|LOINC||Codes for TEG|
Thromboelastography (TEG) is a method of testing the efficiency of blood coagulation. It is a test mainly utilized in surgery and anesthesiology, although few centers are capable of performing it. More common tests of blood coagulation include prothrombin time (PT,INR) and partial thromboplastin time (aPTT) which measure coagulation factor function, but TEG also can assess platelet function, clot strength, and fibrinolysis which these other tests cannot.
Thromboelastometry (TEM), previously named rotational thromboelastography (ROTEG) or rotational thromboelastometry (ROTEM), is another version of TEG in which it is the sensor shaft, rather than the cup, that rotates.
A small sample of blood is taken from the selected person and rotated gently through 4º 45', six times a minute, to imitate sluggish venous flow and activate coagulation. A thin wire probe is used to measure, which the clot forms around. The speed and strength of clot formation is measured in various ways, typically by computer. The speed at which the sample coagulates depends on the activity of the plasma coagulation system, platelet function, fibrinolysis and other factors which can be affected by genetics, illness, environment and medications. The patterns of changes in strength and elasticity in the clot provide information about how well the blood can perform hemostasis, and how well or poorly different factors are contributing to clot formation.
Four values that represent clot formation are determined by this test: the reaction time (R value), the K value, the angle and the maximum amplitude (MA). The R value represents the time until the first evidence of a clot is detected. The K value is the time from the end of R until the clot reaches 20mm and this represents the speed of clot formation. The angle is the tangent of the curve made as the K is reached and offers similar information to K. The MA is a reflection of clot strength. A mathematical formula determined by the manufacturer can be used to determine a Coagulation Index (CI) (or overall assessment of coagulability) which takes into account the relative contribution of each of these 4 values into 1 equation.
Thromboelastometry (TEM), previously named rotational thromboelastography (ROTEG) or rotational thromboelastometry (ROTEM), is another version of TEG in which it is the sensor shaft, rather than the cup, that rotates. Blood (300 µl, anticoagulated with citrate) is placed into the disposable cuvette using an electronic pipette. A disposable pin is attached to a shaft which is connected with a thin spring (the equivalent to Hartert’s torsion wire in thrombelastography) and slowly oscillates back and forth. The signal of the pin suspended in the blood sample is transmitted via an optical detector system. The test is started by adding appropriate reagents. The instrument measures and graphically displays the changes in elasticity at all stages of the developing and resolving clot. The typical test temperature is 37°C, but different temperatures can be selected, e.g. for patients with hypothermia.
Sonoclot is the latest version of Thromboelastography which takes into account the initial viscosity changes (which typically happens before Fibrin polymerization) and later the elastic changes of the developed clot.
- Trapani, Linda (January 2013). "Thromboelastography: Current Applications, Future Directions". Open Journal of Anesthesiology. doi:10.4236/ojanes.2013.31007.
- "Thrombelastography (TEG®): practical considerations on its clinical use in trauma resuscitation". Scandinavian Journal of Trauma, Resuscitation and Emergency Medicine. doi:10.1186/1757-7241-21-29.
- Donahue SM, Otto CM, Thromboelastography: a tool for measuring hypercoagulability, hypocoagulability, and fibrinolysis, Journal of Veterinary Emergency and Critical Care:15(1), March 2005, Pages: 9-16
- Dirkmann D, Hanke AA, Görlinger K, Peters J. Hypothermia and acidosis synergistically impair coagulation in human whole blood.Anesth Analg. 2008;106:1627-32
- ["Sonoclot Analysis, DA Hett" http://bja.oxfordjournals.org/content/75/6/771.full.pdf ]
- Wenker, Olivier C.; Wojciechowski, Zbigniew; Sheinbaum, Roy; Zisman, Eli. "Thrombelastography". The Internet Journal of Anesthesiology. 2000. Volume 1 Number 3.
- Whitten, Charles W. M.D.; Greilich, Philip E. M.D. "Thromboelastography: Past, Present, and Future". Anesthesiology. 92(5):1226, May 2000.
CBC Mean platelet volumevWF:Ristocetin-induced platelet aggregation clotting factors: other/general: fibrinolysis: Red blood cell indices
CBC ratios: Fetal hemoglobin: Other CFU-GM Other