The free end of the Chest Drainage Device is usually attached to an underwater seal, below the level of the chest. This allows the air or fluid to escape from the pleural space, and prevents anything returning to the chest.
A chest tube (chest drain, thoracic catheter, tube thoracostomy, or intercostal drain) is a flexible plastic tube that is inserted through the chest wall and into the pleural space or mediastinum. It is used to remove air (pneumothorax) or fluid (pleural effusion, blood, chyle), or pus (empyema) from the intrathoracic space. It is also known as a Bülau drain or an intercostal catheter.
The concept of chest drainage was first advocated by Hippocrates when he described the treatment of empyema by means of incision, cautery, and insertion of metal tubes. However, the technique was not widely used until the influenza epidemic of 1917 to drain post-pneumonic empyema, which was first documented by Dr. C. Pope, on "Joel", a 22-month-old infant. The use of chest tubes in postoperative thoracic care was reported in 1922, and they were regularly used post-thoracotomy in World War II, though they were not routinely used for emergency tube thoracostomy following acute trauma until the Korean War.
- Pneumothorax: accumulation of air or gas in the pleural space.
- Pleural effusion: accumulation of fluid in the pleural space
- Chylothorax: a collection of lymphatic fluid in the pleural space
- Empyema: a pyogenic infection of the pleural space
- Hemothorax: accumulation of blood in the pleural space
- Hydrothorax: accumulation of serous fluid in the pleural space
- Postoperative blood evacuation after cardiac surgery, thoracic surgery and trauma surgery to prevent retained blood syndrome
Contraindications to chest tube placement include refractory coagulopathy and presence of a diaphragmatic hernia, as well as hepatic hydrothorax. Additional contraindications include scarring in the pleural space (adhesions).
The most frequent complication associated with chest tubes is chest tube clogging, which is commonly caused by thrombus formation inside the chest tube, and can cause major subsequent complications. After cardiac surgery, chest tube clogging has been observed in 36% of patients, and is significantly associated with increased rates of atrial fibrillation and renal failure. Since a portion of the chest tube is inside the patient, 86% of the time chest tube clogging goes unobserved. In the setting of bleeding, chest tube clogging can cause blood to be retained around the heart and lungs, known as retained blood syndrome, or RBS. This can lead to a complex of complications including pericardial tamponade, hemothorax, pleural effusion, decreased cardiac output, and death. In the presence of air production, chest tube clogging can cause tension pneumothorax, and in the setting of infection, empyema. Chest tube clogging after cardiac surgery is more common in reoperations and urgent procedures, and is associated with statistically significantly higher rates of atrial fibrillation. The composite of complications caused by clogged chest tubes is referred to as Retained Blood Syndrome.
To reduce the potential for chest tube clogging, surgeons often employ larger diameter tubes, which can contribute significantly to chest tube related pain, though these too can clog. Chest tube clogging in channel style drains has also caused life-threatening complications when bleeding in the chest goes unrecognized because of an occluded drain. New evidence suggests that keeping chest tubes free of clogging with an active clearance can reduce Retained Blood Syndrome, the composite of tamponade, hemothorax, and pleural effusions and pericardial effusions as well as postoperative atrial fibrillation.
Major insertion complications include hemorrhage, infection, and reexpansion pulmonary edema. Injury to the liver, spleen or diaphragm is possible if the tube is placed inferior to the pleural cavity. Injuries to the thoracic aorta and heart can also occur.
Minor complications include a subcutaneous hematoma or seroma, anxiety, shortness of breath (dyspnea), and cough (after removing large volume of fluid). In most cases, the chest tube related pain goes away after the chest tube is removed, however, chronic pain related to chest tube induced scarring of the intercostal space is not uncommon.
Subcutaneous emphysema indicates backpressure created by undrained air, often caused by a clogged drain or insufficient negative pressure.
Chest tubes are commonly made from clear plastics like PVC and soft silicone. Chest tubes are made in a range of sizes measured by their external diameter from 6 Fr to 40 Fr. Chest tubes, like most catheters, are measured in French catheter scale. For adults, 20 Fr to 40 Fr (6.7 to 13.3mm external diameter) are commonly used, and 6 Fr to 26 Fr for children. Conventional chest tubes feature multiple drainage fenestrations in the section of the tube which resides inside the patient, as well as distance markers along the length of the tube, and a radiopaque stripe which outlines the first drainage hole. Chest tubes are also provided in right angle, trocar, flared, and tapered configurations for different drainage needs. As well, some chest tubes are coated with heparin to help prevent thrombus formation, though the effect of this is disputed.
