Poly Implant Prothèse
|Fate||Liquidation in 2011|
|Headquarters||La Seyne-sur-Mer, France|
|Jean-Claude Mas (President)|
Number of employees
Poly Implant Prothèse (PIP) was a French company founded in 1991 that produced silicone gel breast implants. The company was preemptively liquidated in 2010 following the revelation that they had been illegally manufacturing and selling breast implants made from cheaper industrial-grade silicone since 2001 (instead of the mandated medical-grade silicone they had previously used). The hundreds of thousands of unapproved implants sold globally by PIP from 2001 to 2010 were found to have a 500% higher risk of rupturing or leaking than approved models, as well as being implicated in at several deaths due to systemic toxicity and several cases of induced breast cancer. The scandal, which produced fears of a massive health disaster, prompted a full recall of the company's implants by the French health ministry in 2010, by which time the company was already defunct.
- 1 History
- 2 Scandal
- 3 Timeline
- 4 Issues
- 5 Chest, testicle and buttock implants
- 6 Impact per country
- 7 New company
- 8 Legal process
- 9 Aftermath
- 10 See also
- 11 References
- 12 External links
PIP was founded in 1991 by the Frenchman Jean-Claude Mas, born in 1939, a former butcher and later medical sales representative for the Bristol Myers company for 15 years. Mas had previously teamed up with plastic surgeon Henri Arion, who had introduced breast implants to France in 1965. After Arion died in a plane crash in 198, Mas went on alone and launched PIP in 1991.
Starting in 1991, the company produced approximately 2 million sets of silicone breast implants over a 20-year period. The implants were exported to Latin American countries such as Brazil, Venezuela and Argentina, Western European markets including Britain (25 000), Germany, Spain and Italy, as well as Australia (8900).
Following the FDA ruling in 2000 which banned silicone breast implants in the US market (leading to a slump in sales worldwide), Mas sought to "tighten PIP's belt" and recoup some of its lost market share by severely cutting costs. In particular, Mas was credited with the idea that by switching PIP's silicone from externally purchased medical-grade to in-house produced industrial-grade, huge savings (on the order of 90%) could be generated that would ensure profits remained high no matter the market. As one PIP engineer later noted, this decision involved a relatively small change in the formula for the silicone, enough so that only superficial differences in the end product were noted. However, none of the relevant regulations were followed for manufacturing medical implants, and no pre-launch tests were conducted.
2003: the first signs of legal problems and financial losses can be traced by regulatory filings.
2009: Concerns surfaced in France first in 2009 when surgeons started reporting an abnormally high rupture rates. Which resulted in a flood of legal complaints and the company's bankruptcy.
- A former PIP worker and union chief, Eric Mariaccia said "You had to have been a chemist to have noticed anything,". He also said that "The responsible ones aren't the workers but the heads of the company, notably the four who were linked to production and thus responsible for their quality,".
2010: In March 2010 PIP was placed into liquidation with losses of 9 million EUR after the French medical safety agency recalled its implants. In a subsequent inspection of the manufacturing site, the company was found to use unapproved industrial-grade, silicone, with a cost of only 10% of an approved gel.
2011: On December 20 French officials say that an action plan is underway following the death of a woman from ALCL. The French government recommended on December 23, 2011 that 30 000 women in France seek removal of breast implants made of a suspect silicone gel by the worldwide exporting PIP firm.
Medical problems arise when breast implants rupture and cause inflammation and irritation. Further, impure gel may also release toxic substances. Removing implants "carries risks in itself", says the French government. By the end of 2011, eight cases of breast tumours had been uncovered in women after removing the implants, yet, while of concern, according to the French government, there is no evidence of any increased cancer risk.
The UK medical watchdog, the Medicines and Healthcare products Regulatory Authority (MHRA), says France has reported rupture rates of around 5% for PIP implants, compared with 1% in the UK. But a "leading cosmetic surgery firm" has privately warned ministers that the proportion of women at risk is as high as 8%. Symptoms can include lumps around the implant or in the under-arm, inflammation in the breast tissue and a hardening of the breast. Patient UK says French regulatory authority has reported a rupture rate of up to 10%, with a gel leak in 11% of cases.
