||This article needs attention from an expert in meteorology. (November 2008)|
A Stevenson screen or instrument shelter is an enclosure to shield meteorological instruments against precipitation and direct heat radiation from outside sources, while still allowing air to circulate freely around them. It forms part of a standard weather station. The Stevenson screen holds instruments that may include thermometers (ordinary, maximum/minimum), a hygrometer, a psychrometer, a dewcell, a barometer and a thermograph. Stevenson screens may also be known as a cotton region shelter, an instrument shelter, a thermometer shelter, a thermoscreen or a thermometer screen. Its purpose is to provide a standardised environment in which to measure temperature, humidity, dewpoint and atmospheric pressure.
It was designed by Thomas Stevenson (1818–1887), a British civil engineer and father of the author Robert Louis Stevenson. Stevenson’s development of his small thermometer screen with double louvered walls on all sides and no floor was reported in 1864. After comparisons with other screens in the UK, Stevenson’s original design was modified. The modifications by Mawley of the Royal Meteorological Society in 1884 included a double roof, a floor with slanted boards, and a modification of the double louvers. This design was adopted by the British Meteorological Office and eventually other national services such as Canada. The national services developed their own variations such as the single louvered Cotton Region design in the United States.
The traditional Stevenson Screen is a box shape, constructed of wood, in a double-louvered design. However, it is possible to construct a screen using other materials and shapes, such as a pyramid. The World Meteorological Organization (WMO) agreed standard for the height of the thermometers is between 1.25 m (4 ft 1 in) and 2 m (6 ft 7 in) above the ground.
The interior size of the screen will depend on the number of instruments that are to be used. A single screen may measure 76.5 by 61 by 59.3 centimetres (30.1 by 24.0 by 23.3 in) and a double screen 76.5 by 105 by 59.3 centimetres (30.1 by 41.3 by 23.3 in). The unit is either supported by four metal or wooden legs or a wooden post.
The top of the screen was originally composed of two asbestos boards with an air space between them. These asbestos boards have generally been replaced by a laminate for health and safety reasons. The whole screen is painted with several coats of white to reflect sunlight radiation and will usually require repainting every two years.
The siting of the screen is very important to avoid data degradation by the effects of ground cover, buildings and trees: WMO 2010 recommendations, if incomplete, are a sound basis. In addition, Environment Canada, for example, recommends that the screen be placed at least twice the distance of the height of the object, e.g., 20 m from any tree that is 10 m high, or 40 ft from one 20 ft high. In the northern hemisphere, the door of the screen should always face north so as to prevent direct sunlight on the thermometers. In polar regions with twenty-four hour sunlight the observer must take care to shield the thermometers from the sun and at the same time avoiding a rise in temperature being caused by the observer's body heat.
A special type of Stevenson screen with an eye bolt on the roof is used on a ship. The unit is hung from above and remains vertical despite the movement of the vessel.
In some areas the use of single unit automatic weather stations is supplanting the Stevenson screen (and other stand-alone meteorological equipment).
- "Stevenson Screen". Environment Canada. Retrieved 2007-12-04.
- New Description of Box for Holding Thermometers, Thomas C.E. Stevenson, Journal of Scottish Meteorological Society, Vol.1, page122, June 1864
- A History of the Thermometer and Its Uses in Meteorology, W.E. Knowles Middleton, Johns Hopkins Press, Baltimore, 1966
- Edward Mawley (January 1884). "Report on Temperatures in Two Different Patterns of Stevenson Screens". Quarterly Journal of the Royal Meteorological Society. Vol.X (No.49).
- The Effect of Thermometer Screen Design on the Observed Temperature, W.R. Sparks, WMO-315, World Meteorological Society, Geneva, 1972
-  World Meteorological Organization Commission for Instruments and Methods of Observation, Fifteenth session, (CIMO-XV, 2010) WMO publication Number 1064
|Wikimedia Commons has media related to Stevenson screen.|
- Article - Temperature measurement and the Stevenson screen
- Example of a non-traditional home built Stevenson screen