The Broselow Tape and associated system
The Broselow Tape, also called the Broselow pediatric emergency tape, is a color-coded length-based tape measure that is used throughout the world for pediatric emergencies. The Broselow Tape relates a child’s height as measured by the tape to his/her weight to provide medical instructions including medication dosages, the size of the equipment that should be used, and the level of shock voltage when using a defibrillator. Particular to children is the need to calculate all these therapies for each child individually. In an emergency the time required to do this detracts from valuable time needed to evaluate, initiate, and monitor patient treatment. The Broselow Tape is designed for children up to approximately 12 years of age who have a maximum weight of roughly 36 kg (80 pounds). The Broselow Tape is recognized in most medical textbooks and publications as a standard for the emergency treatment of children.
To use the Broselow Tape effectively the child must be lying down. Use one hand to hold the red end of the tape so it is even with the child’s head. (Remember: “red to head”). While maintaining one hand on the red portion at the top of the child’s head, use your free hand to run the tape down the length of the child’s body until it is even with his/her heels (not toes). The tape that is level with the child’s heels will provide his/her approximate weight in kilograms and his/her color zone.
As the tape is not completely accurate, care is required with its use.
The Broselow Tape is based on the relationship between weight and length across all ages; each color zone estimates the 50th percentile weight for length, which for practical purposes estimates the ideal body weight (IBW) for emergency dosing. Because of the recent obesity epidemic, concerns have been raised as to the accuracy of the tape to determine acceptable weights and subsequently acceptable doses of emergency medications.
The most recent version of the Broselow Tape incorporates updated length/weight zones based on the most current National Health and Nutrition Examination Survey data set. Utilizing this data set to examine Broselow Tape predictions of actual body weight with the revised zones reveals that approximately 65% of the time the patient’s measured length places them in the correct zone for actual weight. Of the remaining 35%, ~20% fall into the heavier Broselow-Luten zone above and 13% fall into the lighter zone below, with < 1% outliers falling greater than 1 zone from predicted. If the healthcare provider incorporates a visual estimate of body habitus into the prediction, the accuracy of the estimate of actual patient weight is improved as confirmed in multiple studies. Specifically, for drug dosing the patient’s length-based dosing zone can be adjusted up one color zone if the child appears overweight. Thus incorporating a visual estimate of whether the child is over-weight provides a simple method to predict actual patient weight that appears to be clinically relevant given the rise in obesity in the U.S..
Although some medications are best dosed by actual body weight (e.g., succinylcholine), most resuscitation medications are distributed in lean body mass (e.g., epinephrine, sodium bicarbonate, calcium, magnesium, etc.) so that IBW as accurately predicted by length, not the actual body weight, would appear preferable for dosing. For most resuscitation medications, the optimal dose is not known and doses based on IBW or actual weight are likely equally effective.
The PALS guidelines comment on this issue: "There are no data regarding the safety or efficacy of adjusting the doses of resuscitation medications in obese patients. Therefore, regardless of the patient’s habitus, use the actual body weight for calculating initial resuscitationdrug doses or use a body length tape with pre-calculated doses."
Studies on the accuracy of predicting endotracheal tube sizes consistently demonstrate the superiority of length predictions over other methods. Unlike medication dosing, body habitus therefore does not affect the accuracy of the prediction.
- If a child appears overweight consider utilizing one zone higher for dosing only.
- Always use the tape measured length zone for equipment selection regardless of body habitus.
Multiple studies have been conducted regarding the effectiveness of the Broselow Tape. Recently, in 2012, a study with 572 enrolled subjects published that paramedic Broselow measurements correlated well with both scale and ED measurements, underscoring its utility in the prehospital setting. In spite of the debates about accuracy in actual weight estimation and its relevance as noted above, the tape still remains the best tool for predicting actual body weight.
The original Broselow tape was divided into 25 kg zones for medication doses and eight color zones for equipment selection. Subsequent versions of the tape combined dosing and equipment zones such that the eight color zones contained both dosing and equipment information, thus creating a simple visual system for medication and equipment which is used in most hospitals and ambulances.
