Matrix-assisted laser desorption/ionization
Matrix-assisted laser desorption/ionization (MALDI) is a soft ionization technique used in mass spectrometry, allowing the analysis of biomolecules (biopolymers such as DNA, proteins, peptides and sugars) and large organic molecules (such as polymers, dendrimers and other macromolecules), which tend to be fragile and fragment when ionized by more conventional ionization methods. It is similar in character to electrospray ionization (ESI) in that both techniques are relatively soft ways of obtaining large ions in the gas phase, though MALDI produces far fewer multiply charged ions.
MALDI is a two-step process. First, desorption is triggered by a UV laser beam. Matrix material heavily absorbs UV laser light, leading to the ablation of upper layer (~1 μm) of the matrix material. The hot plume produced during ablation contains many species: neutral and ionized matrix molecules, protonated and deprotonated matrix molecules, matrix clusters and nanodroplets. Second, the analyte molecules are ionized (more accurately protonated or deprotonated) in the hot plume. Ablated species may participate in the ionization of analyte, though the mechanism of MALDI is still debated.
- 1 Matrix
- 2 Laser
- 3 Ionization mechanism
- 4 Atmospheric pressure matrix-assisted laser desorption/ionization
- 5 Mass spectrometer
- 6 History
- 7 Applications
- 8 Reproducibility and performance
- 9 See also
- 10 References
- 11 Bibliography
- 12 External links
|Compound||Other Names||Solvent||Wavelength (nm)||Applications|
|2,5-dihydroxy benzoic acid||DHB, Gentisic acid||acetonitrile, water, methanol, acetone, chloroform||337, 355, 266||peptides, nucleotides, oligonucleotides, oligosaccharides|
|3,5-dimethoxy-4-hydroxycinnamic acid||sinapic acid; sinapinic acid; SA||acetonitrile, water, acetone, chloroform||337, 355, 266||peptides, proteins, lipids|
|4-hydroxy-3-methoxycinnamic acid||ferulic acid||acetonitrile, water, propanol||337, 355, 266||proteins|
|α-Cyano-4-hydroxycinnamic acid||CHCA||acetonitrile, water, ethanol, acetone||337, 355||peptides, lipids, nucleotides|
|3-hydroxy picolinic acid||HPA||Ethanol||337, 355||oligonucleotides|
The matrix consists of crystallized molecules, of which the three most commonly used are 3,5-dimethoxy-4-hydroxycinnamic acid (sinapinic acid), α-cyano-4-hydroxycinnamic acid (alpha-cyano or alpha-matrix) and 2,5-dihydroxybenzoic acid (DHB). A solution of one of these molecules is made, often in a mixture of highly purified water and an organic solvent (normally acetonitrile (ACN) or ethanol). Trifluoroacetic acid (TFA) may also be added. A good example of a matrix-solution would be 20 mg/mL sinapinic acid in ACN:water:TFA (50:50:0.1).
The identification of suitable matrix compounds is determined to some extent by trial and error, but they are based on some specific molecular design considerations:
- They are of a fairly low molecular weight (to allow easy vaporization), but are large enough (with a low enough vapor pressure) not to evaporate during sample preparation or while standing in the spectrometer.
- They are often acidic, therefore act as a proton source to encourage ionization of the analyte. Basic matrices have also been reported.
- They have a strong optical absorption in either the UV or IR range, so that they rapidly and efficiently absorb the laser irradiation. This efficiency is commonly associated with chemical structures incorporating several conjugated double bonds, as seen in the structure of cinnamic acid.
- They are functionalized with polar groups, allowing their use in aqueous solutions.
- They typically contain a chromophore.
The matrix solution is mixed with the analyte (e.g. protein-sample). A mixture of water and organic solvent allows both hydrophobic and water-soluble (hydrophilic) molecules to dissolve into the solution. This solution is spotted onto a MALDI plate (usually a metal plate designed for this purpose). The solvents vaporize, leaving only the recrystallized matrix, but now with analyte molecules embedded into MALDI crystals. The matrix and the analyte are said to be co-crystallized. Co-crystallization is a key issue in selecting a proper matrix to obtain a good quality mass spectrum of the analyte of interest.
