Desorption atmospheric pressure photoionization
Desorption atmospheric pressure photoionization (DAPPI) is an ambient ionization technique for mass spectrometry that uses hot solvent vapor for desorption in conjunction with photoionization. Ambient Ionization techniques allow for direct analysis of samples without pretreatment. The direct analysis technique, such as DAPPI, eliminates the extraction steps seen in most nontraditional samples. DAPPI can be used to analyze bulkier samples, such as, tablets, powders, resins, plants, and tissues. The first step of this technique utilizes a jet of hot solvent vapor. The hot jet thermally desorbs the sample from a surface. The vaporized sample is then ionized by the vacuum ultraviolet light and consequently sampled into a mass spectrometer. DAPPI can detect a range of both polar and non-polar compounds, but is most sensitive when analyzing neutral or non-polar compounds. This technique also offers a selective and soft ionization for highly conjugated compounds.
The history of desorption atmospheric pressure photoionization is relatively new, but can be traced back through developments of ambient ionization techniques dating back to the 1970s. DAPPI is a combination of popular techniques, such as, atmospheric pressure photoionziation (APPI) and surface desorption techniques. The photoionization techniques were first developed in the late 1970s and began being used in atmospheric pressure experiments in the mid 1980s. Early developments in the desorption of open surface and free matrix experiements were first reported in literature in 1999 in an experiment using desorption/ionization on silicon (DIOS). DAPPI replaced techniques such as desorption electrospray ionization (DESI) and direct analysis in real time (DART). This generation of techniques are all recent developments seen in the 21st century. DESI was discovered in 2004 at Purdue University, while DART was discovered in 2005 by Laramee and Cody. DAPPI was developed soon after in 2007 at the University of Helsinki, Finland. The development of DAPPI widened the range of detection for nonpolar compounds and added a new dimension of thermal desorption of direct analysis samples.
Principle of operation
The first operation to occur during desorption atmospheric pressure photoionization is desorption. Desorption of the sample is initiated by a hot jet of solvent vapor that is targeted onto the sample by a nebulizer microchip. The nebulizer microchip is a glass device bonded together by pyrex wafers with flow channels embedded from a nozzle at the edge of the chip. The microchip is heated to 250-350C in order to vaporize the entering solvent and create dopant molecules. Dopant molecules are added to help facilitate the ionization of the sample. Some of the common solvents include: nitrogen, toluene, acetone, and anisole. The desorption process can occur by two mechanisms: thermal desorption or momentum transfer/liquid spray. Thermal desorption uses heat to volatilize the sample and increase the surface temperature of the substrate. As the substrate's surface temperature is increased, the higher the sensitivity of the instrument. While studying the substrate temperature, it was seen that the solvent did not have a noticeable effect on the final temperature or heat rate of the substrate. Momentum transfer or liquid spray desoprtion is based on the solvent interaction with the sample, causing the release of specific ions. The momentum transfer is propagated by the collision of the solvent with the sample along with the transfer of ions with the sample. The transfer of positive ions, such as protons and charge transfers, are seen with the solvents: toluene and anisole. Toluene goes through a charge exchange mechanism with the sample, while acetone promotes a proton transfer mechanism with the sample. A beam of 10 eV photons that are given off by a UV lamp is directed at the newly desorbed molecules, as well as the dopant molecules. Photoionization then occurs, which knocks out the molecule's electron and produces an ion. This technique alone is not highly efficient for different varieties of molecules, particularly those that are not easily protonated or deprotonated. In order to completely ionize samples, dopant molecules must help. The gaseous solvent can also undergo photoionization and act as an intermediate for ionization of the sample molecules. Once dopant ions are formed, proton transfer can occur with the sample, creating more sample ions. The ions are then sent to the mass analyzer for analysis.
The main desorption mechanism in DAPPI is thermal desorption due to rapid heating of the surface. Therefore, DAPPI only works well for surfaces of low thermal conductivity. The ionization mechanism depends on the analyte and solvent used. For example, the following analyte (M) ions may be formed: [M + H]+, [M - H]−, M+•, M−•.