The following list identifies which color zones correlate with each estimated weight zone in kilograms (kg) and pounds (lbs).
|Grey||3–5 kg||6-11 lbs|
|Pink||6–7 kg||13-15 lbs|
|Red||8–9 kg||17-20 lbs|
|Purple||10–11 kg||22-24 lbs|
|Yellow||12–14 kg||26-30 lbs|
|White||15–18 kg||33-40 lbs|
|Blue||19–23 kg||42-50 lbs|
|Orange||24–29 kg||53-64 lbs|
|Green||30–36 kg||66-80 lbs|
Emergency physicians James Broselow and Robert Luten struggled with solving these issues related to the emergency treatment of children in the early 1980s. The result was the invention by Dr. Broselow of a home-made prototype version of the tape in 1985. Dr. Broselow joined with Dr. Luten, an academic physician from the University of Florida and member of the newly formed Pediatric Advanced Life Support (PALS) subcommittee, to do the foundational studies upon which the tape was based and to develop and update the tape over the years.
The tape provides pre-calculated medication doses effectively eliminating the potential errors associated with pediatric emergent dosing preparation and administration. This benefit has had major implications in recent years given the prevalence and magnitude of medication errors. Medical errors are a greater threat to children than adults because their organs are smaller and still developing. An estimated 35% of pediatric patients are incorrectly dosed by EMS providers. Tenfold mathematical errors due to incorrect calculations are a much greater threat to children than adults. Due to the high level of incorrect calculation errors, alternative pediatric emergency tapes that can be customized have gained popularity. A tenfold adult overdose of a standard adult medication would require multiple syringes and tends, therefore, to be obvious to a caregiver, effectively warning of the error. In contrast, for a small child both a 1x correct dose and a 10x overdose of a drug can be administered in the same syringe thus providing no clue as to a potential error. Furthermore, pediatric emergency care is especially prone to error due to the chaotic nature and stress associated with the emergency setting.
- Luten R, Wears RL, Broselow J, Croskerry P, Joseph MM, Frush K (August 2002). "Managing the unique size-related issues of pediatric resuscitation: reducing cognitive load with resuscitation aids". Academic Emergency Medicine. 9 (8): 840–7. doi:10.1197/aemj.9.8.840. PMID 12153892.
- Google Book Search for Broselow Tape references[unreliable medical source?]
- Frush, Karen. "Study Packet for the Correct Use of the Broselow™ Pediatric Emergency Tape" (PDF). Duke University Medical Center.
- Wells, M; Goldstein, LN; Bentley, A; Basnett, S; Monteith, I (25 September 2017). "The accuracy of the Broselow tape as a weight estimation tool and a drug-dosing guide - A systematic review and meta-analysis". Resuscitation. 121: 9–33. doi:10.1016/j.resuscitation.2017.09.026. PMID 28958796.
- Kleinman ME, Chameides L, Schexnayder SM, et al. (November 2010). "Part 14: pediatric advanced life support: 2010 American Heart Association Guidelines for Cardiopulmonary Resuscitation and Emergency Cardiovascular Care". Circulation. 122 (18 Suppl 3): S876–908. doi:10.1161/CIRCULATIONAHA.110.971101. PMID 20956230.
- Heyming, Theodore; Bosson, Nichole; Kurobe, Aileen; Kaji, Amy H.; Gausche-Hill, Marianne (6 June 2012). "Accuracy of Paramedic Broselow Tape Use in the Prehospital Setting". Prehospital Emergency Care. 16 (3): 374–380. doi:10.3109/10903127.2012.664247.
- Meguerdichian MJ, Clapper TC (August 2012). "The Broselow tape as an effective medication dosing instrument: a review of the literature". Journal of Pediatric Nursing. 27 (4): 416–20. doi:10.1016/j.pedn.2012.04.009. PMID 22579781.
- Luten RC, Wears RL, Broselow J, et al. (August 1992). "Length-based endotracheal tube and emergency equipment in pediatrics". Annals of Emergency Medicine. 21 (8): 900–4. doi:10.1016/S0196-0644(05)82924-5. PMID 1497153.
- Lubitz DS, Seidel JS, Chameides L, Luten RC, Zaritsky AL, Campbell FW (June 1988). "A rapid method for estimating weight and resuscitation drug dosages from length in the pediatric age group". Annals of Emergency Medicine. 17 (6): 576–81. doi:10.1016/S0196-0644(88)80396-2. PMID 3377285.
- Kaufmann, Jost; Laschat, Michael; Wappler, Frank (2016-10-27). "Medication Errors in Pediatric Emergencies". Deutsches Ärzteblatt International. 109 (38): 609–616. doi:10.3238/arztebl.2012.0609. ISSN 1866-0452. PMC 3471264. PMID 23093991.
- Park, Kyung S. (1997). "Human error". In Salvendy, Gavriel. Handbook of human factors and ergonomics (2nd ed.). New York: Wiley. pp. 150–73. ISBN 978-0-471-11690-5.