This article needs additional citations for verification. (June 2012) (Learn how and when to remove this template message)
The thermal effects of the incident IR radiation can be followed through many temperature dependent phenomena. Bolometers and microbolometers are based on changes in resistance. Thermocouples and thermopiles use the thermoelectric effect. Golay cells follow thermal expansion. In IR spectrometers the pyroelectric detectors are the most widespread.
The response time and sensitivity of photonic detectors can be much higher, but usually these have to be cooled to cut thermal noise. The materials in these are semiconductors with narrow band gaps. Incident IR photons can cause electronic excitations. In photoconductive detectors, the resistivity of the detector element is monitored. Photovoltaic detectors contain a p-n junction on which photoelectric current appears upon illumination.
A few detector materials:
- Mercury cadmium telluride (MCT)
- Indium antimonide
- Indium arsenide
- Lead selenide
- Lithium tantalate (LiTaO3)
- Triglycine sulfate (TGS)
- "Revolutionary New High-speed Infrared Detector Sees First Light". Retrieved 15 June 2015.