Nigel Mercer of the British Association of Aesthetic and Plastic Surgeons (BAAPS) says the French move was "certainly not unreasonable" but the British cosmetic surgeon Kevin Hancock said the divergent government responses would cause distress for British women. Plastic surgeon Hancock, of the Liverpool Women's Hospital, says there were concerns in the profession over a high rupture rate and that "We are worried about the rupture risk because it is the rupture that brings the contents into direct contact with the body's tissues," .. "We know that the contents were not what they were supposed to be. So in general we agree with the (French) decision to remove them.".
Patients are used as "guinea pigs", said Nigel Mercer, a former president of the British Association of Aesthetic and Plastic Surgeons (BAAPS), because medical implants lack the requirement of an independent clinical trial.
The company sold a high-end product that used approved silicone for wealthier clients and products with a price 5 times lower,[clarification needed] for which the lawyer (Haddad) representing the founder says "There is a product made by PIP which did not formally receive the (regulators') approval and in this regard there was a violation of regulations,". The company did not ask the France's medical device safety regulators for approval of their in house manufactured product, but expected it would have gotten approval if the company asked for it.
Cancer has been found in 20 women with the allegedly faulty PIP breast implants but French health authorities (Agence française de sécurité sanitaire des produits de santé (AFSSAPS)) insisted there was still no proven link with the disease.
AFSSAPS has as of 28 December,[when?] registered 15 cases of breast adenocarcinoma, the most frequent form of breast cancer, 1 case of breast lymphoma, 2 cases of other lymphoma, 1 case of lung cancer and 1 case of acute myelogenous leukemia in women with the implants.
A French woman with a PIP breast implant where capsular scar tissue formed in the breast, which is a common occurrence with implants. This tissue developed anaplastic large-cell lymphoma (ALCL), a rare form of cancer that affects the immune system cells. The cancer caused death in this case and caused the French authorities to act.
During the 672 registered preventive extractions, in 43 cases (6.4%) the implants had ruptured and in 14 cases (2.1%) they were "oozing", AFSSAPS said.
The PIP implants have been resold as "Rofil M-implants". British Association of Aesthetic and Plastic Surgeons (BAAPS) believes many British women may have had the bad Rofil M-implants used when they bought cut-price breast enlargement surgery from clinics in Eastern and Central Europe.
The PIP factory was shut down and their products banned after it was found they had used the chemicals Baysilone, Silopren and Rhodorsil in their implants. These chemicals are normally used as fuel additives or in the manufacture of industrial rubber tubing.
In 1993, there was an availability crisis of silicone materials for use in medical long-term implant applications. In 1993, Mas ordered the staff to "hide the truth" and "We did it for 13 years without a problem". The PIP gel cost 5 EUR per litre while the authorized American gel Nusil cost 35 EUR per litre. The PIP implants were made with 75% non-authorised gel and 25% Nusil. "The material was better than that used to make the officially authorised gel," Mas said. "Usually, from 1997 onwards we hid the products used to make the PIP gel. I wasn't allowed to buy these products because they were not authorised. We organised everything to escape being monitored." Investigators interviewed Mas on 18 and 19 November 2010 after the French health authority AFSSAPS discovered the PIP breast implants did not conform to regulations. They passed the case to a judge who ordered a police inquiry for "aggravated deception".
Chest, testicle and buttock implants
- "Three people were specially trained to work with a machine that made silicon testicles," said a former PIP worker.
Impact per country
The French government recommended per December 23, 2011 that 30 000 women in France seek removal of breast implants made by PIP. For women that want to keep their implants, the state will pay for a 6-monthly ultrasound scan.
French public health care funds will be used to finance the recommended implant removals, at a cost estimated at 60 million EUR. New implants will be paid for in cases where they were used initially for medical reasons, others will need to finance a new implant by themselves. Associations representing women with PIP implants demand that public funds cover all cases. "This announcement is just a smokescreen and the victims of PIP are angry," says Alexandra Blachere, head of the association of PIP implant users in France.
