature and requires ample chewing to achieve a paste consistency prior to swallowing. This texture will maximise the effect of the band, rather than choosing easier wet foods, such as soups, casseroles and smoothies, which pass through the band quickly and easily resulting in greater caloric intake.
It is very important to discuss post-surgical care and diet plans with your weight loss team if you are considering this surgery. Recommendations can vary dramatically from team to team and it is important to find a weight loss team with a good post-surgical plan. Some teams offer support groups, but unfortunately many of them mix RNY and gastric bypass patients with gastric banding patients. Some gastric band patients have criticized this approach because while many of the underlying issues related to obesity are the same, the needs and challenges of the two groups are very different, as are their early rates of weight loss. Some gastric band recipients feel the procedure is a failure when they see that RNY patients generally lose weight faster.
History and development of gastric banding
At the end of the 1970s, Wilkinson developed several surgical approaches whose common aim was to limit food intake without disrupting the continuity of the gastro-intestinal tract.
The early 1980s saw further developments, with Kolle (Norway), Molina & Oria (US), Naslund (Sweden), Frydenberg (Australia) and Kuzmack (United States) implanting non-adjustable gastric bands made from a variety of different materials, including marlex mesh, dacron vascular prosthesis, silicone covered mesh and Gore-Tex, among others. In addition, Bashour developed the "gastro-clip", a 10.5 cm polypropylene clip with a 50cc pouch and a fixed 1.25 cm stoma, which was later abandoned due to high rates of gastric erosion
All these early attempts at restriction using meshes, bands and clips showed a high failure rate due to difficulty in achieving correct stomal diameter, stomach slippage, erosion, food intolerance, intractable vomiting and pouch dilatation. Despite these difficulties, an important ancillary observation was that silicone was identified as the best tolerated material for a gastric device, with far fewer adhesions and tissue reactions than other materials. Nevertheless, adjustability became the “Holy Grail” of these early pioneers.
The development of the modern adjustable gastric band is a tribute both to the vision and persistence of the early pioneers, particularly Lubomyr Kuzmak and a sustained collaborative effort on the part of bio-engineers, surgeons and scientists.
Early research on the concept of band “adjustability” can be traced back to the early work of G. Szinicz (Austria) who experimented with an adjustable band, connected to a subcutaneous port, in animals.
In 1986, Lubomyr Kuzmak, a Ukrainian surgeon who had emigrated to the United States in 1965, reported on the clinical use of the “adjustable silicone gastric band” (ASGB) via open surgery. Kuzmak, who from the early 1980s had been searching for a simple and safe restrictive procedure for severe obesity, modified his original silicone non-adjustable band he had been using since 1983, by adding an adjustable portion. His clinical results showed an improved weight loss and reduced complication rates compared with the non-adjustable band he had started using in 1983. Kuzmak’s major contributions were the application of Mason’s teachings about VBG to the development of the gastric band; the volume of the pouch; the need to overcome staple line disruption; the ratification of the use of silicone and the essential element of adjustability.
Separately, but in parallel with Kuzmak, Hallberg and Forsell in Stockholm, Sweden also developed an adjustable gastric band. After further work and modifications this eventually became known as the Swedish Adjustable Gastric Band (SAGB).
In early 1985, Dr. Dag Hallberg applied for a patent for the Swedish Adjustable Gastric Band (SAGB) within Scandinavian countries. In late March, Dr. Hallberg presented his idea of the "balloon band" at the Swedish Surgical Society and started to use the SAGB in a controlled series of 50 procedures. During this time, laparoscopic surgery was not common and Dr. Hallberg and his assistant, Dr. Peter Forsell, started performing the open technique to implant the SAGB.
In 1992, Dr. Forsell was in contact with different surgeons in Switzerland, Italy and Germany who began to implant the SAGB with the laparoscopic technique. Dr. Forsell fully owned the patent at this time. In 1994, Dr. Forsell presented the SAGB at an international workshop for bariatric surgery in Sweden, and from then on, the SAGB started to be implanted laparoscopically. During this time, the SAGB was manufactured by a Swedish company, ATOS Medical.
