Liquid mirror telescope
Liquid mirror telescopes are telescopes with mirrors made with a reflective liquid. The most common liquid used is mercury, but other liquids will work as well (for example, low melting alloys of gallium). The container for the liquid is rotating so that the liquid assumes a paraboloidal shape. A paraboloidal shape is precisely the shape needed for the primary mirror of a telescope. The rotating liquid assumes the paraboloidal shape regardless of the container's shape. To reduce the amount of liquid metal needed, and thus weight, a rotating mercury mirror uses a container that is as close to the necessary parabolic shape as possible. Liquid mirrors can be a low cost alternative to conventional large telescopes. Compared to a solid glass mirror that must be cast, ground, and polished, a rotating liquid metal mirror is much less expensive to manufacture.
Isaac Newton noted that the free surface of a rotating liquid forms a circular paraboloid and can therefore be used as a telescope, but he could not actually build one because he had no way to stabilize the speed of rotation (the electric motor did not exist yet). The concept was further developed by Ernesto Capocci of the Naples Observatory (1850), but it was not until 1872 that Henry Skey of Dunedin, New Zealand constructed the first working laboratory liquid mirror telescope.
Another difficulty is that a telescope with a liquid metal mirror can only be used in zenith telescopes that look straight up at the zenith, so it is not suitable for investigations where the telescope must remain pointing at the same location of space (a possible exception to this rule may exist for a mercury mirror space telescope, where the effect of Earth's gravity is replaced by artificial gravity, perhaps by rotating the telescope on a very long tether, or propelling it gently forward with rockets). Only a telescope located at the North Pole or South Pole would offer a relatively static view of the sky, although the freezing point of mercury and the remoteness of the location would need to be considered. A very large telescope already exists at the South Pole, but the North Pole is located in the Arctic Ocean.
Currently, the mercury mirror of the Large Zenith Telescope in Canada is the largest liquid metal mirror in operation. It has a diameter of six meters, and rotates at a rate of about 8.5 revolutions per minute.
- 1 Explanation of the equilibrium
- 2 Liquid mirror telescopes
- 3 Advantages and disadvantages
- 4 List of liquid mirror telescopes
- 5 Notes
- 6 See also
- 7 References
- 8 External links
Explanation of the equilibrium
In fluid mechanics, the state when no part of the fluid has motion relative to any other part of the fluid is called 'solid body rotation'. When the liquid has reached a state of solid body rotation, then the dynamic equilibrium can be understood as a balance of two energies: gravitational potential energy, and rotational kinetic energy. When a fluid is in solid body rotation it is the lowest state of energy that is available, because in a state of solid body rotation there is no friction to dissipate any of the energy. In an inertial reference frame, the dynamic equilibrium cannot be understood in terms of an equilibrium of forces. This is because when the liquid is rotating, there is an unbalanced force acting on the liquid – the force of gravity is acting in a vertical direction on the liquid, and the surface of the parabolic dish exerts a normal force on the liquid resting on it. The resultant force is a net centripetal force toward the axis of rotation.
The following discussion is for the case of the liquid as it is rotating in solid body rotation.
The kinetic energy of a parcel of liquid given by the formula:
In the case of circular motion the relation holds ( is in radians per second), hence
The gravitational potential energy is given by
where is the acceleration of gravity and is the height of the liquid's surface above some arbitrary elevation, for instance, we can set to be the lowest liquid surface.
We set the potential energy equal to the kinetic energy to find the liquid's shape:
This is of the form , where is a constant, which is, by definition, a paraboloid.
It is not necessary to invoke the equality of rotational kinetic energy and gravitational potential energy. With reference to the force diagram above, the vertical component of the normal force (green arrow) must equal the weight of the parcel (red arrow), which is , and the horizontal component of the normal force must equal the centripetal force (blue arrow) that keeps the parcel in circular motion, which is . Since the green arrow is perpendicular to the surface of the liquid, the slope of the surface must equal the quotient of these forces:
Cancelling the 's, integrating, and setting when leads to
which is identical to the result obtained by the previous method, and likewise shows that the liquid surface is paraboloidal.
Rotation speed and focal length
The equation of the paraboloid in terms of its focal length (derived here) can be written as:
where is the focal length, and and are defined as above.
Dividing this equation by the last one above it eliminates and and leads to:
which relates the angular velocity of the rotation of the liquid to the focal length of the paraboloid that is produced by the rotation. Note that no other variables are involved. The density of the liquid, for example, has no effect on the focal length of the paraboloid. The units must be consistent, e.g. may be in metres, in radians per second, and in metres per second-squared. The angle unit in must be radians.
On the Earth's surface, where is approximately 9.81 metres per second-squared, this last equation reduces to the approximation:
where is the focal length in metres, and is the rotation speed in revolutions per minute (RPM).
