Laboratory robotics is the act of using robots in biology or chemistry labs. For example, pharmaceutical companies employ robots to move biological or chemical samples around to synthesize novel chemical entities or to test pharmaceutical value of existing chemical matter. Advanced laboratory robotics can be used to completely automate the process of science, as in the Robot Scientist project.
Laboratory processes are suited for robotic automation as the processes are composed of repetitive movements (e.g. pick/place, liquid & solid additions, heating/cooling, mixing, shaking, testing).
Automated synthesis originated with peptide and oligonucleotide synthesis. One early example is the polymerase chain reaction (PCR) which is able to amplify DNA strands using a thermal cycler to micromanage DNA synthesis by adjusting temperature using a pre-made computer program. Since then, automated synthesis has been applied to organic chemistry and expanded into three categories: reaction-block systems, robot-arm systems, and non-robotic fluidic systems. The primary objective of any automated workbench is high-throughput processes and cost reduction. This allows a synthetic laboratory to operate with a fewer number of people working more efficiently.
Biological Laboratory Robotics
Biological and chemical samples, in either liquid or solid state, are stored in vials, plates or tubes. Often, they need to be frozen and/or sealed to avoid contamination or to retain their biological and/or chemical properties. Specifically, the life science industry has standardized on a plate format, known as the microtiter plate, to store such samples.
The microtiter plate standard was formalized by the Society for Biomolecular Screening in 1996. It typically has 96, 384 or even 1536 sample wells arranged in a 2:3 rectangular matrix. The standard governs well dimensions (e.g. diameter, spacing and depth) as well as plate properties (e.g. dimensions and rigidity).
A number of companies have developed robots to specifically handle SBS microplates. Such robots may be liquid handlers which aspirates or dispenses liquid samples from and to these plates, or "plate movers" which transport them between instruments.
Other companies have pushed integration even further: on top of interfacing to the specific consumables used in biology, some robots (Andrew by Andrew Alliance, see picture) have been designed with the capability of interfacing to volumetric pipettes used by biologists and technical staff. Essentially, all the manual activity of liquid handling can be performed automatically, allowing humans spending their time in more conceptual activities.
Instrument companies have designed plate readers which can carry out detect specific biological, chemical or physical events in samples stored in these plates. These readers typically use optical and/or computer vision techniques to evaluate the contents of the microtiter plate wells.
One major area where automated synthesis has been applied is structure determination in pharmaceutical research. Processes such as NMR and HPLC-MS can now have sample preparation done by robotic arm. Additionally, structural protein analysis can be done automatically using a combination of NMR and X-ray crystallography. Crystallization often takes hundreds to thousands of experiments to create a protein crystal suitable for X-ray crystallography. An automated micropipet machine can allow nearly a million different crystals to be created at once, and analyzed via X-ray crystallography.
Combinatorial Library Synthesis
Robotics have applications with Combinatorial Chemistry which has great impact on the pharmaceutical industry. The use of robotics has allowed for the use of much smaller reagent quantities and mass expansion of chemical libraries. The "parallel synthesis" method can be improved upon with automation. The main disadvantage to "parallel-synthesis" is the amount of time it takes to develop a library, automation is typically applied to make this process more efficient.
The main types of automation are classified by the type of solid-phase substrates, the methods for adding and removing reagents, and design of reaction chambers. Polymer resins may be used as a substrate for solid-phase. It is not a true combinatorial method in the sense that "split-mix" where a peptide compound is split into different groups and reacted with different compounds. This is then mixed back together split into more groups and each groups is reacted with a different compound. Instead the "parallel-synthesis" method does not mix, but reacts different groups of the same peptide with different compounds and allows for the identification of the individual compound on each solid support. A popular method implemented is the reaction block system due to its relative low cost and higher output of new compounds compared to other "parallel-synthesis" methods. Parallel-Synthesis was developed by Mario Geysen and his colleagues and is not a true type of combinatorial synthesis, but can be incorporated into a combinatorial synthesis. This group synthesized 96 peptides on plastic pins coated with a solid support for the solid phase peptide synthesis. This method uses a rectangular block moved by a robot so that reagents can be pipetted by a robotic pipetting system. This block is separated into wells which the individual reactions take place. These compounds are later cleaved from the solid-phase of the well for further analysis. Another method is the closed reactor system which uses a completely closed off reaction vessel with a series of fixed connections to dispense. Though the produce fewer number of compounds than other methods, its main advantage is the control over the reagents and reaction conditions. Early closed reaction systems were developed for peptide synthesis which required variations in temperature and a diverse range of reagents. Some closed reactor system robots have a temperature range of 200 °C and over 150 reagents.