MALDI techniques typically employ the use of UV lasers such as nitrogen lasers (337 nm) and frequency-tripled and quadrupled Nd:YAG lasers (355 nm and 266 nm respectively). Although not as common, infrared lasers are used due to their softer mode of ionization. IR-MALDI also has the advantage of greater material removal (useful for biological samples), less low-mass interferences, and compatibility with other matrix-free laser desorption mass spectrometry methods.
The laser is fired at the matrix crystals in the dried-droplet spot. The matrix absorbs the laser energy and it is thought that primarily the matrix is desorbed and ionized (by addition of a proton) by this event. The matrix is then thought to transfer proton to the analyte molecules (e.g., protein molecules), thus charging the analyte. Ions observed after this process consist of a neutral molecule [M] and an added or removed ion. Together, they form a quasimolecular ion, for example [M+H]+ in the case of an added proton, [M+Na]+ in the case of an added sodium ion, or [M-H]- in the case of a removed proton. MALDI is capable of creating singly charged ions, but multiply charged ions ([M+nH]n+) can also be created, as a function of the matrix, the laser intensity and/or the voltage used. Note that these are all even-electron species. Ion signals of radical cations (photoionized molecules) can be observed, e.g., in case of matrix molecules and other organic molecules.
Atmospheric pressure matrix-assisted laser desorption/ionization
Atmospheric pressure (AP) matrix-assisted laser desorption/ionization (MALDI) is an ionization technique (ion source) that in contrast to vacuum MALDI operates at normal atmospheric environment. The main difference between vacuum MALDI and AP-MALDI is the pressure in which the ions are created. In vacuum MALDI, ions are typically produced at 10 mTorr or less while in AP-MALDI ions are formed in atmospheric pressure. In the past the main disadvantage of AP MALDI technique compared to the conventional vacuum MALDI has been its limited sensitivity; however, ions can be transferred into the mass spectrometer with high efficiency and attomole detection limits have been reported.
AP-MALDI is used in mass spectrometry (MS) in a variety of applications ranging from proteomics to drug discovery. Popular topics that are addressed by AP-MALDI mass spectrometry include: proteomics; mass analysis of DNA, RNA, PNA, lipids, oligosaccharides, phosphopeptides, bacteria, small molecules and synthetic polymers, similar applications as available also for vacuum MALDI instruments.
The type of a mass spectrometer most widely used with MALDI is the TOF (time-of-flight mass spectrometer), mainly due to its large mass range. The TOF measurement procedure is also ideally suited to the MALDI ionization process since the pulsed laser takes individual 'shots' rather than working in continuous operation. MALDI-TOF instrument or reflectron is equipped with an "ion mirror" that reflects ions using an electric field, thereby doubling the ion flight path and increasing the resolution. Today, commercial reflectron TOF instruments reach a resolving power m/Δm of well above 20,000 FWHM (full-width half-maximum, Δm defined as the peak width at 50% of peak height).
The term matrix-assisted laser desorption ionization (MALDI) was coined in 1985 by Franz Hillenkamp, Michael Karas and their colleagues. These researchers found that the amino acid alanine could be ionized more easily if it was mixed with the amino acid tryptophan and irradiated with a pulsed 266 nm laser. The tryptophan was absorbing the laser energy and helping to ionize the non-absorbing alanine. Peptides up to the 2843 Da peptide melittin could be ionized when mixed with this kind of “matrix”. The breakthrough for large molecule laser desorption ionization came in 1987 when Koichi Tanaka of Shimadzu Corp. and his co-workers used what they called the “ultra fine metal plus liquid matrix method” that combined 30 nm cobalt particles in glycerol with a 337 nm nitrogen laser for ionization. Using this laser and matrix combination, Tanaka was able to ionize biomolecules as large as the 34,472 Da protein carboxypeptidase-A. Tanaka received one-quarter of the 2002 Nobel Prize in Chemistry for demonstrating that, with the proper combination of laser wavelength and matrix, a protein can be ionized. Karas and Hillenkamp were subsequently able to ionize the 67 kDa protein albumin using a nicotinic acid matrix and a 266 nm laser. Further improvements were realized through the use of a 355 nm laser and the cinnamic acid derivatives ferulic acid, caffeic acid and sinapinic acid as the matrix. The availability of small and relatively inexpensive nitrogen lasers operating at 337 nm wavelength and the first commercial instruments introduced in the early 1990s brought MALDI to an increasing number of researchers. Today, mostly organic matrices are used for MALDI mass spectrometry.
