Single-photon avalanche diode
A single-photon avalanche diode (SPAD) (also known as a Geiger-mode APD or G-APD) is a solid-state photodetector in which a photon-generated carrier can trigger an avalanche current due to the impact ionization mechanism. This device is able to detect low intensity signals (down to the single photon) and to signal the arrival times of the photons with a jitter of a few tens of picoseconds.
SPADs, like avalanche photodiodes (APDs), exploit the photon-triggered avalanche current of a reverse biased p-n junction to detect an incident radiation. The fundamental difference between SPAD and APD is that SPADs are specifically designed to operate with a reverse-bias voltage well above the breakdown voltage. This kind of operation is also called Geiger mode in literature, in analogy with the Geiger counter.
|This section relies largely or entirely upon a single source. (December 2010)|
SPADs are semiconductor devices based on a p-n junction reverse-biased at a voltage Va that exceeds breakdown voltage VB of the junction (Figure 1). [clarification needed]  "At this bias, the electric field is so high [higher than 3×105 V/cm] that a single charge carrier injected into the depletion layer can trigger a self-sustaining avalanche. The current rises swiftly [sub-nanosecond rise-time] to a macroscopic steady level in the milliampere range. If the primary carrier is photo-generated, the leading edge of the avalanche pulse marks [with picosecond time jitter ] the arrival time of the detected photon." The current continues until the avalanche is quenched by lowering the bias voltage VD down to or below VB: the lower electric field is no longer able to accelerate carriers to impact-ionize with lattice atoms, therefore current ceases. In order to be able to detect another photon, the bias voltage must be raised again above breakdown.
"This operation requires a suitable circuit, which has to:
- sense the leading edge of the avalanche current.
- generate a standard output pulse synchronous with the avalanche build-up.
- quench the avalanche by lowering the bias down to the breakdown voltage.
- restore the photodiode to the operative level.
This circuit is usually referred to as a quenching circuit."
The simplest quenching circuit is commonly called Passive Quenching Circuit and composed of a single resistor in series to the SPAD. This experimental setup has been employed since the early studies on the avalanche breakdown in junctions. The avalanche current self-quenches simply because it develops a voltage drop across a high-value ballast load RL (about 100 kΩ or more). After the quenching of the avalanche current, the SPAD bias VD slowly recovers to Va, and therefore the detector is ready to be ignited again. A detailed description of the quenching process is reported by Zappa et al.
A more advanced quenching scheme is called active quenching. In this case a fast discriminator senses the steep onset of the avalanche current across a 50 Ω resistor and provides a digital (CMOS, TTL, ECL, NIM) output pulse, synchronous with the photon arrival time. It then quickly reduces the bias voltage to below breakdown, then relatively quickly returns bias to above the breakdown voltage ready to sense the next photon.
Photon counting and timing
The intensity of the signal is obtained by counting (photon counting) the number of output pulses within a measurement time slot, while the time-dependent waveform of the signal is obtained by measuring the time distribution of the output pulses (photon timing). The latter is obtained by means of operating the Single Photon Avalanche Diode (SPAD) detector in Time Correlated Single Photon Counting (TCSPC) mode.
While the avalanche recovery circuit is quenching the avalanche and restoring bias, the SPAD cannot detect photons. Any photons that reach the detector during this brief period are not counted. As the number of photons increases such that the (statistical) time interval between photons gets within a factor of ten or so of the avalanche recovery time, missing counts become statistically significant and the count rate begins to depart from a linear relationship with detected light level. At this point the SPAD begins to saturate. If the light level were to increase further, ultimately to the point where the SPAD immediately avalanches the moment the avalanche recovery circuit restores bias, the count rate reaches a maximum defined purely by the avalanche recovery time (hundred million counts per second or more). This can be harmful to the SPAD as it will be experiencing avalanche current nearly continuously.
Besides photon-generated carriers, thermally-generated carriers (through generation-recombination processes within the semiconductor) can also fire the avalanche process. Therefore, it is possible to observe output pulses when the SPAD is in complete darkness. The resulting average number of counts per second is called dark count rate and is the key parameter in defining the detector noise. It is worth noting that the reciprocal of the dark count rate defines the mean time that the SPAD remains biased above breakdown before being triggered by an undesired thermal generation. Therefore, in order to work as a single-photon detector, the SPAD must be able to remain biased above breakdown for a sufficiently long time (e.g., a few milliseconds, corresponding to a count rate well under a thousand counts per second, cps).
If a SPAD is observed by an analogue curve-tracer, it is possible to observe a bifurcation of the current-voltage characteristics beyond breakdown, during the voltage sweeps applied to the device. When the avalanche is triggered, the SPAD sustains the avalanche current (on-branch), instead when no carrier has been generated (by a photon or a thermal generation), no charge flows through the SPAD (off-branch). If the SPAD is triggered during a sweep above breakdown, a transition from the off-branch to the on-branch can be easily observed (like a "flickering").
Comparison with APDs
Both APDs and SPADs are reverse biased semiconductor p-n junctions. However, APDs are biased close to, but below the breakdown voltage of the semiconductor. This high electric field provides an internal multiplication gain only on the order of few hundreds, since the avalanche process is not diverging as in SPADs. The resulting avalanche current intensity is linearly related to the optical signal intensity. A SPAD however operates with a bias voltage above the breakdown voltage. Because the device is operating in this unstable above-breakdown regime, a single photon (or a single dark current electron) can set off a significant avalanche of electrons. Practically, this means that in an APD a single photon produces only tens or few hundreds of electrons, but in a SPAD a single photon triggers a current in the mA region (billions of billions of electrons per second) that can be easily "counted".
Therefore, while the APD is a linear amplifier for the input optical signal with limited gain, the SPAD is a trigger device so the gain concept is meaningless.
- F. Zappa, S. Cova; M. Ghioni; A. Lacaita; C. Samori (1996-04-20). "Avalanche photodiodes and quenching circuits for single-photon detection" (PDF). Applied Optics 35 (12): 1956–1976. doi:10.1364/AO.35.001956. PMID 21085320.
- Eisele, A.; Henderson, R.; Schmidtke, B.; Funk, T.; Grant, L.; Richardson, J.; Freude, W.: 185 MHz count rate, 139 dB dynamic range single-photon avalanche diode with active quenching circuit in 130 nm CMOS technology Intern. Image Sensor Workshop (IISW'11), Hokkaido, Japan; Paper R43; June 2011
- Fishburn, Matthew (2012). Fundamentals of CMOS Single-Photon Avalanche Diodes. Delft, the Netherlands: Delft University of Technology. pp. 4–5. ISBN 978-94-91030-29-1. Retrieved October 10, 2012.