Liquid mirror telescopes
Conventional land-based liquid mirror telescopes
These are made of liquid stored in a cylindrical container made of a composite material, such as Kevlar. The cylinder is spun until it reaches a few revolutions per minute. The liquid gradually forms a paraboloid, the shape of a conventional telescopic mirror. The mirror's surface is very precise and small imperfections in the cylinder's shape do not affect it. The amount of mercury used is small, less than a millimeter in thickness.
Moon-based liquid mirror telescopes
Low temperature ionic liquids (below 130 kelvins) have been proposed as the fluid base for an extremely large diameter spinning liquid mirror telescope to be based on the Earth's moon. Low temperature is advantageous in imaging long wave infrared light which is the form of light (extremely red-shifted) that arrives from the most distant parts of the visible universe. Such a liquid base would be covered by a thin metallic film that forms the reflective surface.
Space-based ring liquid mirror telescopes
The Rice liquid mirror telescope design is similar to conventional liquid mirror telescopes. It will only work in space; but in orbit, gravity will not distort the mirror's shape into a paraboloid. The design features a liquid stored in a flat-bottomed ring-shaped container with raised interior edges. The central focal area would be rectangular, but a secondary rectangular-parabolic mirror would gather the light to a focal point. Otherwise the optics are similar to other optical telescopes. The light gathering power of a Rice telescope is equivalent to approximately the width times the diameter of the ring, minus a percentage based on optics, superstructure design, etc.
Advantages and disadvantages
The greatest advantage of a liquid mirror telescopic is its small cost, about 1% of a conventional mirror. This cuts down the cost of the entire telescope at least 95%. The University of British Columbia's 6 meter Large Zenith Telescope cost about a fiftieth as much as a conventional telescope with a glass mirror. Unfortunately, the mirror can only be pointed straight up. If it tilts from the zenith, it loses its shape. The mirror's view changes as the Earth rotates and objects cannot be physically tracked. An object can be briefly electronically tracked while in the field of view by applying a voltage to the CCD to shift electrons across it at the same speed as the image moves; this tactic is called "drift scanning." However, some types of astronomical research are unaffected by these limitations, such as long-term sky surveys and supernova searches. Since the Universe is believed to be isotropic and homogeneous (this is called the Cosmological Principle), the investigation of its structure by cosmologists can also use liquid mirror telescopes. However, research is underway to develop telescopes that can be tilted.
Since mercury metal and its vapor are both toxic to humans and animals there remains a problem for its use in any telescope where it may affect its users and others in its area. The less toxic metal gallium may be used instead of mercury but has the disadvantage of high cost. Recently Canadian researchers have proposed the substitution of magnetically deformable liquid mirrors composed of a suspension of iron and silver nanoparticles in ethylene glycol. In addition to low toxicity and relatively low cost, such a mirror would have the advantage of being easily and rapidly deformable using variations of magnetic field strength.
List of liquid mirror telescopes
Various prototypes exist historically. Following a resurgence of interest in the technology in the 1980s, several projects came to fruition.
- UBC/Laval LMT, 2.65 m, 1992
- NASA-LMT, 3 m, 1995-2002
- LZT, 6 m, 2003-
- ILMT, 4 m, 2011 test
- Borra, Ermanno F.; et al. (21 June 2007). "Deposition of metal films on an ionic liquid as a basis for a lunar telescope". Nature 447 (7147): 979–981. Bibcode:2007Natur.447..979B. doi:10.1038/nature05909. PMID 17581579. Retrieved 2008-10-11.
- "Liquid-mirror telescope set to give stargazing a new spin". Govert Schilling. 2003-03-14. Archived from the original on 2003-08-18. Retrieved 2008-10-11.
- American Chemical Society (2008, November 12) (November 12, 2008). "'Liquid Mirror' Advance May Lead To Better Eye Exams, Improved Telescopes". Science News (Science Daily (online)). Retrieved November 24, 2009.
- Déry, J. P.; Borra, E. F.; Ritcey, A. M. (2008). "Ethylene Glycol Based Ferrofluid for the Fabrication of Magnetically Deformable Liquid Mirrors". Chemistry of Materials 20 (20): 6420. doi:10.1021/cm801075u.
- List of telescope parts and construction
- List of telescope types
- Mercury glass, internally silvered decorative glass products named for their resemblance to mercury
- Mercury silvering, a technique to apply a thin layer of a precious metal to a base metal object
- Rotating furnace, used to make large glass mirrors
- Liquid mirror space telescope
- Solar cooker
- Specular reflection
- The Economist - Mirror, Mirror
- The 4m International Liquid Mirror Telescope Project
- The Large Zenith Telescope
- Gallium Liquid Mirror
- Gibson, B. K. (1991). " PDF (138 KiB)". Journal of the Royal Astronomical Society of Canada 85 (4): 158–171. Bibcode:1991JRASC..85..158G.
- Hickson, Paul (May–June 2007). "An old idea for astronomical imaging is undergoing a technology-driven renaissance" (PDF). American Scientist. Retrieved 2007-04-23.
- Large Zenith Telescope A 6 meter diameter mercury mirror telescope.