In proteomics, MALDI is used for the rapid identification of proteins isolated by using gel electrophoresis: SDS-PAGE, size exclusion chromatography, affinity chromatography, strong/weak ion exchange, isotope coded protein labelling (ICPL),and two-dimensional gel electrophoresis. Peptide mass fingerprinting is the most popular analytical application of MALDI-TOF mass spectrometers. MALDI TOF/TOF mass spectrometers are used to reveal amino acid sequence of peptides using post-source decay or high energy collision-induced dissociation (further use see mass spectrometry).
Loss of sialic acid has been identified in papers when DHB has been used as a matrix for MALDI MS analysis of glycosylated peptides. Using sinapinic acid, 4-HCCA and DHB as matrices, S. Martin studied loss of sialic acid in glycosylated peptides by metastable decay in MALDI/TOF in linear mode and reflector mode. A group at SHIMADZU CORPORATION derivatized the sialic acid by an amidation reaction as a way to improve detection sensitivity and also demonstrated that ionic liquid matrix reduces a loss of sialic acid during MALDI/TOF MS analysis of sialylated oligosaccharides. THAP, DHAP, and a mixture of 2-aza-2-thiothymine and phenylhydrazine  have been identified as matrices that could be used to minimize loss of sialic acid during MALDI MS analysis of glycosylated peptides.
It has been reported that a reduction in loss of some post-translational modifications can be accomplished if IR MALDI is used instead of UV MALDI 
In molecular biology, a mixture of 5-methoxysalicylic acid and spermine can be used as a matrix for oligonucleotides analysis in MALDI mass spectrometry, for instance after oligonucleotide synthesis.
Some synthetic macromolecules, such as catenanes and rotaxanes, dendrimers and hyperbranched polymers, and other assemblies, have molecular weights extending into the thousands or tens of thousands, where most ionization techniques have difficulty producing molecular ions. MALDI is a simple and fast analytical method that can allow chemists to rapidly analyze the results of such syntheses and verify their results.
In polymer chemistry MALDI can be used to determine the molar mass distribution. Polymers with polydispersity greater than 1.2 are difficult to characterize with MALDI due to the signal intensity discrimination against higher mass oligomers. A good matrix for polymers is dithranol and AgTFA. The sample must first be mixed with dithranol and the AgTFA added afterwards; otherwise the sample would precipitate out of solution.
MALDI/TOF spectra are used for the identification of microorganisms such as bacteria or fungi. A colony of the microbe in question is smeared directly on the sample target and overlayed with matrix. The mass spectra generated are analyzed by dedicated software and compared with stored profiles. Species diagnosis by this procedure is much faster, more accurate and cheaper than other procedures based on immunological or biochemical tests. MALDI/TOF may become the standard method for species identification in medical microbiological laboratories over the next few years.
Reproducibility and performance
The sample preparation for MALDI is important for both sensitivity, reproducibility, and quantification of mass analysis. Inorganic salts which are also part of protein extracts interfere with the ionization process. The salts can be removed by solid phase extraction or by washing the dried-droplet MALDI spots with cold water. Both methods can also remove other substances from the sample. The matrix-protein mixture is not homogenous because the polarity difference leads to a separation of the two substances during co-crystallization. The spot diameter of the target is much larger than that of the laser, which makes it necessary to make many laser shots at different places of the target, to get the statistical average of the substance concentration within the target spot. The matrix chemical composition, the addition of trifluoroacetic acid, formic acid, fructose, delay time between the end of laser pulse and start of ion acceleration in the ion source (in vacuum MALDI sources), laser wavelength, UV energy (as well as its density and homogeneity)in a focused light spot produced by pulsed laser, and the impact angle of the laser on the target are among critical parameters for the quality and reproducibility of the MALDI-TOF MS method.