Types of component geometries
Considered the normal or conventional geometry of DAPPI, this mode is ideal for solid samples that do not need any former preparation. The microchip is parallel to the MS inlet. The microchip heater is aimed to hit the samples at . The UV lamp is directly above the sample and it releases photons to interact with the desorbed molecules that are formed. The conventional method generally uses a higher heating power and gas flow rate for the nebulizer gas, while also increasing the amount of dopant used during the technique. These increases can cause higher background noise, analyte interference, substrate impurities, and more ion reactions from excess dopant ions.
This mode is specialized for analyzing liquid samples, with a metal or polymer mesh replacing the sample plate in reflection geometry. The mesh is oriented from the nebulizer microchip and the mass spec inlet, with the lamp directing photons to the area where the mesh releases newly desorbed molecules. The analyte is thermally desorbed as both the dopant vapor and nebulizer gas are directed through the mesh. It has been seen that steel mesh with low density and narrow strands produces better signal intensities. This type of mesh allows for larger openings in the surface and quicker heating of strands. Transmission mode uses a lower microchip heating power which eliminates some of the issues seen with the reflection geometry above, including low signal noise. This method can also improve the S/N ratio of smaller non-polar compounds.
Thin layer chromatography (TLC) is a simple separation technique that can be coupled with DAPPI-MS to identify lipids. Some of the lipids that were seen to be separated and ionized include: cholesterol, triacylglycerols, 1,2-diol diesters, wax esters, hydrocarbons, and cholesterol esters. TLC is normally coupled with instruments in vacuum or atmospheric pressure, but vacuum pressure gives poor sensitivity for more volatile compounds and has minimal area in the vacuum chambers. DAPPI was used for its ability to ionize neutral and non-polar compounds, and was seen to be a fast and efficient method for lipid detection as it was coupled with both NP-TLC and HPTLC plates.
Laser desorption is normally used in the presence of a matrix, such as matrix assisted laser desorption ionization (MALDI), but research has combined techniques of laser desoprtion in atmospheric pressure conditions to produce a method that does not use a matrix or discharge. This method is able to help with smaller compounds, and generates both positive and negative ions for detection. A transmission geometry is taken as the beam and spray are guided at a angle into the coupled MS. Studies have shown the detection of organic compounds such as: farnesene, squalene, tetradecahydro anthracene, 5-alpha cholestane, perylene, benzo perylene, coronene, tetra decylprene, dodecyl sulfide, benzodiphenylene sulfide, dibenzosuberone, carbazole, and elipticine. This method was also seen to be coupled with the mass spectroscopy technique, FTICR, to detect shale oils and some smaller nitrogen containing aromatics.
Fourier transform ion cyclotron resonance (FTICR) is a technique that is normally coupled with electrospray ionization (ESI), DESI, or DART, which allows for the detection of polar compounds. DAPPI allows for a broader range of polarities to be detected, and a range of molecular weights. Without separation or sample preparation, DAPPI is able to thermally desorb compounds such as oak biochars. The study did cite an issue with DAPPI. If the sample is not homogeneous, then the neutral ions will ionize only the surface, which does not provide an accurate detection for the substance. The scanning of the FTICR allows for the detection of complex compounds with high resolution, which leads to the ability to analyze elemental composition.
DAPPI can analyze both polar (e.g. verapamil) and nonpolar (e.g. anthracene) compounds. This technique has an upper detection limit of 600 Da. Compared to desorption electrostray ionization (DESI), DAPPI is less likely to be contaminated by biological matrices. DAPPI was also seen to be more sensitive and contain less background noise than popular techniques such as direct analysis in real time (DART). Performance of DAPPI has also been demonstrated on direct analysis of illicit drugs. Other applications include lipid detection and drug analysis sampling. Lipids are can be detected through a coupling procedure with orbitrap mass spectroscopy. DAPPI has also been known to couple with liquid chromotography and gas chromotography mass spectroscopy for the analysis of drugs and aerosol compounds. Studies have also shown where DAPPI has been used to find harmful organic compounds in the environment and in food, such as polycyclic aromatic hydrocarbons (PAH) and pesticides.
