Quantitative computed tomography
|Quantitative computed tomography|
Quantitative computed tomography (QCT) is a medical technique that measures bone mineral density (BMD) using a standard X-ray Computed Tomography (CT) scanner with a calibration standard to convert Hounsfield Units (HU) of the CT image to bone mineral density values. Quantitative CT scans are primarily used to evaluate bone mineral density at the lumbar spine and hip.
In general, solid phantoms placed in a pad under the patient during CT image acquisition are used for calibration. These phantoms contain materials that represent a number of different equivalent bone mineral densities. Usually either calcium hydroxyapatite (CaHAP) or potassium phosphate (K2HPO4) are used as the reference standard.
QCT was invented at University of California San Francisco (UCSF) during the 1970s. Douglas Boyd, PhD and Harry Genant, MD used a CT head scanner to do some of the seminal work on QCT. At the same time, CT imaging technology progressed rapidly and Genant and Boyd worked with one of EMI’s first whole body CT systems in the late 1970s and early 1980s to apply the quantitative CT method to the spine, coining the term "QCT." Genant later published several articles on spinal QCT in the early 1980s with Christopher E. Cann, PhD. Today, QCT is being used in hundreds of medical imaging centers around the world, both clinically and as a powerful research tool.
Three-dimensional QCT imaging
Originally, conventional 2D QCT used individual, thick CT slice images through each of multiple vertebrae which involved tilting the CT scanner gantry to align the slice with each vertebra. Today, modern 3D QCT uses the ability of CT scanners to rapidly acquire multiple slices to construct three-dimensional images of the human body. Using 3D imaging substantially reduced image acquisition time, improved reproducibility and enabled QCT bone density analysis of the hip.
QCT exams are typically used in the diagnosis and monitoring of osteoporosis.
At the spine, QCT is used to measure the bone mineral density of only the spongy interior bone separately from the dense cortical bone that forms the exterior walls of the vertebrae. The trabecular bone has much higher metabolic activity than the cortical bone and so is affected by age, disease and therapy-related changes earlier and to a greater degree than cortical bone. This means that QCT of the spine has an advantage compared to other bone density tests because earlier changes in bone mineral density may be detected .
Clinically, QCT is used at the hip to produce areal BMD measurements and T-Scores that are equivalent to DXA measurements. The exam can be done without particular attention to the positioning of the patient’s limbs because the software allows the hip anatomy to be manipulated after the image is captured, allowing the exam to be performed on patients with arthritic hips who may find traditional exams uncomfortable.
Contraindications for use
QCT bone densitometry should not be used for patients who have the following conditions:
- Patients who have recently had another radiological procedure that includes the introduction of high density contrast material (barium, iodine, thorotrast, thorium) or radio-opaque catheters and tubes.
- Patients who are pregnant or may be pregnant.
QCT scan protocols are low-dose and can limit the amount of radiation exposure to between 200-400µSv for a spine exam This is comparable to a set of mammograms and typically substantially less than a standard CT exam. Using other non-IV contrast abdominal or pelvic scans such as a Virtual Colonography studies, the QCT exam can be performed without requiring any further image acquisition or consequent radiation dose to the patient.
QCT enables spine BMD measurements on patients with scoliosis, which cannot usually be measured using Dual-energy X-ray absorptiometry (DXA). In addition, QCT can avoid the artiﬁcially high BMD measurements that can confuse the results from DXA in arthritic patients, patients who are obese, who suffer from disc space narrowing or spinal degenerative diseases, aortic calciﬁcation or osteophytes.
Dual use of CT images
Several studies have shown that bone density may be measured by QCT using CT images that were ordered for other purposes. Using pre-existing images, including CT colonography exams, QCT allows for bone density screening without submitting the patient to any additional radiation exposure. The feasibility of using routine abdominal contrast-enhanced CT scans for the evaluation of bone density by QCT has also been demonstrated.
Average bone mineral density is calculated and then compared to age and sex matched controls. At the spine, a volumetric BMD measurement is made using QCT and rather than using T-Scores, it should be compared to guideline thresholds from the American College of Radiology (ACR): a BMD < 80 mg/cm3 indicates osteoporosis; a BMD < 120 mg/cm3 and > 80 mg/cm3 indicates osteopenia; and a BMD above 120 mg/cm3 is considered normal.
At the hip, a DXA-equivalent T-score may be calculated for comparison to the WHO classiﬁcation at the proximal femur as normal, osteopenia (T-Score < -1.0 and > -2.5) or osteoporosis (T-Score < -2.5). This T-Score may also be used for fracture risk probability calculation in the WHO FRAX® tool with "T-Score" as the appropriate DXA setting.
Peripheral quantitative computed tomography
In medicine, peripheral quantitative computed tomography, commonly abbreviated pQCT, is a type of quantitative computed tomography (QCT), used for making measurements of the bone mineral density (BMD) in a peripheral part of the body, such as the forearms or legs as opposed to QCT that measures bone mineral density at the hip and spine. It is useful for measuring bone strength.