- Strupat, K.; Karas, M.; Hillenkamp, F. (1991). "2,5-Dihidroxybenzoic acid: A new matrix for laser desorption—ionization mass spectrometry". International Journal of Mass Spectrometry and Ion Processes 72 (111): 89–102. Bibcode:1991IJMSI.111...89S. doi:10.1016/0168-1176(91)85050-V.
- Beavis, R. C.; Chait, B. T.; Standing, K. G. (1989). "Matrix-assisted laser-desorption mass spectrometry using 355 nm radiation". Rapid Communications in Mass Spectrometry 3 (12): 436–9. doi:10.1002/rcm.1290031208. PMID 2520224.
- Beavis, R. C.; Chait, B. T.; Fales, H. M. (1989). "Cinnamic acid derivatives as matrices for ultraviolet laser desorption mass spectrometry of proteins". Rapid Communications in Mass Spectrometry 3 (12): 432–5. doi:10.1002/rcm.1290031207. PMID 2520223.
- Beavis, R. C.; Chaudhary, T.; Chait, B. T. (1992). "α-Cyano-4-hydroxycinnamic acid as a matrix for matrix-assisted laser desorption mass spectrometry". Organic Mass Spectrometry 27 (2): 156–8. doi:10.1002/oms.1210270217.
- Tang, K.; Taranenko, N. I.; Allman, S. L.; Cháng, L. Y.; Chen, C. H.; Lubman, D. M. (1994). "Detection of 500-nucleotide DNA by laser desorption mass spectrometry". Rapid Communications in Mass Spectrometry 8 (9): 727–30. doi:10.1002/rcm.1290080913. PMID 7949335.
- Wu, K. J.; Steding, A.; Becker, C. H. (1993). "Matrix-assisted laser desorption time-of-flight mass spectrometry of oligonucleotides using 3-hydroxypicolinic acid as an ultraviolet-sensitive matrix". Rapid Communications in Mass Spectrometry 7 (2): 142–6. doi:10.1002/rcm.1290070206. PMID 8457722.
- Fitzgerald, M. C.; Parr, G. R.; Smith, L. M. (1993). "Basic matrixes for the matrix-assisted laser desorption/ionization mass spectrometry of proteins and oligonucleotides". Analytical Chemistry 65 (22): 3204–11. doi:10.1021/ac00070a007. PMID 8291672.
- Zenobi, R.; Knochenmuss, R. (1998). "Ion formation in MALDI mass spectrometry". Mass Spectrometry Reviews 17 (5): 337–366. doi:10.1002/(SICI)1098-2787(1998)17:5<337::AID-MAS2>3.0.CO;2-S.
- "Infrared MALDI". Murray Group at Louisiana State University. Archived from the original on 2011-07-24. Retrieved 2010-01-20.
- Knochenmuss, R (2006). "Ion formation mechanisms in UV-MALDI". Analyst 131 (9): 966–986. Bibcode:2006Ana...131..966K. doi:10.1039/b605646f. PMID 17047796.
- Laiko, V. V.; Baldwin, M. A.; Burlingame, A. L. (2000). "Atmospheric pressure matrix-assisted laser desorption/ionization mass spectrometry". Analytical Chemistry 72 (4): 652–7. doi:10.1021/ac990998k. PMID 10701247.
- Strupat, K.; Scheibner, O.; Arrey, T.; Bromirski, M. (2011). "Biological Applications of AP MALDI with Thermo Scientific Exactive Orbitrap MS". 443-539-1710. Thermo Scientific. Retrieved 17 June 2011.
- Laiko, V. V.; Moyer, S. C.; Cotter, R. J. (2000). "Atmospheric pressure MALDI/ion trap mass spectrometry". Analytical Chemistry 72 (21): 5239–43. doi:10.1021/ac000530d. PMID 11080870.
