Photoemission spectroscopy (PES), also known as photoelectron spectroscopy, refers to energy measurement of electrons emitted from solids, gases or liquids by the photoelectric effect, in order to determine the binding energies of electrons in a substance. The term refers to various techniques, depending on whether the ionization energy is provided by an X-ray photon, an EUV photon, or an ultraviolet photon. Regardless of the incident photon beam however, all photoelectron spectroscopy revolves around the general theme of surface analysis by measuring the ejected electrons.
Types of Photoemission Spectroscopy
X-ray photoelectron spectroscopy (XPS) was developed by Kai Siegbahn starting in 1957  and is used to study the energy levels of atomic core electrons, primarily in solids. Siegbahn referred to the technique as Electron Spectroscopy for Chemical Analysis (ESCA), since the core levels have small chemical shifts depending on the chemical environment of the atom which is ionized, allowing chemical structure to be determined. Siegbahn was awarded the Nobel Prize in 1981 for this work. XPS is sometimes referred to as PESIS (photoelectron spectroscopy for inner shells) whereas the lower-energy radiation of UV light is referred to as PESOS (outer shells) because it cannot excite core electrons.
In the ultraviolet and visible region, the method is usually referred to as photoelectron spectroscopy for the study of gases, and photoemission spectroscopy for solid surfaces.
Ultraviolet photoelectron spectroscopy (UPS) is used to study valence energy levels and chemical bonding; especially the bonding character of molecular orbitals. The method was developed originally for gas-phase molecules in 1962 by David W. Turner, and other early workers included David C.Frost, J.H.D. Eland and K. Kimura. Later, Richard Smalley modified the technique and used a UV laser to excite the sample, in order to measure the binding energy of electrons in gaseous molecular clusters.
Two-photon photoelectron spectroscopy (2PPE) extends the technique to optically excited electronic states through the introduction of a pump-and-probe scheme.
Extreme ultraviolet photoelectron spectroscopy (EUPS) lies in between XPS and UPS. It is typically used to assess the valence band structure. Compared to XPS it gives better energy resolution, and compared to UPS the ejected electrons are faster, resulting in a better spectrum signal.
The physics behind the PES technique is an application of the photoelectric effect. The sample is exposed to a beam of UV or XUV light inducing photoelectric ionization. The energies of the emitted photoelectrons are characteristic of their original electronic states, and depend also on vibrational state and rotational level. For solids, photoelectrons can escape only from a depth on the order of nanometers, so that it is the surface layer which is analyzed.
Because of the high frequency of the light, and the substantial charge and energy of emitted electrons, photoemission is one of the most sensitive and accurate techniques for measuring the energies and shapes of electronic states and molecular and atomic orbitals. Photoemission is also among the most sensitive methods of detecting substances in trace concentrations, provided the sample is compatible with ultra-high vacuum and the analyte can be distinguished from background.
Typical PES (UPS) instruments use helium gas sources of UV light, with photon energy up to 52 eV (corresponding to wavelength 23.7 nm). The photoelectrons that actually escaped into the vacuum are collected, energy resolved, slightly retarded and counted, which results in a spectrum of electron intensity as a function of the measured kinetic energy. Because binding energy values are more readily applied and understood, the kinetic energy values, which are source dependent, are converted into binding energy values, which are source independent. This is achieved by applying Einstein's relation . The term of this equation is due to the energy (frequency) of the UV light that bombards the sample. Photoemission spectra are also measured using synchrotron radiation sources.
The binding energies of the measured electrons are characteristic of the chemical structure and molecular bonding of the material. By adding a source monochromator and increasing the energy resolution of the electron analyzer, peaks appear with full width at half maximum (FWHM) less than 5–8 meV.
- Angle resolved photoemission spectroscopy AR-PES
- Inverse photoemission spectroscopy IPS
- Ultra high vacuum UHV
- X-ray photoelectron spectroscopy XPS
- Ultraviolet photoelectron spectroscopy UPS
- Two-photon photoelectron spectroscopy 2PPE
- Vibronic spectroscopy
- Covariance mapping
- IUPAC, Compendium of Chemical Terminology, 2nd ed. (the "Gold Book") (1997). Online corrected version: (2006–) "photoelectron spectroscopy (PES)".
- Hercules, D. M.; Hercules, S.H. Al (1984). "Analytical chemistry of surfaces. Part I. General aspects". Journal of Chemical Education 61 (5): 402. Bibcode:1984JChEd..61..402H. doi:10.1021/ed061p402.
- Nordling, Carl; Sokolowski, Evelyn; Siegbahn, Kai (1957). "Precision Method for Obtaining Absolute Values of Atomic Binding Energies". Physical Review 105 (5): 1676. Bibcode:1957PhRv..105.1676N. doi:10.1103/PhysRev.105.1676.
- Sokolowski E.; Nordling C.; Siegbahn K. (1957). Ark. Fysik. 12: 301. Missing or empty
- Ghosh, P.K. (1983). Introduction to Photoelectron Spectroscopy. John Wiley & Sons. ISBN 0-471-06427-0.
- Turner, D. W.; Jobory, M. I. Al (1962). "Determination of Ionization Potentials by Photoelectron Energy Measurement". The Journal of Chemical Physics 37 (12): 3007. Bibcode:1962JChPh..37.3007T. doi:10.1063/1.1733134.
- Bauer, M.; Lei, C.; Read, K.; Tobey, R.; et al. (2001). "Direct Observation of Surface Chemistry Using Ultrafast Soft-X-Ray Pulses" (PDF). Physical Review Letters 87 (2): 025501. Bibcode:2001PhRvL..87b5501B. doi:10.1103/PhysRevLett.87.025501.