Removable partial denture
|Removable partial denture|
A removable partial denture (RPD) is a denture for a partially edentulous patient who desires to have replacement teeth for functional or aesthetic reasons and who cannot have a bridge (a fixed partial denture) for any reason, such as a lack of required teeth to serve as support for a bridge (i.e. distal abutments) or financial limitations.
This type of prosthesis is referred to as a removable partial denture because patients can remove and reinsert it when required without professional help. Conversely, a "fixed" prosthesis can and should be removed only by a dental professional.
Partially edentulous conditions
Depending on where in the mouth teeth are missing, edentulous situations can be grouped under four different categories, as defined by Dr. Edward Kennedy in his classification of partially edentulous arches.
- Class I (bilateral free ended partially edentulous)
- Class II (unilateral free ended partially edentulous)
- Class III (unilateral bounded partially edentulous)
- Class IV (bilateral bounded anterior partially edentulous)
Kennedy Class I RPDs are fabricated for people who are missing some or all of their posterior teeth on both sides (left and right) in a single arch (either mandibular or maxillary), and there are no teeth posterior to the edentulous area. In other words, Class I RPDs clasp onto teeth that are more towards the front of the mouth, while replacing the missing posterior teeth on both sides with false denture teeth. The denture teeth are composed of either plastic or porcelain.
Class II RPDs are fabricated for people who are missing some or all of their posterior teeth on one side (left or right) in a single arch, and there are no teeth behind the edentulous area. Thus, Class II RPDs clasp onto teeth that are more towards the front of the mouth, as well as on teeth that are more towards the back of the mouth of the side on which teeth are not missing, while replacing the missing more-back-of-the-mouth teeth on one side with false denture teeth.
Class III RPDs are fabricated for people who are missing some teeth in such a way that the edentulous area has teeth remaining both posterior and anterior to it. Unlike Class I and Class II RPDs which are both tooth-and-tissue-borne (meaning they both clasp onto teeth, as well as rest on the posterior edentulous area for support), Class III RPDs are strictly tooth-borne, which means they only clasp onto teeth and do not need to rest on the tissue for added support. This makes Class III RPDs exceedingly more secure as per the three rules of removable prostheses that will be mentioned later, namely: support, stability and retention. (See the article on dentures for a more thorough review of these three fundamentals of removable prosthodontics.)
However, if the edentulous area described in the previous paragraph crosses the anterior midline (that is, at least both central incisors are missing), the RPD is classified as a Class IV RPD. By definition, a Kennedy Class IV RPD design will possess only one edentulous area.
Class I, II and III RPDs that have multiple edentulous areas in which replacement teeth are being placed are further classified with modification states that were defined by Oliver C. Applegate. Kennedy classification is governed by the most posterior edentulous area that is being restored. Thus if, for example, a maxillary arch is missing teeth #1, 3, 7-10 and 16, the RPD would be Kennedy Class III mod 1. It would not be Class I, because missing third molars are generally not restored in an RPD (although if they were, the classification would indeed be Class I), and it would not be Class IV, because modification spaces are not allowed for Kennedy Class IV.
The results of a study conducted in Saudi Arabia, showed that the occurrence of Kennedy Class III partial edentulism was 67.2 % in the maxillary arch and 64.1% in the mandibular arch. Followed by Class II in both maxillary and mandibular arch with an average of 16.3 % in maxillary arch and14.8% in the mandibular arch. Based on these results, class III has the highest prevalence in younger group of patient (31- 40 years). Class I and class II have the highest incidence among older group of Patients (41-50 years).
Components of an RPD
Rather than lying entirely on the edentulous ridge like complete dentures, removable partial dentures possess clasps of cobalt-chrome or titanium metal or plastic that "clip" onto the remaining teeth, making the RPD more stable and retentive.
The parts of an RPD can be listed as follows (and are exemplified by the picture above):
- Major connector (The thick metal "U" in the RPD image above is a lingual bar, a type of major connector)
- Minor connector (See the small struts protruding from the lingual bar at roughly 90 degree angles.)
- Direct retainer (Examples are in the upper left of upper photo and lower right of lower photo; the clasp arms act to hug the teeth and keep the RPD in place. The metal clasp and rest immediately adjacent to the denture teeth is also a direct retainer.)
- Indirect retainer (An example is the little metal piece coming off the "U" at a 90 degree angle near the top of the upper photo, which is a cingulum rest on a canine.)
- Physical retainer (This is a mesh of metal that allows the pink base material to connect to the metal framework of the RPD. Some consider physical retainers their own component (making a total of seven), while others consider them within the indirect retainer category (thus making a total of six components.)
- Base (the pink material, mimicking gingiva)
- Teeth (plastic or porcelain formed in the shape of teeth)
Direct retainers may come in various designs:
- Cast circumferential clasp (suprabulge)
- Half and half
- Ring clasp
- Wrought wire clasp
- Roach clasp (infrabulge)
Both cast circumferential and wrought wire clasps are supra bulge clasps, in that they engage an undercut on the tooth by originating coronal to the height of contour, while Roach clasps are infrabulge clasps and engage undercuts by approaching from the gingival.
In addition there are a couple of specific theories which include the clasp design:
- RPI: mesial rest, distolingual guide plate, I-bar
- The RPI design was made for clasping a bilateral free end extension. These clasps are unique because they have to take into account extra torque force due to being tissue borne (and not tooth borne) at the posterior.
- Described by Kratochvil in 1963 and modified by Krol in 1973
- Kratochvil designed the abutment tooth with a long rest (from the mesial marginal ridge to the distal pit), long guide plane, and a regular I-bar clasp.
- Krol modified this design with a short occlusal rest, short guide plane (touching only from occlusal to middle third), and a mesial-shifted I-bar. The theory behind Krol's decision was to allow for movement of the partial denture without placing too much torque on the abutment tooth.
- An illustration of the RPI design function
- RPA: mesial rest, distolingual guide plate, Akers' clasp-style retentive arm
- RPC: mesial rest, distolingual guide plate, other type of cast circumferential clasp
- So named in response to the RPI Philosophy introduced by Kratochvil and Kroll
- Edward Kennedy. 1942. Partial Denture Construction
- Oliver C. Applegate. 1949. Essentials of partial denture prosthesis.
- Davis Henderson; Victor L. Steffel. 1973. McCracken's Removable partial prosthodontics. 4th Ed.
- Fayad, MostafaI; Baig, MohamedN; Alrawaili, AbdulrazaqM (2016-12-01). "Prevalence and pattern of partial edentulism among dental patients attending College of Dentistry, Aljouf University, Saudi Arabia". Journal of International Society of Preventive and Community Dentistry. 6 (9). doi:10.4103/2231-0762.197189.