- Ruotolo, B. T.; Gillig, K. J.; Woods, A. S.; Egan, T. F.; Ugarov, M. V.; Schultz, J. A.; Russell, D. H. (2004). "Analysis of Phosphorylated Peptides by Ion Mobility-Mass Spectrometry". Analytical Chemistry 76 (22): 6727–6733. doi:10.1021/ac0498009.
- Ruotolo, B. T.; Verbeck, G. F.; Thomson, L. M.; Woods, A. S.; Gillig, K. J.; Russell, D. H. (2002). "Distinguishing between Phosphorylated and Nonphosphorylated Peptides with Ion Mobility−Mass Spectrometry". Journal of Proteome Research 1 (4): 303–306. doi:10.1021/pr025516r.
- Pasa-Tolic, L.; Huang, Y.; Guan, S.; Kim, H. S.; Marshall, A. G. (1995). "Ultrahigh-resolution matrix-assisted laser desorption/ionization Fourier transform ion cyclotron resonance mass spectra of peptides". Journal of Mass Spectrometry 30 (6): 825–833. doi:10.1002/jms.1190300607.
- Karas, M.; Bachmann, D.; Hillenkamp, F. (1985). "Influence of the Wavelength in High-Irradiance Ultraviolet Laser Desorption Mass Spectrometry of Organic Molecules". Analytical Chemistry 57 (14): 2935–9. doi:10.1021/ac00291a042.
- Karas, M.; Bachmann, D.; Bahr, U.; Hillenkamp, F. (1987). "Matrix-Assisted Ultraviolet Laser Desorption of Non-Volatile Compounds". International Journal of Mass Spectrometry and Ion Processes 78: 53–68. doi:10.1016/0168-1176(87)87041-6.
- Tanaka, K.; Waki, H.; Ido, Y.; Akita, S.; Yoshida, Y.; Yoshida, T.; Matsuo, T. (1988). "Protein and Polymer Analyses up to m/z 100 000 by Laser Ionization Time-of flight Mass Spectrometry". Rapid Communications in Mass Spectrometry 2 (20): 151–3. doi:10.1002/rcm.1290020802.
- Markides, K.; Gräslund, A. (9 October 2002). "Advanced information on the Nobel Prize in Chemistry 2002". The Royal Swedish Academy of Sciences. pp. 1–13. Retrieved 2013-08-28.
- Karas, M.; Hillenkamp, F. (1988). "Laser desorption ionization of proteins with molecular masses exceeding 10,000 daltons". Analytical Chemistry 60 (20): 2299–301. doi:10.1021/ac00171a028. PMID 3239801.
- Karas, M.; Bahr, U. (1990). "Laser Desorption Ionization Mass Spectrometry of Large Biomolecules". Trends in Analytical Chemistry 9 (10): 321–5. doi:10.1016/0165-9936(90)85065-F.
- Huberty, M. C.; Vath, J. E.; Yu, W.; Martin, S. A. (1993). "Site-specific carbohydrate identification in recombinant proteins using MALD-TOF MS". Analytical Chemistry 65 (20): 2791–2800. doi:10.1021/ac00068a015.
- Sekiya, S.; Wada, Y.; Tanaka, K. (2005). "Derivatization for Stabilizing Sialic Acids in MALDI-MS". Analytical Chemistry 77 (15): 4962–4968. doi:10.1021/ac050287o.
- Fukuyama, Y.; Nakaya, S.; Yamazaki, Y.; Tanaka, K. (2008). "Ionic Liquid Matrixes Optimized for MALDI-MS of Sulfated/Sialylated/Neutral Oligosaccharides and Glycopeptides". Analytical Chemistry 80 (6): 2171–2179. doi:10.1021/ac7021986.
- Papac, D. I.; Wong, A.; Jones, A. J. S. (1996). "Analysis of Acidic Oligosaccharides and Glycopeptides by Matrix-Assisted Laser Desorption/Ionization Time-of-Flight Mass Spectrometry". Analytical Chemistry 68 (18): 3215–3223. doi:10.1021/ac960324z.
- Harvey, D. J. (1999). "Matrix-assisted laser desorption/ionization mass spectrometry of carbohydrates". Mass Spectrometry Reviews 18 (6): 349–451. doi:10.1002/(SICI)1098-2787(1999)18:6<349::AID-MAS1>3.0.CO;2-H.
