|This article does not cite any sources. (December 2009) (Learn how and when to remove this template message)|
A snow gauge is a type of instrument used by meteorologists and hydrologists in Canada to gather and measure the amount of solid precipitation (as opposed to liquid precipitation that is measured by a rain gauge) over a set period of time.
The snow gauge consists of two parts, a copper catchment container and the funnel shaped gauge itself. The actual gauge is mounted on a pipe outdoors and is approximately 1.5 m (4 ft 11 in) high, while the container is 51.5 cm (4 ft 2.25 in) long.
When snow is collected, the container is removed and replaced with a spare one.
The snow is then melted while it is still in the container, and then poured into a glass measuring graduate. While the depth of snow is normally measured in centimetres, the measurement of melted snow (water equivalent) is in millimetres.
An estimate of the snow depth can be obtained by multiplying the water equivalent by ten. However, this multiplier can vary over a wide range (many say the range is 5 to 30, but the National Snow and Ice Data Center has quoted a range as wide as 3 to 100), depending on the water content of the snow (how "dry" it is), so this only provides, at best, a rough estimate of snow depth.
The snow gauge suffers from the same problem as that of the rain gauge when conditions are windy. If the wind is strong enough, then the snow may be blown across the wind gauge and the amount of snow fallen will be under-reported. However, due to the shape and size of the funnel this is a minor problem.
If the wind is very strong and a blizzard occurs then extra snow may be blown into the gauge and the amount of snow fallen will be over-reported. In this case the observer must judge how much of the water is from snow blown into the container and how much is fallen snow.
Another problem occurs when both snow and rain fall before the observer has time to change the gauge. In all of these cases the observer must judge how much of the water is snow and how much is rain.
Other snow gauges
Remote reading gauges, such as used by weather stations, work similarly to rain gauges. They have a large catch area (such as a drum sawn in half, top to bottom) which collects snow until a given weight is collected. When this critical weight is reached, it tips and empties the snow catch. This dumping trips a switch, sending a signal. The collection then repeats. If the catch container has a heater in it, it measures the snow weight accurately. It is also possible to tip based on volume instead of weight, with appropriate fill sensing.
Another snow sensor called a snow pillow looks like a round bag lying on the ground. Inside the pillow is a liquid such as an environmentally safe antifreeze. Usually the snow pillow will be connected to a manometer. The manometer reading will vary based on how much snow is sitting on the pillow. This type of sensor works well for many locations but is more difficult to use in areas of hard blowing snow.
|Wikimedia Commons has media related to Snow gauge.|