- Atmospheric pressure chemical ionization
- Desorption atmospheric pressure chemical ionization
- Haapala M, Pól J, Saarela V, Arvola V, Kotiaho T, Ketola RA, Franssila S, Kauppila TJ, Kostiainen R (2007). "Desorption Atmospheric Pressure Photoionization". Anal. Chem. 79 (20): 7867–7872. doi:10.1021/ac071152g. PMID 17803282.
- Huang, Min-Zong; Cheng, Sy-Chi; Cho, Yi-Tzu; Shiea, Jentaie (2011-09-19). "Ambient ionization mass spectrometry: A tutorial". Analytica Chimica Acta. 702 (1): 1–15. doi:10.1016/j.aca.2011.06.017.
- Kauppila TJ, Arvola V, Haapala M, Pól J, Aalberg L, Saarela V, Franssila S, Kotiaho T, Kostiainen R (2008). "Direct analysis of illicit drugs by desorption atmospheric pressure photoionization". Rapid Commun. Mass Spectrom. 22 (7): 979–985. doi:10.1002/rcm.3461. PMID 18320545.
- Weston, Daniel J. (2010-03-22). "Ambient ionization mass spectrometry: current understanding of mechanistic theory; analytical performance and application areas". The Analyst. 135 (4). doi:10.1039/b925579f. ISSN 1364-5528.
- Vestal, Marvin L. (2001-02-01). "Methods of Ion Generation". Chemical Reviews. 101 (2): 361–376. doi:10.1021/cr990104w. ISSN 0009-2665.
- Raffaelli, Andrea; Saba, Alessandro (2003-09-01). "Atmospheric pressure photoionization mass spectrometry". Mass Spectrometry Reviews. 22 (5): 318–331. doi:10.1002/mas.10060. ISSN 1098-2787.
- Buriak, Jillian M.; Wei, Jing; Siuzdak, Gary. "http://www.nature.com/doifinder/10.1038/20400". Nature. 399 (6733): 243–246. doi:10.1038/20400. External link in
- Takáts, Zoltán; Wiseman, Justin M.; Gologan, Bogdan; Cooks, R. Graham (2004-10-15). "Mass Spectrometry Sampling Under Ambient Conditions with Desorption Electrospray Ionization". Science. 306 (5695): 471–473. doi:10.1126/science.1104404. ISSN 0036-8075. PMID 15486296.
- Ambient Ionization Mass Spectrometry. 2014-11-21. doi:10.1039/9781782628026. ISBN 9781849739269.
- Chen, Huanwen; Gamez, Gerardo; Zenobi, Renato (2009-11-01). "What can we learn from ambient ionization techniques?". Journal of the American Society for Mass Spectrometry. 20 (11): 1947–1963. doi:10.1016/j.jasms.2009.07.025. ISSN 1044-0305.
- Saarela, Ville; Haapala, Markus; Kostiainen, Risto; Kotiaho, Tapio; Franssila, Sami (2007-05-02). "Glass microfabricated nebulizer chip for mass spectrometry". Lab on a Chip. 7 (5). doi:10.1039/b700101k. ISSN 1473-0189.
- Harris, Glenn A.; Galhena, Asiri S.; Fernández, Facundo M. (2011-06-15). "Ambient Sampling/Ionization Mass Spectrometry: Applications and Current Trends". Analytical Chemistry. 83 (12): 4508–4538. doi:10.1021/ac200918u. ISSN 0003-2700.
- Ifa, Demian R.; Jackson, Ayanna U.; Paglia, Giuseppe; Cooks, R. Graham (2009-08-01). "Forensic applications of ambient ionization mass spectrometry". Analytical and Bioanalytical Chemistry. 394 (8): 1995–2008. doi:10.1007/s00216-009-2659-2. ISSN 1618-2642.
- Parshintsev, Jevgeni; Vaikkinen, Anu; Lipponen, Katriina; Vrkoslav, Vladimir; Cvačka, Josef; Kostiainen, Risto; Kotiaho, Tapio; Hartonen, Kari; Riekkola, Marja-Liisa (2015-07-15). "Desorption atmospheric pressure photoionization high-resolution mass spectrometry: a complementary approach for the chemical analysis of atmospheric aerosols". Rapid Communications in Mass Spectrometry. 29 (13): 1233–1241. doi:10.1002/rcm.7219. ISSN 1097-0231.
