A load cell is a type of force gauge. It consists of a transducer that is used to create an electrical signal whose magnitude is directly proportional to the force being measured. The various load cell types include hydraulic, pneumatic, and strain gauge.
- 1 Strain gauge load cell
- 2 Piezoelectric load cell
- 3 Hydraulic load cell
- 4 Pneumatic load cell
- 5 Other types
- 6 Ringing
- 7 Uses
- 8 Load cells weighing performances
- 9 Load cell calibration
- 10 See also
- 11 References
Strain gauge load cell
Strain gauge load cells are the most common in industry. These load cells are particularly stiff, have very good resonance values, and tend to have long life cycles in application. Strain gauge load cells work on the principle that the strain gauge (a planar resistor) deforms when the material of the load cells deforms appropriately. Deformation of the strain gauge changes its electrical resistance, by an amount that is proportional to the strain. The change in resistance of the strain gauge provides an electrical value change that is calibrated to the load placed on the load cell.
A load cell usually consists of four strain gauges in a Wheatstone bridge configuration. Load cells of one strain gauge (quarter bridge) or two strain gauges (half bridge) are also available. The electrical signal output is typically in the order of a few millivolts (mV) and requires amplification by an instrumentation amplifier before it can be used. The output of the transducer can be scaled to calculate the force applied to the transducer. Sometimes a high resolution ADC, typically 24-bit, can be used directly.
The gauges themselves are bonded onto a beam or structural member that deforms when weight is applied. In most cases, four strain gauges are used to obtain maximum sensitivity and temperature compensation. Two of the gauges are usually in tension can be represented as T1 and T2, and two in compression can be represented as C1 and C2, and are wired with compensation adjustments. The strain gauge load cell is fundamentally a spring optimized for strain measurement. Gauges are mounted in areas that exhibit strain in compression or tension. When weight is applied to the load cell, gauges C1 and C2 compress decreasing their resistances. Simultaneously, gauges T1 and T2 are stretched increasing their resistances. The change in resistances causes more current to flow through C1 and C2 and less current to flow through T1 and T2. Thus a potential difference is felt between the output or signal leads of the load cell. The gauges are mounted in a differential bridge to enhance measurement accuracy. When weight is applied, the strain changes the electrical resistance of the gauges in proportion to the load. Other load cells are fading into obscurity, as strain gauge load cells continue to increase their accuracy and lower their unit costs.
Common types of load cells
There are several types of load cells:
- Shear beam, a straight block of material fixed on one end and loaded on the other
- Bending beam, a compact, extremely versatile load cell that can easily fit into components where space is limited. These devices are commonly used in industrial and commercial measurement applications.
- Pancake, low-profile compression and wheel tension load cells that have a wide surface area for absorbing shock loads.
- Single point / platform load cells, allow the construction of a weighing scale or machine with only one load cell. Used in small / medium-sized platform scales.
- Double-ended shear beam, a straight block of material fixed at both ends and loaded in the center
- Compression load cell, a block of material designed to be loaded at one point or area in compression
- S-type load cell, a S-shaped block of material that can be used in both compression and tension (load links and tension load cells are designed for tension only)
- Rope clamps, an assembly attached to a rope and measures its tension. Rope clamps are popular in hoist, crane and elevator applications due to the ease of their installation; they have to be designed for a large range of loads, including dynamic peak loads, so their output for the rated load tends to be lower than of the other types
- Loadpin, used for sensing loads on e.g. axles
- Mechanical mounting: the cells have to be properly mounted. All the load force has to go through the part of the load cell where its deformation is sensed. Friction may induce offset or hysteresis. Wrong mounting may result in the cell reporting forces along undesired axis, which still may somewhat correlate to the sensed load, confusing the technician.
- Overload: Within its rating, the load cell deforms elastically and returns to its shape after being unloaded. If subjected to loads above its maximum rating, the material of the load cell may plastically deform; this may result in a signal offset, loss of linearity, difficulty with or impossibility of calibration, or even mechanical damage to the sensing element (e.g. delamination, rupture).
- Wiring issues: the wires to the cell may develop high resistance, e.g. due to corrosion. Alternatively, parallel current paths can be formed by ingress of moisture. In both cases the signal develops offset (unless all wires are affected equally) and accuracy is lost.
- Electrical damage: the load cells can be damaged by induced or conducted current. Lightning hitting the construction, or arc welding performed near the cells, can overstress the fine resistors of the strain gauges and cause their damage or destruction. For welding nearby, it is suggested to disconnect the load cell and short all its pins to the ground, nearby the cell itself. High voltages can break through the insulation between the substrate and the strain gauges.
