His collimator was meant to replace a level, or plumb-line, in astronomical observations, and to furnish a ready and perfectly exact method of determining the position of the horizontal or zenith point on the limb of a circle or zenith sector. Its principle is the invariability, with respect to the horizon, of the position assumed by any body of invariable figure and weight floating on a fluid.
Floating collimator consists of a rectangular box containing mercury, on which is floated a mass of cast-iron about twelve inches long, four broad, and half an inch thick, having two short uprights or Y's of equal height, cast in one piece with the rest. On these is firmly attached a small telescope, furnished with cross wires, or, what is better, crossed portions of the fine balance-spring of a watch, set flat-ways, and adjusted very exactly in the sidereal focus of object glass.
The float is browned with nitric acid to prevent the adhesion of the mercury, and is prevented from moving laterally by two smoothly polished iron pins, projecting from its sides in the middle of its length, which play freely in vertical grooves of polished iron in the sides of the box.
When this instrument is used, it is placed at a short distance from the circle whose horizontal point is to be ascertained, on either side, suppose the north, of its centre, and the telescopes of the circle and of the collimator are so adjusted as to look mutually at each other's cross wires, in the manner practised by astronomers Gauss and Friedrich Bessel, first of all coarsely, by trial, applying the eye to the eyeglasses of the two instruments alternately; and, finally, by illuminating the cross wires of the collimator by a lantern and oiled paper, taking care to exclude false light by a black screen, having an aperture equal to that of the collimator, and making the coincidence in the manner of an astronomical observation, by the fine motion of the circle.
The microscopes on the limb are then read off, and thus the apparent zenith distance of the collimating point, intersection of the wires, is found. The collimator is then transferred to the other (south) side of the circle, and a corresponding observation made without reversing the circle, but merely by the motion of the telescope on the limb. The difference of the two zenith distances so read off is double the error of the zenith or horizontal point of the graduation, and their semi-sum is the true zenith distance of the collimating point, or the co-inclination of the axis of the collimating telescope to the horizon.
By the experiments detailed in captain Kater's paper, read before the Royal Society in 1825, it appears that the error to be feared in the determination of the horizontal point by this instrument, can rarely amount to half a second, if a mean of four or five observations be taken. In 151 single trials, two only gave an error of two seconds, and one of these was made with a wooden float.
- The Description of a Floating Collimator. By Captain Henry Kater. Read January 13, 1825. [Phil. Trans. 1825, p. 147.]
This article is based on the item in The London encyclopaedia: or, Universal dictionary of science, art, literature, and practical mechanics, Volume 6, Printed for Thomas Tegg, 1829. (Google eBook)