In Britain, there are an estimated 30 000 - 40 000 affected women. The chief Medical Officer Dame Sally Davies says: "Women with PIP implants should not be unduly worried. We have no evidence of a link to cancer or an increased risk of rupture. If women are concerned they should speak to their surgeon.". Removing the implant "carries risks in itself," she also said.
On 6 January 2012, NHS officials began to offer examinations and possible removal - in instances where a doctor ruled that necessary. They also called on private medical centers to offer the same deal to patients who had paid for private sector cosmetic surgery.
On 8 January 2012, The Sunday Times reported that leading cosmetic surgeon Jan Stanek had conducted a study with colleague Mike Berry into PIP implants. Their report to the Journal of Plastic and Aesthetic Surgery revealed that, in an audit of 453 patients, between 16% and 34% of the PIP implants ruptured. This compared to a failure rate of less than 1% in other implants.
Germany and Brazil
Germany's medical safety board and Brazil's health watchdog called for users of PIP implants to visit their doctor for checks.
The bad breast implants were sold to about 1000 Dutch women under the name "M-implants" by the Dutch importer Rofil.  The Dutch healthcare spokeswoman Diana Bouhys says "We have advised them to consult their physician", but also declined to disclose the name of the company. PIP was the 3rd largest maker of breast implants in the world.
Approximately 5000 implants have already been sold, corresponding to an estimated 2500 Swedish women with the controversial implants. Nine clinics in Sweden has sold them says Gert Bruse of the Swedish Medical Products Agency (Läkemedelsverket) told the Dagens Nyheter (DN) newspaper this week. The agency also recommends that the implants to be removed. Sweden banned the company's implants in March 2010, along with many other European countries, when it was discovered that substandard silicone gel was causing unusually many implants to burst. "We've had cases where the implants have burst in Sweden, but it's fewer than in France in relation to how many women have these implants in both countries," and "Implants shouldn't burst and they shouldn't have an irritating effect that spreads throughout the body. What this company has done is criminal, simply put." says Bruse. PIP's implants have only been used in cosmetic breast augmentation surgery, not in reconstructive surgery.
Finland and Estonia
Patients from Finland often visit Tallinn to get cosmetic surgery. In October 2011 (two months before 22 December 2011), a Finnish woman in her 30s had received a different implant than what was agreed to in advance. An M-implant with the same type code as the implant from PIP was inserted instead of a promised Allergan implant. At least two Estonian clinics operating in Tallinn have offered M-implants as recently as December 2011.
Implants were sold up until May 2000 through Heritage Worldwide, documents filed with the US government show. When the FDA launched a moratorium on silicone implants. At the time, the US market accounted for 40% of Heritage Worldwide's revenues, or 3 million EUR, according to corporate documents filed in 2009 with the U.S. Securities and Exchange Commission (SEC). In 2007 significant losses started as both users and distributors filed complaints. Between 1996 - 2009, PIP was targeted by several dozen lawsuits in the United States, filed by users and business partners, claiming breach of contract or unmet payments. Starting in 2003 dozens of women began filing lawsuits mainly for product liability against PIP. But as of 2009, no trial date had yet been set. Many of the lawsuits were later dismissed.
On 29 December Health Minister Renato Balduzzi said that hospitals and clinics are required to compile a list of women who has received breast implants from PIP. Even clinics that didn't use any PIP implants will be required to send a declaration stating that.
Czech Republic health ministry said about 2000 women with potentially faulty implants should have them removed.
According to the Icelandic Directorate of Health, 440 women underwent breast surgery where PIP breast implants were used. The Ministry of Welfare have stated that health insured women will be offered counsel and ultrasound examination, free of charge. The Ministry of Welfare also stated that in the event that the implant is leaking, the government will pay a portion of the cost to have it removed.
Latin and South America
- Costa Rica - breast implants banned and taken off the market in March 2010
- Brazil - banned the breast implants on 30 December 2011, taken off the market in 2010
- Bolivia - implants replaced free of charge for some women on 29 December 2011
- Venezuela - promised to remove them free of charge
- Ecuador - sales banned on 29 December 2011
- Argentina - ban
- Chile - ban
- Colombia - ban
Ministry of Health and Medical Education banned the breast implants and took them off the market in 2012. They advised women that used the products to consult with a doctor and if deemed a health threat, to have them removed.