- Lattova, E.; Chen, V. C.; Varma, S.; Bezabeh, T.; Perreault, H. (2007). "Matrix-assisted laser desorption/ionization on-target method for the investigation of oligosaccharides and glycosylation sites in glycopeptides and glycoproteins". Rapid Communications in Mass Spectrometry 21 (10): 1644–1650. doi:10.1002/rcm.3007.
- Tajiri, M.; Takeuchi, T.; Wada, Y. (2009). "Distinct Features of Matrix-Assisted 6 μm Infrared Laser Desorption/Ionization Mass Spectrometry in Biomolecular Analysis". Analytical Chemistry 81 (16): 6750–6755. doi:10.1021/ac900695q.
- Distler, A. M.; Allison, J. (2001). "5-Methoxysalicylic acid and spermine: A new matrix for the matrix-assisted laser desorption/ionization mass spectrometry analysis of oligonucleotides". Journal of the American Society for Mass Spectrometry 12 (4): 456–62. doi:10.1016/S1044-0305(01)00212-4. PMID 11322192.
- Schrepp, W.; Pasch, H. (2003). MALDI-TOF Mass Spectrometry of Synthetic Polymers. Springer-Verlag. ISBN 3-540-44259-6.
- Nielen, M. W. F.; Malucha, S. (1997). "Characterization of polydisperse synthetic polymers by size-exclusion chromatography/matrix-assisted laser desorption/ionization time-of-flight mass spectrometry". Rapid Communications in Mass Spectrometry 11 (11): 1194. doi:10.1002/(SICI)1097-0231(199707)11:11<1194::AID-RCM935>3.0.CO;2-L.
- Wu, K. J.; Odom, R. W. (1998). "Characterizing synthetic polymers by MALDI MS". Analytical Chemistry 70 (13): 456A–461A. doi:10.1021/ac981910q. PMID 9666717.
- Schriemer, D. C.; Li, L. (1997). "Mass Discrimination in the Analysis of Polydisperse Polymers by MALDI Time-of-Flight Mass Spectrometry. 2. Instrumental Issues". Analytical Chemistry 69 (20): 4176. doi:10.1021/ac9707794.
- Seng, P.; Drancourt, M.; Gouriet, F.; La Scola, B.; Fournier, P. E.; Rolain, J. M.; Raoult, D. (2009). "Ongoing revolution in bacteriology: routine identification of bacteria by matrix-assisted laser desorption ionization time-of-flight mass spectrometry". Clinical Infectious Diseases 49 (4): 552–3. doi:10.1086/600885. PMID 19583519.
- Hillenkamp, F.; Karas, M.; Beavis, R. C.; Chait, B. T. (1991). "Matrix-assisted laser desorption/ionization mass spectrometry of biopolymers". Analytical Chemistry 63 (24): 1193A–1203A. doi:10.1021/ac00024a002. PMID 1789447.
- Ragoussis, J.; Elvidge, G. P.; Kaur, K.; Colella, S. (2006). "Matrix-Assisted Laser Desorption/Ionisation, Time-of-Flight Mass Spectrometry in Genomics Research". PLoS Genetics 2 (7): e100. doi:10.1371/journal.pgen.0020100. PMC 1523240. PMID 16895448.
- Hardouin, J. (2007). "Protein sequence information by matrix-assisted laser desorption/ionization in-source decay mass spectrometry". Mass Spectrometry Reviews 26 (5): 672–82. doi:10.1002/mas.20142. PMID 17492750.
- Peter-Katalinic, J.; Hillenkamp, F. (2007). MALDI MS: A Practical Guide to Instrumentation, Methods and Applications. Wiley-VCH. ISBN 3-527-31440-7.
- Schrepp, W.; Pasch, H. (2003). MALDI-TOF Mass Spectrometry of Synthetic Polymers. Springer-Verlag. ISBN 3-540-44259-6.
- Primer on Matrix-Assisted Laser Desorption Ionization (MALDI) National High Magnetic Field Laboratory