|This article is an orphan, as no other articles link to it. Please introduce links to this page from ; try the Find link tool for suggestions. (February 2009)|
|This article does not cite any sources. (May 2014) (Learn how and when to remove this template message)|
Pilatus is the name of a series of silicon pixel detectors developed at the Swiss Light Source, and commercialised by Dectris. The common factor is that X-rays are converted to an electrical signal by the photoelectric effect in silicon subject to a substantial bias voltage, and then counted directly by a series of cells in an ASIC bonded to the silicon detector, rather than relying on a phosphor. That is, each pixel has its own amplifier, discriminator (for distinguishing X-rays of the desired energy from noise) and counter circuit; this is possible because the desired pixel size for X-ray crystallography is around 100 micrometres, and a 100 micrometre square in a contemporary CMOS process can contain a significant quantity of electronics.
Design challenges include making the bonding of the ASIC to the sensor sufficiently reliable; the ASIC has to be manufactured in a radiation-tolerant process since the sensor layer by no means absorbs all the incident X-rays.
Other issues involve combining ASICs to make a sensor of usable size - protein crystallography applications tend to want an active area of around 100,000 mm², though powder diffraction work can use a rather smaller active area. One detector inclines the ASICs so that the view from the point of view of the crystal is of an unbroken sphere of detector.