Electron beam ion trap
This article includes a list of references, but its sources remain unclear because it has insufficient inline citations. (July 2014) (Learn how and when to remove this template message)
Electron beam ion trap (EBIT) is an electromagnetic bottle that produces and confines highly charged ions. An EBIT uses an electron beam focused with a powerful magnetic field to ionize atoms to high charge states by successive electron impact.
The positive ions produced in the region where the atoms intercept the electron beam are tightly confined in their motion by the strong attraction exerted by the negative charge of the electron beam. Therefore, they orbit around the electron beam, crossing it frequently and giving rise to further collisions and ionization. To restrict the ion motion along the direction of the electron beam axis, trapping electrodes carrying positive voltages with respect to a central electrode are used.
The strong charge needed for radial confinement of the ions requires large electron beam currents of tens up to hundreds of milliampere. At the same time, high voltages (up to 200 kilovolts) are used for accelerating the electrons in order to achieve high charge states of the ions.
To avoid charge reduction of ions by collisions with neutral atoms from which they can capture electrons, the vacuum in the apparatus is usually maintained at UHV levels, with typical pressure values of only 10−12 torr, (~10−10 pascal).
EBITs are used to investigate the fundamental properties of highly charged ions e. g. by photon spectroscopy in particular in the context of relativistic atomic structure theory and quantum electrodynamics (QED). Their suitability to prepare and reproduce in a microscopic volume the conditions of high temperature astrophysical plasmas and magnetic confinement fusion plasmas make them very appropriate research tools. Other fields include the study of their interactions with surfaces and possible applications to microlithography.
- Levine, M. A., et al., The Electron Beam Ion Trap: A New Instrument for Atomic Physics Measurements, Physica Scripta, T22, p. 157 (1988)
- Marrs, R. E., et al., "Production and Trapping of Hydrogenlike and Bare Uranium Ions in an Electron Beam Ion Trap", Phys. Rev. Lett., 72, p. 4082 (1994)
- Roscoe E. Marrs, Peter Beiersdorfer, and Dieter Schneider, Physics Today, 27 (October 1994) – Description of the Electron Beam Ion Trap
- R. E. Marrs, M. A. Levine, D. A. Knapp, J. R. Henderson, Phys. Rev. Lett. 60, 1715 (1988) – First EBIT atomic spectroscopy measurement
- C. A. Morgan et al., Observation of Visible and UV Magnetic Dipole Transitions in Highly-Charged Xenon and Barium, Phys. Rev. Lett. 74, 1716 (1995)
- H. P. Cheng, J. D. Gillaspy, Nanoscale modification of silicon surfaces via Coulomb explosion, Phys. Rev. B 55, 2628 (1997)
- J. D. Gillaspy, D. C. Parks, L. P. Ratliff, Masked ion beam lithography with highly charged ions, Journal of Vacuum Science and Technology B, 16, 3294 (1998)
- F. J. Currell et al., A new versatile electron-beam ion trap, Journal of the Physical Society of Japan 65, 3186 (1996)
- H. F. Beyer, H.-J. Kluge, V. P. Shevelko, in: X-ray Radiation of Highly Charged Ions, Springer Series on Atoms and Plasmas, Springer Verlag, Berlin-Heidelberg 1997
- "EBIT". National Institute of Standards and Technology. 7 December 2010. Retrieved 26 November 2012.
- "Electron Beam Ion Trap (EBIT)". Lawrence Livermore National Laboratory. 14 April 2009. Retrieved 26 November 2012.
- "Other EBITs Around the Globe". National Institute of Standards and Technology. 14 December 2011. Retrieved 26 November 2012.