Chest tube have an end hole (proximal, toward the patient) and a series of side holes. The number of side holes is generally 6 on most chest tubes. The length of tube that has side holes is the effective drainage length (EDL). In chest tubes designed for pediatric heart surgery, the EDL is shorter, generally by only having 4 side holes. 
Channel style chest drains, also called Blake drains, are so-called silastic drains made of silicone and feature open flutes that reside inside the patient. Drainage is thought to be achieved by capillary action, allowing the fluids to travel through the open grooves into a closed cross section, which contains the fluid and allows it to be suctioned through the tube. Though these chest tubes are more expensive than conventional ones, they are theoretically less painful. No studies have found these to be superior to conventional lumen chest tubes. They are generally better for fluid and serosanguinous drainage, but less effective for blood, especially when it starts to clot, as the channel holes in the transition zone can become clogged leading retained blood syndrome.
Chest drainage canister
A chest drainage canister device is typically used to collect chest drainage (air, blood, effusions). Most commonly, drainage canisters use three chambers which are based on the three-bottle system. The first chamber allows fluid that is drained from the chest to be collected. The second chamber functions as a "water seal", which acts as a one way valve allowing gas to escape, but not reenter the chest. Air bubbling through the water seal chamber is usual when the patient coughs or exhales but may indicate, if continual, a pleural or system leak that should be evaluated critically. It can also indicate a leak of air from the lung. The third chamber is the suction control chamber. The height of the water in this chamber regulates the negative pressure applied to the system. A gentle bubbling through the water column minimizes evaporation of the fluid and indicates that the suction is being regulated to the height of the water column. In this way, increased wall suction does not increase the negative pressure of the system. Newer drainage canisters eliminate the water seal using a mechanical check-valve, and some also use a mechanical regulator to regulate the suction pressure. Systems which employ both these are dubbed "dry" systems, whereas systems that retain the water seal but use a mechanical regulator are called "wet-dry" systems. Systems which use a water seal and water column regulator are called "wet" systems. Dry systems are advantageous as tip-overs of wet systems can spill and mix with blood, mandating the replacement of the canister. Even newer systems are smaller and more ambulatory so the patient can be sent home for drainage if indicated.
The insertion technique for emergency pleural drainage is described in detail in an article of the NEJM. The free end of the tube is usually attached to an underwater seal, below the level of the chest. This allows the air or fluid to escape from the pleural space, and prevents anything returning to the chest. Alternatively, the tube can be attached to a flutter valve. This allows patients with pneumothorax to remain more mobile.
British Thoracic Society recommends the tube is inserted in an area described as the "safe zone", a region bordered by: the lateral border of pectoralis major, a horizontal line inferior to the axilla, the anterior border of latissimus dorsi and a horizontal line superior to the nipple. More specifically, the tube is inserted into the 5th intercostal space slightly anterior to the mid axillary line.
Chest tubes are usually inserted under local anesthesia. The skin over the area of insertion is first cleansed with antiseptic solution, such as iodine, before sterile drapes are placed around the area. The local anesthetic is injected into the skin and down to the muscle, and after the area is numb a small incision is made in the skin and a passage made through the skin and muscle into the chest. The tube is placed through this passage. If necessary, patients may be given additional analgesics for the procedure. Once the tube is in place it is sutured to the skin to prevent it falling out and a dressing applied to the area. Once the drain is in place, a chest radiograph will be taken to check the location of the drain. The tube stays in for as long as there is air or fluid to be removed, or risk of air gathering.
Chest tubes can also be placed using a trocar, which is a pointed metallic bar used to guide the tube through the chest wall. This method is less popular due to an increased risk of iatrogenic lung injury. Placement using the Seldinger technique, in which a blunt guidewire is passed through a needle (over which the chest tube is then inserted) has been described.
The placement technique for postoperative drainage (e.g. cardiac surgery) differs from the technique used for emergency situations. At the completion of open cardiac procedures, chest tubes are placed through separate stab incisions, typically near the inferior aspect of the sternotomy incision. In some instances multiple drains may be used to evacuate the mediastinal, pericardial, and pleural spaces. The drainage holes are placed inside the patient and the chest tube is passed out through the incision. Once the tube is in place, it is sutured to the skin to prevent movement. The chest tube is then connected to the drainage canister using additional tubing and connectors and connected to a suction source, typically regulated to -20 cm of water.