The laparoscopic era
The advent of surgical laparoscopy has transformed the field of bariatric surgery and made the gastric band a more appealing option for the surgical management of obesity. In 1992, Prof. Guy-Bernard Cadière was the first to apply an adjustable band (the Kuzmak ASGB device) by the laparoscopic approach. Over the next couple of years, the Kuzmak ASGB was modified to make it suitable for laparoscopic implantation, eventually emerging as the modern lap band. This landmark innovation was driven by Belachew, Cadière, Favretti and O’Brien; the Inamed Development Company engineered the device. The first human laparoscopic implantation of the newly developed lap band was performed by Belachew and le Grand on 1 September 1993 in Huy, Belgium, followed on 8 September, by Cadière and Favretti in Padua, Italy. In 1993, Broadbent in Australia and Catona in Italy, implanted non-adjustable (Molina-type) gastric bands by laparoscopy. In 1994, the first international laparoscopic band workshop was held in Belgium and the first on the SAGB in Sweden.
Single port laparoscopy (SPL) is an advanced, minimally invasive procedure in which the surgeon operates almost exclusively through a single entry point, typically the navel. Special articulating instruments and access ports obviate the need to place trocars externally for triangulation, thus allowing the creation of a small, solitary portal of entry into the abdomen. The SPL technique has been used to perform many types of surgery, including adjustable gastric banding and sleeve gastrectomy.
In 2003, the American Institute of Gastric Banding (AIGB) True Results opened the first outpatient surgery center using advanced laparoscopic technology to be recognized by the American College of Surgeons as an Accredited Outpatient Bariatric Center in the United States. The first outpatient Lap Band procedure in the United States was performed in Texas at AIGB True Results' Surgery Center of Richardson in 2003 and since that time the AIGB True Results has performed more than 30,000 outpatient Lap Band procedures.
There is currently a request for a congressional investigation by members of the U.S. Congress into lapband safety, prompted by recent patient deaths after lapband surgeries at clinics affiliated with the 1-800-GET-THIN advertising campaign in Southern California.
In popular culture
As with many developments in approaches to weight loss, some high-profile and well-publicized public figures have increased public opinion and awareness of gastric banding:
- Khaliah Ali (daughter of Muhammad Ali)
- Fern Britton
- Chris Christie - New Jersey Governor
- Guillermo Del Toro - Director
- Brian Dennehy - Actor
- Anne Diamond - Radio presenter, Journalist
- John Daly (golfer) - U.S. golfer
- Joseph R. Gannascoli
- Harry Knowles
- Ira Joe Fisher Television announcer and weatherman
- Sharon Osbourne
- Mikey Robins
- Rex Ryan - NY Jets Football Coach
- Caitlin Van Zandt - Guiding Light actress
- Ann Wilson - Musician
- Tina Malone - Actress
- Benedict Francis - Actor
- Lee Strafford - Sheffield Wednesday chairman
- Corey Harrison - Reality Star Pawn Stars
- Lauren Manzo - Reality Star The Real Housewives of New Jersey
- Carnie Wilson - Singer Wilson Phillips
- "Most Popular E-mail Newsletter". USA Today. 2011-02-17.
- "Gastric Band/Lap Band". University of California San Diego.
- Bariatric Surgery for Severe Obesity, National Institutes of Health. 2009;08-4006. Retrieved 2010-07-30.
- Buchwald H, Avidor Y, Braunwald E, et al. (2004). "Bariatric surgery: a systematic review and meta-analysis". JAMA 292 (14): 1724–37. doi:10.1001/jama.292.14.1724. PMID 15479938.
- Lee C, Zieve, D. Laparoscopic gastric banding. National Institutes of Health. Retrieved 2010-07-08.
- Surgical Procedures & Innovations: Gastric Bandings Columbia University, Center for Metabolic and Weight Loss Surgery; New York, NY: 2007. Retrieved 2010-07-09.
- FDA Approval: LAP-BAND Adjustable Gastric Banding (LAGB) System - P000008, fda.gov 2001-06-05. Retrieved 2010-01-21.
