Clinic of Zaragoza radiotherapy accident
This article needs additional citations for verification. (January 2014)
In the accident, at least 27 patients were injured, and 11 of them died due to the overexposure, according to International Atomic Energy Agency (IAEA). All of the injured were cancer patients receiving external beam radiotherapy.
On 7 December 1990, a technician performed maintenance on an electron accelerator at the Clinic of Zaragoza. On 10 December, it returned to service after the repairs.
Affected patients immediately suffered burns on the skin of the irradiated area, as well as inflammation of the internal organs and bone marrow. The first overexposed patient died on 16 February 1991, two months after irradiation. Fatalities increased until, on 25 December 1991, the last of a total of 25 patients died. The IAEA established that 11 of the deaths were due to the faulty maintenance.
On 19 December, the Nuclear Safety Council (CSN) was scheduled to make its annual review of the device but, due to bureaucratic reasons, this review was delayed. The CSN found the electron accelerator power was too high. The unit was deactivated on 20 December 1990 and restarted on 8 March 1991.
The radiotherapy unit was repaired without following the correct instructions. The unit, in service 14 years at the time of the failure, had a breakdown in the electron beam accelerator control system ("deviator"). Repairs incorrectly increased output power, so patients that should have received therapy at 7 MeV were instead treated at 40 MeV.
- González, Abel J. "Strengthening the Safety of Radiation Sources & the Security of Radioactive Materials: Timely Action". IAEA Bulletin (in English, Chinese, French, Russian, and Spanish). Vol. 41 no. 3. Archived (PDF) from the original on June 8, 2009.
- Argos, Lucía (October 12, 1991). "El accidente del Clínico de Zaragoza, una cadena de fallos humanos única en el mundo, según los expertos" [The accident at the Clinic of Zaragoza, a chain of human failures unique in the world, according to experts]. El País (in Spanish). Archived from the original on October 10, 2012.
- Garriga, Josep; Ortega, Javier (February 23, 1991). "El CSN no controló el aparato radiactivo del Clínico de Zaragoza en un año y medio" [The CSN did not control the radioactive apparatus of the Zaragoza Clinic in a year and a half]. El País (in Spanish). Archived from the original on October 10, 2012.
- Serrano, Sebastian (February 26, 1991). "Un técnico revisó el acelerador del Clínico de Zaragoza tres días antes del accidente" [A technician checked the accelerator of the Zaragoza Clinic three days before the accident]. El País (in Spanish). Archived from the original on October 10, 2012.