The adequacy of regulations on silicone-filled breast implants, investigation by the regulatory authorities and advice given to women implanted with PIP silicone pre-filled breast implants was criticized in December 2012 in a report examining information publicly available from the regulatory authorities.
A new firm has been registered under the name France Implant Technologie (FIT), with Mas's son Nicolas Lucciardi, 27, and daughter, Peggy Lucciardi, 24, at the address of their mother, Dominique Lucciardi, who was Mas's former civil partner.
The local paper Nice-Matin obtained the business plan in which Mas is named as a "technical-commercial consultant" to the company and described as "a creative genius". Two former PIP managers were listed in senior positions in the company. With the objective to export implants to the "European, South American and Chinese markets". An investment of 2 million EUR was planned to put the former PIP plant back into operation, aiming to manufacture 400 implants a day with about 20 workers. But Nicolas Lucciardi said that the project has collapsed due to media coverage of the scandal, adding, "That's obvious.".
The lawyer Yves Haddad is representing Mas. Legal complaints have been filed by 2500 women in France and 250 women in Britain. Mas is wanted in Costa Rica for "life and health" offenses, a warrant issued by Interpol.
By February 2011 TÜV Rheinland sued PIP, claiming it had been "thoroughly and continuously misled" regarding the silicone used. Many executives are expected to face charges of aggravated fraud in an ongoing court case in France likely to be scheduled for October 2012, which carries the possibility of jail terms of up to 5-years. The death of a cancer victim who had such implants may result in more serious charges of involuntary manslaughter.
Frédéric Van Roekeghem, general director for La Caisse Nationale d'Assurance Maladie (French state social insurance agency, CNAM) will go to court regarding the complaints of bad implants. A criminal charge will be made within a few days he said to TF1 on 25 December 2011. The decision has been made in consultation with minister of health, Xavier Bertrand.
Mas has recently undergone difficult surgery regarding a vascular problem that prevents him from walking said his lawyer Haddad. Sources say a Marseilles court could soon announce fraud charges against 4 -6 former PIP employees.
A fraud case was filed against PIP on 30 December 2011 by France’s state-run health insurance fund (CNAM) officials in Marseille announced. Which is near the company’s laboratory at La Seyne-sur-Mer. On 26 January 2012 at just before 07:00 the founder Mas was arrested in the home of a friend in the Six-Fours in the Var located in South of France and taken into custody. Investigators searched the house. A deputy chief executive was also arrested at his home.
Philippe Courtois, a lawyer represents 1300 women in France. Laurent Gaudon, represents 4 women and say a surgeon should inform about complications from both the surgery and devices used.
Nathalie Lozano, lawyer, represents more than 1400 women from Colombia, Argentina, United Kingdom and Venezuela. She claims TÜV Rheinland's gross negligence in the process of certifying the quality process.
On 17 April 2013 the trial of 5 executives in begun in Marseille, France. Those charged are the company founder Mr Mas, who was booed in the court, his deputy Claude Couty, the quality director Hannelore Font, technical director Loic Gossart and products chief Thierry Brinon. They face up to 5 years in prison. The trial is expected to last until 17 May.
On 10 December 2013, Jean-Claude Mas was sent to Jail for 4 years he was also fined 75,000 euros (£63,000) by a court in Marseille, France.
British Association of Aesthetic Plastic Surgeons (BAAPS) says that poor postmarketing surveillance is the root behind the crisis and proposes significantly more stringent monitoring of all medical devices including breast implants and all cosmetic injectables, via compulsory, regular reporting of adverse effects and mystery shopping. Professor Sir Bruce Keogh, the NHS Medical Director, will look into whether the cosmetic surgery industry needs to be more effectively regulated.
- reuters.com - Regulators had not approved most PIP implants: lawyer, 2011-12-27
- thestar.com.my - France, UK issue clashing advice on breast implants, 2011-12-24
- straitstimes.com - Interpol seeks arrest of breast implant company founder, 2011-12-24
- thelocal.fr - Breast implant tycoon hits back over scandal, 2011-12-29
- gmanetwork.com - Sub-standard breast implants sold to Dutch firm under new name, 2011-12-27
- Steve Bird (December 31, 2011). "The terrifying story of how a former French butcher earned millions from selling faulty breast implants made from mattress filling". Daily Mail. Retrieved December 31, 2011.