- Laparoscopic Adjustable Gastric Banding: Surgical Technique, thinforlife.med.nyu.edu
- FDA Approval: REALIZE Band - P070009, fda.gov 2007-09-28. Retrieved 2010-01-21.
- REALIZE Gastric Band Comparison, Ethicon Endo-Surgery Retrieved 2010-01-21.
- Médical Innovation Développement, Médical Innovation Développement. Retrieved 2010-01-21.
- Laparoscopic Adjustable Gastric Banding Using the Midband, Monkhouse, S., Johnson, A., Bates, S., Sawyer, L., Lord, K., Morgan, J. Departments of Surgery, Endocrinology and Dietetics, Southmead Hospital, Bristol. UK. Retrieved 2010-01-21.
- Functioning of the Héliogast band, Helioscopie. Retrieved 2010-01-21.
- Metabolic Surgery Expected to Play Bigger Role in Treating Type 2 Diabetes And Other Metabolic Diseases, American Society of Metabolic & Bariatric Surgery 2007. Retrieved 2010-08-26.
- The Kid I Want To Be, July 2007
- Dixon J, Dixon M, O’Brien P. Pregnancy after Lap-Band Surgery: Management of the Band to Achieve Healthy Weight Outcomes. Obesity Surgery. 2001 Feb;11(1):59-65. Retrieved 2010-08-26.
- Can weight loss increase fertility?, Discovery Fit & Health. Retrieved 2013-12-06.
- Maggard M, Shugarman L, Suttorp M, et al. Meta-analysis: surgical treatment of obesity. Annals of Internal Medicine. 2005;142:547-559. Retrieved 2010-09-16.
- Aim for a Healthy Weight, National Institutes of Health 2005;05-5213. Retrieved 2010-09-16.
- Maggard M, Shugarman L, Suttorp M, et al. LAP BAND Diet Guidelines. University of California, San Diego; UCSD Medical Center, Center for the Treatment of Obesity; San Diego, CA: 2007. Retrieved 2010-09-16.
- Brief History and Summary of Bariatric Surgery, American Society of Metabolic & Bariatric Surgery, 200. Retrieved 2010-09-16.
- Suter M, Calmes J, Paroz A, et al. A 10-year experience with laparoscopic gastric banding for morbid obesity: high long-term complication and failure rates. Obesity Surgery. 2006;16: 829-835. Retrieved 2010-09-16.
- Pfeifer, Stuart (September 23, 2011). "Another patient dies after Lap-Band surgery". Los Angeles Times.
- Surgical procedures and innovations. Columbia University Center for Metabolic and Weight Loss Surgery. Retrieved 2010-10-01.
- "CADTH: Laparoscopic Adjustable Gastric Banding for Weight Loss in Obese Adults: Clinical and Economic Review". Archived from the original on 22 October 2007. Retrieved 2007-09-24.
- Cherian P, Tentzeris V, Sigurdsson A. Variation of outcome in weight loss with band volume adjustments under clinical and radiological control following laparoscopic adjustable gastric banding. Obesity Surgery. 2010;20:13–18. Retrieved 2010-10-01.
- Chevallier J, Paita M, Rodde-Dunet M. Predictive Factors of Outcome After Gastric Banding: A Nationwide Survey on the Role of Center Activity and Patients' Behavior. Annals of Surgery. 2007;6:1034–1039. Retrieved 2010-10-01.
- Denos, Matt. "Lap Band vs Gastric Bypass". Bariatric Surgery Information Guide. bariatricguide.org. Retrieved 19 May 2013.
- Wilkinson LH (1980). "Reduction of gastric reservoir capacity". The American Journal of Clinical Nutrition 33 (2 Suppl): 515–7. PMID 7355831.
- Wilkinson LH, Peloso OA (1981). "Gastric (reservoir) reduction for morbid obesity". Archives of Surgery 116 (5): 602–5. PMID 7235951.