- Nonlinearity: at the low end of their scale, the load cells tend to be nonlinear. This becomes important for cells sensing very large ranges, or with large surplus of load capability to withstand temporary overloads or shocks (e.g. the rope clamps). More points may be needed for the calibration curve.
- The correct selection of a load cell for the application it is to be used in is a critical factor to realise accuracy and reliability. So what parameters need to be considered when selecting a load cell ?
Excitation and rated output
The bridge is excited with stabilized voltage (usually 10V, but can be 20V, 5V, or less for battery powered instrumentation). The difference voltage proportional to the load then appears on the signal outputs. The cell output is rated in millivolts per volt (mV/V) of the difference voltage at full rated mechanical load. So a 2.96 mV/V load cell will provide 29.6 millivolt signal at full load when excited with 10 volts.
Typical sensitivity values are 1 to 3 mV/V. Typical maximum excitation voltage is around 15 volts.
The full-bridge cells come typically in four-wire configuration. The wires to the top and bottom end of the bridge are the excitation (often labelled E+ and E−, or Ex+ and Ex−), the wires to its sides are the signal (labelled S+ and S−). Ideally, the voltage difference between S+ and S− is zero under zero load, and grows proportionally to the load cell's mechanical load.
Sometimes a six-wire configuration is used. The two additional wires are "sense" (Sen+ and Sen−), and are connected to the bridge with the Ex+ and Ex- wires, in a fashion similar to four-terminal sensing. With these additional signals, the controller can compensate for the change in wire resistance due to e.g. temperature fluctuations.
The individual resistors on the bridge usually have resistance of 350 Ω. Sometimes other values (typically 120 Ω, 1,000 Ω) can be encountered.
The bridge is typically electrically insulated from the substrate. The sensing elements are in close proximity and in good mutual thermal contact, to avoid differential signals caused by temperature differences.
Using multiple cells
One or more load cells can be used for sensing a single load.
If the force can be concentrated to a single point (small scale sensing, ropes, tensile loads, point loads), a single cell can be used. For long beams, two cells at the end are used. Vertical cylinders can be measured at three points, rectangular objects usually require four sensors. More sensors are used for large containers or platforms, or very high loads.
If the loads are guaranteed to be symmetrical, some of the load cells can be substituted with pivots. This saves the cost of the load cell but can significantly decrease accuracy.
Load cells can be connected in parallel; in that case, all the corresponding signals are connected together (Ex+ to Ex+, S+ to S+, ...), and the resulting signal is the average of the signals from all the sensing elements. This is often used in e.g. personal scales, or other multipoint weight sensors.
The most common color assignment is red for Ex+, black for Ex−, green for S+, and white for S−.
Less common assignments are red for Ex+, white for Ex−, green for S+, and blue for S−, or red for Ex+, blue for Ex−, green for S+, and yellow for S−. Other values are also possible, e.g. red for Ex+, green for Ex−, yellow for S+ and blue for S−.
Piezoelectric load cell
Piezoelectric load cells work on the same principle of deformation as the strain gauge load cells, but a voltage output is generated by the basic piezoelectric material – proportional to the deformation of load cell. Useful for dynamic/frequent measurements of force. Most applications for piezo-based load cells are in the dynamic loading conditions, where strain gauge load cells can fail with high dynamic loading cycles. The piezoelectric effect is dynamic, that is, the electrical output of a gauge is an impulse function and is not static. The voltage output is only useful when the strain is changing and does not measure static values.
However, depending on conditioning system used, "quasi static" operation can be done. Using a so-called "Charge amplifier " with "Long" time constant allow accurate measurement lasting many hours for large loads to many minutes for small loads. Another advantage of Piezoelectric load cell, conditioned with a Charge amplifier, is the wide measuring range that can be achieved. Users can choose a load cell with a range of hundred of kN and use it for measuring few N of forces with the same Signal/Noise ratio, again this is possible only with the use of a "Charge amplifier" conditioning.
Hydraulic load cell
The hydraulic load cell uses a conventional piston and cylinder arrangement with the piston placed in a thin elastic diaphragm. The piston doesn't actually come in contact with the load cell. Mechanical stops are placed to prevent over strain of the diaphragm when the loads exceed certain limit. The load cell is completely filled with oil. When the load is applied on the piston, the movement of the piston and the diaphragm results in an increase of oil pressure. This pressure is then transmitted to a hydraulic pressure gauge via a high pressure hose. The gauge's Bourdon tube senses the pressure and registers it on the dial. Because this sensor has no electrical components, it is ideal for use in hazardous areas. Typical hydraulic load cell applications include tank, bin, and hopper weighing. By example, a hydraulic load cell is immune to transient voltages (lightning) so these type of load cells might be a more effective device in outdoor environments. This technology is more expensive than other types of load cells. It is a more costly technology and thus cannot effectively compete on a cost of purchase basis.