- rfi.fr - French breast implant fraud boss tries to start new company, 2011-12-30
- thelocal.fr - French breast implant maker faced US lawsuits, 2011-12-27
- chicagotribune.com - French health minister wants implant boss found, 2010-10-14
- guardian.co.uk - French government 'to order women to remove defective breast implants', 2011-12-20
- mhra.gov.uk - Poly Implant Prosthese (PIP) breast implants update, 2011-12-23
- thelocal.se - Swedish women warned over French fake boobs, 2011-12-23
- bbc.co.uk - Breast implants: French boss sought 'for drunk driving', 2011-12-24
- independent.co.uk - Exclusive: Silicone implant risk 'eight times greater, 2012-01-01
- patient.co.uk - 'Breast implants scrutinised after cancer death' - NHS Analysis, 2011-12-21
- bbc.co.uk - Cosmetic surgeon concerned over 'guinea pig' patients, 2010-10-14
- medicalxpress.com - 20 cancer cases in French women with PIP implants: agency, 2011-12-30
- telegraph.co.uk - French implant company 'used same silicone in male chest and testicle implants', 2012-01-05
- nusil.com - Healthcare products, 2012, read 2012-01-26
- guardian.co.uk - French breast implant firm boss unrepentant over unauthorised silicone, 2012-01-06
- nydailynews.com - Men received banned silicone in chest, testicle implants, French PIP scandal grows, Daily News (New York) 2012-01-05
- nzherald.co.nz - Testicle implants also contaminated - report, The New Zealand Herald 2012-01-06
- bellinghamherald.com - Mixed advice in Europe over faulty breast implants, 2012-01-06
- Sarah-Kate Templeton, John Follain, 'Leak Rate Of Faulty Implants Is 1 in 3 In The UK' , The Sunday Times, 8 January 2012
- 'Advocaat: artsen kozen voor die implantaten' (transl.: 'Lawyer: doctors opted for those implants'), http://nos.nl/artikel/2047963-advocaat-artsen-kozen-voor-die-implantaten.html, NOS Nieuws, 20 July 2015
- http://www.hs.fi/english/print/1135270081610 hs.fi - Finnish woman got dangerous breast implant in Estonia, 2011-12-22
- jpost.com - Italy orders list of women with French firm's implants, 2011-12-29
- grapevine.is - Government Will Assist With Breast Implant Removal, 2012-01-10
- costa-rica-forum-cosmetic-plastic-surgery.com - PIP Implants in Costa Rica - La Nacion and Ministerio de Salud Costa Rica - extract of archives, 2012-01-26
- http://www.bbc.co.uk/persian/iran/2012/01/120129_l21_cancer_pip_implantes.shtml , 2012-01-29
- "Quality First International Limited Medical Devices Consultancy". qualityfirstint.com. 2013-01-07. Retrieved 2013-01-17.
- ibtimes.co.uk - France: Faulty Breast Implant Firm PIP Founder Arrested, 2012-01-26
- guardian.co.uk - PIP breast implant boss arrested in south of France, 2012-01-26
- "Afectadas por prÃ³tesis PIP | Lozano Blanco & Asociados". implantespip.com.co. Retrieved 2013-02-06.
- "BBC News - PIP implants scandal: Trial under way in Marseille". bbc.co.uk. Retrieved 2013-04-17.
- "Breast implants: PIP's Jean-Claude Mas gets jail sentence". BBC News. 10 December 2013. Retrieved 10 December 2013.
- "BBC News - PIP breast implants: 'serious lessons must be learned'". bbc.co.uk. 2012-05-14. Retrieved 2013-04-17.
- PIP breast implants – latest official news from the NHS at the National Health Service website
- PIP Breast Implants : Department of Health - Publications
- BBC - Watchdog: PIP breast implants
- guardian.co.uk - The breast implants scandal is bringing out the worst kinds of private sector attitudes, 2012-01-14