- Kolle K. Gastric banding (abstract). OMGI 7th Congress, Stockholm,1982;145:37[verification needed]
- Molina M, Oria HE. Gastric segmentation: a new, safe, effective, simple, readily revised and fully reversible surgical procedure for the correction of morbid obesity (abstract 15). In: 6th Bariatric Surgery Colloquium; Iowa City, IA: June 2–3, 1983[verification needed]
- Näslund E, Granström L, Stockeld D, Backman L (1994). "Marlex Mesh Gastric Banding: A 7-12 Year Follow-up". Obesity Surgery 4 (3): 269–73. doi:10.1381/096089294765558494. PMID 10742785.
- Frydenberg HB (1985). "The surgical management of obesity". Australian Family Physician 14 (10): 1017–8, 1020–2. PMID 3936462.
- Frydenberg HB (1991). "Modification of Gastric Banding, Using a Fundal Suture". Obesity Surgery 1 (3): 315–7. doi:10.1381/096089291765561088. PMID 10775933.
- Kuzmak LI (1986). "Silicone gastric banding: a simple and effective operation for morbid obesity". Contemporary Surgery 28: 13–8.
- Bashour SB, Hill RW (1985). "The gastro-clip gastroplasty: an alternative surgical procedure for the treatment of morbid obesity". Texas Medicine 81 (10): 36–8. PMID 4060078.
- Oria, Horacio E.; Doherty, Cornelius (2007). "Farewell to a Pioneer: Lubomyr Kuzmak". Obesity Surgery 17 (2): 141–2. doi:10.1007/s11695-007-9036-z.
- Szinicz G, Müller L, Erhart W, Roth FX, Pointner R, Glaser K (1989). ""Reversible gastric banding" in surgical treatment of morbid obesity--results of animal experiments". Research in Experimental Medicine 189 (1): 55–60. doi:10.1007/BF01856030. PMID 2711037.
- Cadière GB, Bruyns J, Himpens J, Favretti F (1994). "Laparoscopic gastroplasty for morbid obesity". The British Journal of Surgery 81 (10): 1524. doi:10.1002/bjs.1800811042. PMID 7820493.
- Belachew M, Legrand M, Vincent V, Lismonde M, Le Docte N, Deschamps V (1998). "Laparoscopic adjustable gastric banding". World Journal of Surgery 22 (9): 955–63. doi:10.1007/s002689900499. PMID 9717421.
- Cadiere GB, Favretti F, Bruyns J, et al. Gastroplastie par celiovideoscopie:technique. J Celio Chir 1994;10:27-31.[verification needed]
- Broadbent R, Tracey M, Harrington P (1993). "Laparoscopic Gastric Banding: a preliminary report". Obesity Surgery 3 (1): 63–7. doi:10.1381/096089293765559791. PMID 10757907.
- Catona A, Gossenberg M, La Manna A, Mussini G (1993). "Laparoscopic Gastric Banding: preliminary series". Obesity Surgery 3 (2): 207–9. doi:10.1381/096089293765559610. PMID 10757923.
- Nguyen NT, Hinojosa MW, Smith BR, Reavis KM (2008). "Single laparoscopic incision transabdominal (SLIT) surgery-adjustable gastric banding: a novel minimally invasive surgical approach". Obesity Surgery 18 (12): 1628–31. doi:10.1007/s11695-008-9705-6. PMID 18830779.
- Saber AA, Elgamal MH, Itawi EA, Rao AJ (2008). "Single incision laparoscopic sleeve gastrectomy (SILS): a novel technique". Obesity Surgery 18 (10): 1338–42. doi:10.1007/s11695-008-9646-0. PMID 18688685.
- House members call for probe into Lap-Band safety, marketing - Los Angeles Times 1/20/12
- "She floats like a butterfly, too", thinforlife.med.nyu.edu
- "Britton admits to stomach surgery". BBC Online. 2008-05-31. Retrieved 2008-06-01.
- Sharon Osbourne candid about colorectal cancer, USA Today, 2003-11-14. Retrieved on May 30, 2007
|Wikimedia Commons has media related to Adjustable gastric band.|
- NIH statistics on weight and obesity
- Medical Journal of Australia study
- Education Site for Lap-Band Patients
- American Society for Metabolic and Bariatric Surgery