- Venter, Andre; Nefliu, Marcela; Graham Cooks, R. (2008-04-01). "Ambient desorption ionization mass spectrometry". TrAC Trends in Analytical Chemistry. 27 (4): 284–290. doi:10.1016/j.trac.2008.01.010.
- Ding, Xuelu; Duan, Yixiang (2015-07-01). "Plasma-based ambient mass spectrometry techniques: The current status and future prospective". Mass Spectrometry Reviews. 34 (4): 449–473. doi:10.1002/mas.21415. ISSN 1098-2787.
- D., Lin, C. (1993-01-01). Review of fundamental processes and applications of atoms and ions. World Scientific Publ. ISBN 9810215371. OCLC 832685134.
- Robb, Damon B.; Blades, Michael W. (2008-10-03). "State-of-the-art in atmospheric pressure photoionization for LC/MS". Analytica Chimica Acta. Mass Spectrometry. 627 (1): 34–49. doi:10.1016/j.aca.2008.05.077.
- Van Berkel, Gary J.; Pasilis, Sofie P.; Ovchinnikova, Olga (2008-09-01). "Established and emerging atmospheric pressure surface sampling/ionization techniques for mass spectrometry". Journal of Mass Spectrometry. 43 (9): 1161–1180. doi:10.1002/jms.1440. ISSN 1096-9888.
- Luosujärvi, Laura; Laakkonen, Ulla-Maija; Kostiainen, Risto; Kotiaho, Tapio; Kauppila, Tiina J. (2009-05-15). "Analysis of street market confiscated drugs by desorption atmospheric pressure photoionization and desorption electrospray ionization coupled with mass spectrometry". Rapid Communications in Mass Spectrometry. 23 (9): 1401–1404. doi:10.1002/rcm.4005. ISSN 1097-0231.
- Luosujärvi L, Arvola V, Haapala M, Pól J, Saarela V, Franssila S, Kotiaho T, Kostiainen R, Kauppila TJ (2008). "Desorption and Ionization Mechanisms in Desorption Atmospheric Pressure Photoionization". Anal. Chem. 80 (19): 7460–7466. doi:10.1021/ac801186x. PMID 18778037.
- Harris, Glenn A.; Nyadong, Leonard; Fernandez, Facundo M. (2008-09-09). "Recent developments in ambient ionization techniques for analytical mass spectrometry". The Analyst. 133 (10). doi:10.1039/b806810k. ISSN 1364-5528.
- Vaikkinen, Anu; Hannula, Juha; Kiiski, Iiro; Kostiainen, Risto; Kauppila, Tiina J. (2015-04-15). "Transmission mode desorption atmospheric pressure photoionization". Rapid Communications in Mass Spectrometry. 29 (7): 585–592. doi:10.1002/rcm.7139. ISSN 1097-0231.
- Rejšek, Jan; Vrkoslav, Vladimír; Vaikkinen, Anu; Haapala, Markus; Kauppila, Tiina J.; Kostiainen, Risto; Cvačka, Josef (2016-12-20). "Thin-Layer Chromatography/Desorption Atmospheric Pressure Photoionization Orbitrap Mass Spectrometry of Lipids". Analytical Chemistry. 88 (24): 12279–12286. doi:10.1021/acs.analchem.6b03465. ISSN 0003-2700.
- F., Poole, Colin (2015-01-01). Instrumental thin-layer chromatography. Elsevier. ISBN 9780124172234. OCLC 897437460.
- Han, Yehua; Levkin, Pavel; Abarientos, Irene; Liu, Huwei; Svec, Frantisek; Fréchet, Jean M. J. (2010-03-15). "Monolithic Superhydrophobic Polymer Layer with Photopatterned Virtual Channel for the Separation of Peptides Using Two-Dimensional Thin Layer Chromatography-Desorption Electrospray Ionization Mass Spectrometry". Analytical Chemistry. 82 (6): 2520–2528. doi:10.1021/ac100010h. ISSN 0003-2700. PMC . PMID 20151661.