Simulated distillation, a type of gas chromatography testing method used in the petroleum, can be automated via robotics. An older method used a system called ORCA (Optimized Robot for Chemical Analysis)was used for the analysis of petroleum samples by simulated distillation (SIMDIS). ORCA has allowed for shorter analysis times and has reduced maximum temperature needed to elute compounds. One major advantage of automating purification is the scale at which separations can be done. Using microprocessors, ion-exchange separation can be conducted on a nanoliter scale in a short period of time.
Robotics have been implemented in liquid-liquid extraction (LLE) to streamline the process of preparing biological samples using 96-well plates. This is an alternative method to solid-phase extraction methods and protein precipitation, which has the advantage of being more reproducible and robotic assistance has made LLE comparable in speed to solid phase extraction. The robotics used for LLE can perform an entire extraction with quantities in the microliter scale and performing the extraction in as little as ten minutes.
Advantages and Disadvantages
One of the advantages to automation faster processing, but it is not necessarily faster than a human operator. Repeatability and reproducibility are improved as automated systems as less likely to have variances in reagent quantities and less likely to have variances in reaction conditions. Typically productivity is increased since human constraints, such as time constraints, are no longer a factor. Efficiency is generally improved as robots can work continuously and reduce the amount of reagents used to perform a reaction. Also there is a reduction in material waste. Automation can also establish safer working environments since hazardous compounds do not have to be handled. Additionally automation allows staff to focus on other tasks that are not repetitive.
Typically the cost of a single synthesis or sample assessment are expensive to set up and start up cost for automation can be expensive. Many techniques have not been developed for automation yet. Additionally there is difficultly automating instances where visual analysis, recognition, or comparison is required such as color changes. This also leads to the analysis being limited by available sensory inputs.One potential disadvantage is an increases job shortages as automation may replace staff members who do tasks easily replicated by a robot. Some systems require the use of programming languages such as C++ or Visual Basic to run more complicated tasks.
- Mortimer, James A.; Hurst, W. Jeffrey (1987). Laboratory robotics: a guide to planning, programming, and applications. New York, N.Y: VCH Publishers. ISBN 0-89573-322-6.
- Ward, K. B.; Perozzo, M. A.; Zuk, W. M. (1988). "Automatic preparation of protein crystals using laboratory robotics and automated visual inspection". Journal of Crystal Growth 90: 325. doi:10.1016/0022-0248(88)90328-4.
- King, R. D.; Whelan, K. E.; Jones, F. M.; Reiser, P. G. K.; Bryant, C. H.; Muggleton, S. H.; Kell, D. B.; Oliver, S. G. (2004). "Functional genomic hypothesis generation and experimentation by a robot scientist". Nature 427 (6971): 247–252. doi:10.1038/nature02236. PMID 14724639.
- Nicholas W Hird Drug Discovery Today, Volume 4, Issue 6, p.265-274 (1999) 
- David Cork, Tohru Sugawara. Laboratory Automation in the Chemical Industries. CRC Press, 2002.
- Barsoum, I. S.; Awad, A. Y. (1972). "Microtiter plate agglutination test for Salmonella antibodies". Applied microbiology 23 (2): 425–426. PMC 380357. PMID 5017681.
- "Microplate Standardization, Report 3" submitted by T. Astle Journal of Biomolecular Screening (1996). Vol. 1 No. 4, pp 163-168.
- hands-free use of pipettes, October 2012, retrieved September 30, 2012
- Gary A. McClusky, Brian Tobias. "Automation of Structure Analysis in Pharmaceutical R&D." Journal of Management of Information Systems (1996).
- Heinemann, Udo, Gerd Illing, and Hartmut Oschkinat. "High-Throughput Three-Dimensional Protein Structure Determination." Current Opinion in Biotechnology 12.4 (2001): 348-54.
- Hardin, J.; Smietana, F., Automating combinatorial chemistry: A primer on benchtop robotic systems. Mol Divers 1996, 1 (4), 270-274.
- H. M. Geysen, R. H. Meloen, S. J. Barteling Proc. Natl. Acad. Sci. USA 1984, 81, 3998.
- William F. Berry, V. G., Automated simulated distillation using an articulated laboratory robot system. Journal of Automatic Chemistry 1994, 16 (6), 205-209.
- Paegel, Brian M., Stephanie H. I. Yeung, and Richard A. Mathies. "Microchip Bioprocessor for Integrated Nanovolume Sample Purification and DNA Sequencing." Analytical chemistry 74.19 (2002): 5092-98.
- Peng, S. X.; Branch, T. M.; King, S. L., Fully Automated 96-Well Liquid−Liquid Extraction for Analysis of Biological Samples by Liquid Chromatography with Tandem Mass Spectrometry. Analytical Chemistry 2000, 73 (3), 708-714.
- Cargill, J. F.; Lebl, M., New methods in combinatorial chemistry — robotics and parallel synthesis. Current Opinion in Chemical Biology 1997, 1 (1), 67-71.