Chest tube management
Chest tubes should be kept free of dependent loops, kinks, and obstructions which may prevent drainage. In general, chest tubes are not clamped except during insertion, removal, or when diagnosing air leaks.
When a chest tube is inserted for whatever reason, maintaining patency is critical to avoid complications due to clogging or clotting that obstruct drainage and can result in retained blood syndrome. Manual manipulation, often called milking, stripping, fan folding, or tapping, of chest tubes is commonly performed to clear chest tube obstructions. No conclusive evidence has demonstrated that any of these techniques are more effective than the others, and no method has shown to improve chest tube drainage. Furthermore, chest tube manipulation has proved to increase negative pressure, which may be detrimental, and painful to the patient. For these reasons, many hospitals do not allow these types of manual tube manipulations.
Internal chest tube clearing can be performed to clear chest tube obstructions using an open or closed technique. Open chest tube clearing involves breaking the sterile environment separating the chest tube from the drainage canister tubing. The internal lumen may then be flushed with saline, or a second catheter may be inserted inside the chest tube and suction used to clear the obstructions. These open techniques can be dangerous as they allow the potential to introduce bacteria, or injury to an internal structure. Closed chest tube clearing without breaking the sterile field is much safer, a technique performed using specially designed drainage systems with active clearance mechanisms. These systems use a magnetically driven wire loop to clear obstructions that form inside the chest tube. New evidence suggests that keeping chest tubes free of clogging with an active clearance using the PleuraFlow device can significantly reduce Retained Blood Syndrome, the composite of tamponade, hemothorax, and pleural effusions and pericardial effusions as well as postoperative atrial fibrillation. This can help reduce these complications in patients undergoing ventricular assist device implantation as well. 
- Noppen, M. (2002). "Manual Aspiration versus Chest Tube Drainage in First Episodes of Primary Spontaneous Pneumothorax: A Multicenter, Prospective, Randomized Pilot Study". American Journal of Respiratory and Critical Care Medicine. 165 (9): 1240–1244. doi:10.1164/rccm.200111-078OC. ISSN 1073-449X.
- Hippocrates (1847). Genuine Works of Hippocrates. Sydenham Society.
- Graham, EA; Bell, RD (1918). "Open Pneumothorax: Its relation to the treatment of empyema". Am J Med Sci. 156: 839–871. doi:10.1097/00000441-191812000-00007.
- Lilienthal, H (1922). "Resection of the lung for supportive infections with a report based on 31 consecutive operative cases in which resection was done or intended". Ann Surg. 75: 257–320. doi:10.1097/00000658-192203000-00001.
- Miller, K S. "Chest tubes. Indications, technique, management and complications.". CHEST Journal. 91 (2): 258. doi:10.1378/chest.91.2.258.
- Runyon BA, Greenblatt M, Ming RH (July 1986). "Hepatic hydrothorax is a relative contraindication to chest tube insertion". The American Journal of Gastroenterology. 81 (7): 566–7. PMID 3717119.
- Hunter, Steven; Angelini, Gianni D. (1993). "Phosphatidylcholine-coated chest tubes improve drainage after open heart operation". The Annals of Thoracic Surgery. 56 (6): 1339–1342. doi:10.1016/0003-4975(93)90678-B.
- Karimov, J. H.; Gillinov, A. M.; Schenck, L.; Cook, M.; Kosty Sweeney, D.; Boyle, E. M.; Fukamachi, K. (21 March 2013). "Incidence of chest tube clogging after cardiac surgery: a single-centre prospective observational study". European Journal of Cardio-Thoracic Surgery. 44: 1029–1036. doi:10.1093/ejcts/ezt140. PMID 23520232.
- Dixon, Barry; John Santamaria; et al. (2012). "The association of blood transfusion with mortality after cardiac surgery: cause or confounding?". Transfusion. 53: 19–27. doi:10.1111/j.1537-2995.2012.03697.x.
- Shalli S, Saeed D, Fukamachi K, et al. (2009). "Chest tube selection in cardiac and thoracic surgery: a survey of chest tube-related complications and their management". J Card Surg. 24 (5): 503–9. doi:10.1111/j.1540-8191.2009.00905.x. PMID 19740284.