- Nyadong, Leonard; Mapolelo, Mmilili M.; Hendrickson, Christopher L.; Rodgers, Ryan P.; Marshall, Alan G. (2014-11-18). "Transmission Geometry Laser Desorption Atmospheric Pressure Photochemical Ionization Mass Spectrometry for Analysis of Complex Organic Mixtures". Analytical Chemistry. 86 (22): 11151–11158. doi:10.1021/ac502138p. ISSN 0003-2700.
- Nyadong, Leonard; McKenna, Amy M.; Hendrickson, Christopher L.; Rodgers, Ryan P.; Marshall, Alan G. (2011-03-01). "Atmospheric Pressure Laser-Induced Acoustic Desorption Chemical Ionization Fourier Transform Ion Cyclotron Resonance Mass Spectrometry for the Analysis of Complex Mixtures". Analytical Chemistry. 83 (5): 1616–1623. doi:10.1021/ac102543s. ISSN 0003-2700.
- Cho, Yunju; Jin, Jang Mi; Witt, Matthias; Birdwell, Justin E.; Na, Jeong-Geol; Roh, Nam-Sun; Kim, Sunghwan (2013-04-18). "Comparing Laser Desorption Ionization and Atmospheric Pressure Photoionization Coupled to Fourier Transform Ion Cyclotron Resonance Mass Spectrometry To Characterize Shale Oils at the Molecular Level". Energy & Fuels. 27 (4): 1830–1837. doi:10.1021/ef3015662. ISSN 0887-0624.
- Podgorski, David C.; Hamdan, Rasha; McKenna, Amy M.; Nyadong, Leonard; Rodgers, Ryan P.; Marshall, Alan G.; Cooper, William T. (2012-02-07). "Characterization of Pyrogenic Black Carbon by Desorption Atmospheric Pressure Photoionization Fourier Transform Ion Cyclotron Resonance Mass Spectrometry". Analytical Chemistry. 84 (3): 1281–1287. doi:10.1021/ac202166x. ISSN 0003-2700.
- Suni, Niina M.; Lindfors, Pia; Laine, Olli; Östman, Pekka; Ojanperä, Ilkka; Kotiaho, Tapio; Kauppila, Tiina J.; Kostiainen, Risto (2011-08-05). "Matrix effect in the analysis of drugs of abuse from urine with desorption atmospheric pressure photoionization-mass spectrometry (DAPPI-MS) and desorption electrospray ionization-mass spectrometry (DESI-MS)". Analytica Chimica Acta. 699 (1): 73–80. doi:10.1016/j.aca.2011.05.004.
- Räsänen, Riikka-Marjaana; Dwivedi, Prabha; Fernández, Facundo M.; Kauppila, Tiina J. (2014-11-15). "Desorption atmospheric pressure photoionization and direct analysis in real time coupled with travelling wave ion mobility mass spectrometry". Rapid Communications in Mass Spectrometry. 28 (21): 2325–2336. doi:10.1002/rcm.7028. ISSN 1097-0231.
- Kauppila, Tiina J.; Syage, Jack A.; Benter, Thorsten (2015-05-01). "Recent developments in atmospheric pressure photoionization-mass spectrometry". Mass Spectrometry Reviews: n/a–n/a. doi:10.1002/mas.21477. ISSN 1098-2787.
- Luosujärvi, Laura; Kanerva, Sanna; Saarela, Ville; Franssila, Sami; Kostiainen, Risto; Kotiaho, Tapio; Kauppila, Tiina J. (2010-05-15). "Environmental and food analysis by desorption atmospheric pressure photoionization-mass spectrometry". Rapid Communications in Mass Spectrometry. 24 (9): 1343–1350. doi:10.1002/rcm.4524. ISSN 1097-0231.
- Ifa, Demian R.; Wu, Chunping; Ouyang, Zheng; Cooks, R. Graham (2010-03-22). "Desorption electrospray ionization and other ambient ionization methods: current progress and preview". The Analyst. 135 (4). doi:10.1039/b925257f. ISSN 1364-5528.