Pneumatic load cell
The Load cell is designed to automatically regulate the balancing pressure. Air pressure is applied to one end of the diaphragm and it escapes through the nozzle placed at the bottom of the load cell. A pressure gauge is attached with the load cell to measure the pressure inside the cell. The deflection of the diaphragm affects the airflow through the nozzle as well as the pressure inside the chamber.
Other types include vibrating wire load cells, which are useful in geomechanical applications due to low amounts of drift, and capacitive load cells where the capacitance of a capacitor changes as the load presses the two plates of a capacitor closer together.
Every load cell is subject to "ringing" when subjected to abrupt load changes. This stems from the spring-like behavior of load cells. In order to measure the loads, they have to deform. As such, a load cell of finite stiffness must have spring-like behavior, exhibiting vibrations at its natural frequency. An oscillating data pattern can be the result of ringing. Ringing can be suppressed in a limited fashion by passive means. Alternatively, a control system can use an actuator to actively damp out the ringing of a load cell. This method offers better performance at a cost of significant increase in complexity.
Load cells are used in several types of measuring instruments such as laboratory balances, industrial scales, platform scales and universal testing machines. From 1993 the British Antarctic Survey installed load cells in glass fibre nests to weigh albatross chicks. Load cells are used in a wide variety of items such as the seven-post shaker which is often used to set up race cars.
Load cells weighing performances
Load cells are commonly used to measure weight in an industrial environment. They can be installed on hoppers, reactors...etc... to control their weight capacity, which is often of critical importance for an industrial process. Some performance characteristics of the load cells must be defined and specified to make sure they will cope with the expected service. Among those design characteristics are :
- Combined error
- Minimum verification interval
Load cell calibration
Load cells are an integral part of most weighing systems in industrial, aerospace and automotive industries, enduring rigorous daily use. Over time, load cells will drift, age and misalign therefore they will need to be calibrated regularly to ensure accurate results are maintained. ISO9000 and most other standards specify a maximum period of around 18 months to 2 years between re-calibration procedures, dependant on the level of load cell deterioration. Annual re-calibration is considered best practice by many load cell users for ensuring the most accurate measurements.
Standard calibration tests will use linearity and repeatability as a calibration guideline as these are both used to determine accuracy. Calibration is conducted incrementally starting working in ascending or descending order. For example, if you have a 60 tonne load cell, then you way use specific test weights that measure in 5, 10, 20, 40 and 60 tonne increments – A five step calibration process is usually sufficient for ensuring your device is accurately calibrated. Repeating this five-step calibration procedure 2-3 times is recommended for consistent results.
- "Wheatstone Bridge Diagrams and Equations". Transducer Techniques.
- "Sensors – May 2000 – Getting the Most out of Strain Gauge Load Cells". archives.sensorsmag.com. Retrieved 2018-03-15.
- LLC, Tacuna Systems. "The Essential Guide to Load Cells : Load Cells, Amplifiers, Calibration, Strain Gauges – Tacuna Systems". tacunasystems.com. Retrieved 2017-08-21.
- https://powderprocess.net/Equipments html/Load_cells.html
- "Loadcell trouble shooting".
- DeGlandon, Kathy. "Selectring the Best Hydraulic Pressure Sensor". Drilling Instruments. Drilling Instruments. Retrieved 28 December 2016.
- "Load Cells – Hydraulic". www.centralcarolinascale.com. Retrieved 2018-03-15.
- "Emery Winslow Scale Company – Industrial Scales – Hydrostatic Load Cells For Harsh Environments". www.emerywinslow.com. Retrieved 2018-03-15.
- "2. Where are load cells used? 【Introduction to Load Cells】 A&D". www.aandd.jp. Retrieved 2018-03-15.
- "Load cell testing gets straight to the point". Maritime Journal. Mercator Media. 20 December 2010.
- Highfield, Roger. Antarctic's birds thrive on warmth, The Daily Telegraph 18 August 1993
- "Control of Monitoring & Measuring Equipment". www.iso-9001-checklist.co.uk. Retrieved 2018-07-17.
- "Load Cells: A Beginner's Guide – Measurement Shop UK". www.measurementshop.co.uk. Retrieved 2018-07-17.
- ASTM E4 – Practices for Force Verification of Testing Machines
- ASTM E74 – Practice for Calibration of Force Measuring Instruments for Verifying the Force Indication of Testing Machines
- NTEP – National Conference on Weights and Measures (Certificate of Conformance)