- Clark G, Licker M, Bertin D, Spiliopoulos A (March 2007). "Small size new silastic drains: life-threatening hypovolemic shock after thoracic surgery associated with a non-functioning chest tube". Eur J Cardiothorac Surg. 31 (3): 566–8. doi:10.1016/j.ejcts.2006.12.010. PMID 17215136.
- Sirch, J; Ledwon, M; et al. (2015). "Active clearance of chest drainage catheters reduces retained blood". Journal of Thoracic and Cardiovascular Surgery. 151: 832–8.e1–2. doi:10.1016/j.jtcvs.2015.10.015. PMID 26611748.
- "Archived copy". Archived from the original on 2010-04-08. Retrieved 2009-07-19.
- Kumar, P; D McKee; et al. (1997). "Phosphatidylcholine coated chest drains: are they better than conventional drains after open heart surgery?". European Journal of Cardio-thoracic Surgery. 11: 769–733. doi:10.1016/s1010-7940(96)01145-1.
- Obney, James; Mary Barnes; et al. (2000). "A method for mediastinal drainage after cardiac procedures using small silastic drains". The Annals of Thoracic Surgery. 70: 1109–110. doi:10.1016/s0003-4975(00)01800-2.
- Frankel, T. L. "Silastic Drains vs Conventional Chest Tubes After Coronary Artery Bypass". Chest. 124 (1): 108–113. doi:10.1378/chest.124.1.108.
- Dev SP, Nascimiento B, Simone C, Chien V (October 2007). "Videos in clinical medicine. Chest-tube insertion". N. Engl. J. Med. 357 (15): e15. doi:10.1056/NEJMvcm071974. PMID 17928590.
- Laws, D. "BTS guidelines for the insertion of a chest drain". Thorax. 58 (90002): 53ii–59. doi:10.1136/thorax.58.suppl_2.ii53.
- "Archived copy". Archived from the original on 2009-06-17. Retrieved 2009-09-16.
- Schmelz, JO; Johnson, D; Norton, JM; Andrews, M; Gordon, PA (September 1999). "Effects of position of chest drainage tube on volume drained and pressure.". American Journal of Critical Care. 8 (5): 319–23. PMID 10467469.
- Halejian, BA; Badach, MJ; Trilles, F (December 1988). "Maintaining chest tube patency.". Surgery, gynecology & obstetrics. 167 (6): 521. PMID 3187876.
- Wallen, Margaret; Anne Morrison; et al. (2008). "Mediastinal chest drain clearance for cardiac surgery". The Cochrane Library (3): 2–27. doi:10.1002/14651858.CD003042.pub2.
- Shiose, A.; Takaseya, T.; Fumoto, H.; Arakawa, Y.; Horai, T.; Boyle, E. M.; Gillinov, A. M.; Fukamachi, K. (May 2010). "Improved drainage with active chest tube clearance". Interactive CardioVascular and Thoracic Surgery. 10 (5): 685–688. doi:10.1510/icvts.2009.229393. PMID 20179137.
- Maltais S, Davis ME, Haglund NA, Perrault L, Kushwaha SS, Stulak JM, Boyle EM. Active Clearance of Chest Tubes Reduces Re-Exploration for Bleeding After Ventricular Assist Device Implantation. ASAIO J. 2016 Nov/Dec;62(6):704-709. doi.org/10.1097/MAT.0000000000000437 http://journals.lww.com/asaiojournal/Fulltext/2016/11000/Active_Clearance_of_Chest_Tubes_Reduces.12.aspx
- Catheter drainage is used for empyemas after chest-tube failure: vanSonnenberg, E; S K Nakamoto; P R Mueller; G Casola; C C Neff; P J Friedman; J T Ferrucci, Jr; J F Simeone (May 1984). "CT- and ultrasound-guided catheter drainage of empyemas after chest-tube failure" (PDF). Radiology. 151 (2): 349–353. doi:10.1148/radiology.151.2.6709904.
- Commonly used after Percutaneous CT-Guided Lung Biopsies: Saji, H. (2002). "The Incidence and the Risk of Pneumothorax and Chest Tube Placement After Percutaneous CT-Guided Lung Biopsy* : The Angle of the Needle Trajectory Is a Novel Predictor". Chest. 121 (5): 1521–1526. doi:10.1378/chest.121.5.1521. ISSN 0012-3692.
|Wikimedia Commons has media related to Chest drain.|