Contrast resolution is the ability to distinguish between differences in intensity in an image. The measure is used in medical imaging to quantify the quality of acquired images. It is a difficult quantity to define, because it depends on the human observer as much as the quality of the actual image. For example, the size of a feature affects how easily it is detected by the observer.
One definition of image contrast is:
where SA and SB are signal intensities for signal producing structures A and B in the region of interest. A disadvantage of this definition is that the contrast C can be negative. An alternative definition is:
where Sref is a reference signal intensity, which is independent of the type of signal producing structure under investigation.
In MRI, determining contrast is of high importance for calibration because the operator has a high degree of control of how the signal intensities of various structures vary in the images by using different MRI methods and imaging parameters. Unlike most other imaging modalities, such as x-ray CT in which the Hounsfield units value for water is set to zero, there is no standard reference signal for MRI. Thus the contrast-to-noise ratio is often employed as an index for contrast because this metric does not require a reference signal.
Contrast resolution or contrast-detail is an approach to describing the image quality in terms of both the image contrast and resolution.
Contrast resolution is usually measured by generating a pattern from a test object that depicts how image contrast changes as the structures being imaged get smaller and closer together. The picture below shows one such set of images produced using the low contrast detectability inserts of the phantom employed in the MRI accreditation program of the American College of Radiology.
Contrast detail curves, which depict the functional relationship between image resolution and contrast resolution, can be produced using this type of data.
- The encyclopaedia of medical imaging. ISBN 1-901865-13-4.
- Imaging Systems for Medical Diagnostics: Fundamentals, Technical Solutions and Applications for Systems Applying Ionizing Radiation, Nuclear Magnetic Resonance and Ultrasound. p. 44. ISBN 3-89578-226-2.
- Hendrick, R. E.; Raff, U. (1992). Stark, David D.; Bradley, William G., eds. Magnetic Resonance Imaging (2nd ed.). St. Louis: Mosby. pp. 109–144.
- Edelstein, W. A.; Bottomley, P. A.; Hart, H. R.; Smith, L. S. (1983). "Signal, noise and contrast in nuclear magnetic resonance (NMR) imaging". Journal of Computer Assisted Tomography. 7 (3): 391–401. doi:10.1097/00004728-198306000-00001.
|This article related to medical equipment is a stub. You can help Wikipedia by expanding it.|