Comparison to DXA
Unlike most other common techniques for measuring BMD, a pQCT scan is able to measure volumetric bone mineral density, plus other measures such as the stress-strain index (SSI) and the geometry of the bone. DXA is only able to provide the areal bone mineral density.
- J. E. Adams, "Quantitative computed tomography.," European journal of radiology, vol. 71, no. 3, pp. 415-24, Sep. 2009.
- K G Faulkner, C C Glüer, S Grampp, and H K Genant, “Cross-calibration of liquid and solid QCT calibration standards: Corrections to the UCSF normative data” Osteoporosis International, vol. 3, no. 1, pp. 36-42, 1993.
- Reflections on Development of Quantitative Computed Tomography
- K. Engelke, A. Mastmeyer, V. Bousson, T. Fuerst, J.-D. Laredo, and W. a Kalender, "Reanalysis precision of 3D quantitative computed tomography (QCT) of the spine.," Bone, vol. 44, no. 4, pp. 566-72, Apr. 2009.
- B. C. C. Khoo et al., "Comparison of QCT-derived and DXA-derived areal bone mineral density and T scores.," Osteoporosis International, vol. 20, no.9, pp. 1539-45, Sep. 2009.
- J. S. Bauer, S. Virmani, and D. K. Mueller, "Quantitative CT to assess bone mineral density as a diagnostic tool for osteoporosis and related fractures," MedicaMundi, vol. 54, no. 2, pp. 31-37, 2010.
- R. M. Summers et al., "Feasibility of simultaneous computed tomographic colonography and fully automated bone mineral densitometry in a single examination.," Journal of computer assisted tomography, vol. 35, no. 2, pp. 212-6, 2011.
- A. El Maghraoui and C. Roux, "DXA scanning in clinical practice.," QJM, vol. 101, no. 8, pp. 605 17, Aug. 2008.
- E. W. Yu, B. J. Thomas, J. K. Brown, and J. S. Finkelstein, "Simulated increases in body fat and errors in bone mineral density measurements by DXA and QCT.," Journal of Bone and Mineral Research, vol. 27, no. 1, pp. 119-124, Sep. 2011.
- G. Guglielmi et al., "Eﬀect of Spinal Degenerative Changes on Volumetric Bone Mineral Density of the Central Skeleton as Measured by Quantitative Computed Tomography," Acta Radiologica, vol. 46, no. 3, pp. 269-275, Jan. 2005.
- J. A. Smith, J. A. Vento, R. P. Spencer, and B. E. Tendler, "Aortic Calciﬁcation Contributing to Bone Densitometry Measurement," Journal of Clinical Densitometry, vol. 2, no. 2, pp. 181-183, Jun. 1999.
- G. Liu, M. Peacock, O. Eilam, G. Dorulla, E. Braunstein, and C. C. Johnston, "Eﬀect of osteoarthritis in the lumbar spine and hip on bone mineral density and diagnosis of osteoporosis in elderly men and women.," Osteoporosis International, vol. 7, no. 6, pp. 564-9, Jan. 1997.
- J. C. Prior et al., "Premenopausal ovariectomy-related bone loss: a randomized, double-blind, one-year trial of conjugated estrogen or medroxyprogesterone acetate.," Journal of Bone and Mineral Research, vol. 12, no. 11, pp. 1851-63, Nov. 1997.
- Pickhardt et al., "Simultaneous screening for osteoporosis at CT colonography: Bone mineral density assessment using MDCT attenuation techniques compared with the DXA reference standard," Journal of Bone and Mineral Research, vol. 26, no. 9, pp. 2194-203, 2011, ,
- Bauer et al., "Volumetric quantitative CT of the spine and hip derived from contrast-enhanced MDCT: conversion factors," American Journal of Roentgenology, vol. 188, no.5, pp. 1294-301, 2007.
- American College of Radiology, "ACR Practice Guideline for the Performance of Quantitative Computed Tomography (QCT) Bone Densitometry," 2008.
- E. M. Lewiecki et al., "International Society for Clinical Densitometry 2007 Adult and Pediatric Oﬃcial Positions.," Bone, vol. 43, no. 6, pp. 1115-21, Dec. 2008.
- E. M. Lewiecki et al., "Oﬃcial Positions for FRAX® Bone Mineral Density and FRAX® simpliﬁcation from Joint Oﬃcial Positions Development Conference of the International Society for Clinical Densitometry and International Osteoporosis Foundation on FRAX®.," Journal of Clinical Densitometry, vol. 14, no. 3, pp. 226-36.
- Müller A, Rüegsegger E, Rüegsegger P (1989). "Peripheral QCT: a low-risk procedure to identify women predisposed to osteoporosis.". Phys Med Biol. 34 (6): 741–9. doi:10.1088/0031-9155/34/6/009. PMID 2740441.