Capacitive micromachined ultrasonic transducers
This article has multiple issues. Please help improve it or discuss these issues on the talk page. (Learn how and when to remove these template messages)(Learn how and when to remove this template message)
Capacitive micromachined ultrasonic transducers (CMUT) is a relatively new concept in the field of ultrasonic transducers. Most of the commercial ultrasonic transducers today are based on piezoelectricity. CMUTs are the transducers where the energy transduction is due to change in capacitance. CMUTs are constructed on silicon using micromachining techniques. A cavity is formed in a silicon substrate, and a thin layer suspended on the top of the cavity serves as a membrane on which a metallized layer acts an electrode, together with the silicon substrate which serves as a bottom electrode.
If an AC signal is applied across the biased electrodes, the vibrating membrane will produce ultrasonic waves in the medium of interest. In this way it works as a transmitter. On the other hand, if ultrasonic waves are applied on the membrane of a biased CMUT, it will generate alternating signal as the capacitance of the CMUT is varied. In this way, it works as a receiver of ultrasonic waves.
As CMUTs are micromachined devices, it is easier to construct 2D and 3D arrays of transducers using this technology. This means large numbers of CMUTs could be included in a transducer array providing larger bandwidth compared to other transducer technologies. To achieve a high frequency operation using CMUTs is easier due to its smaller dimensions. The frequency of operation depends on the cell size (cavity of membrane), and on the stiffness of the material used as a membrane. As it is built on silicon, the integration of electronics would be easier for the CMUTs compared to other transducer technologies. The properties to use in high frequency with large bandwidth makes it a good choice to use as a transducer in medical imaging, especially in an intravascular ultrasound (IVUS). Because of its broader bandwidth, it could be used in second-harmonic imaging. Also some experiments have been performed to use CMUTs as hydrophones.
- 1 Fabrication methods
- 2 Benchmarking
- 3 Applications
- 4 Challenges in CMUT operation
- 5 External links
- 6 References
Sacrificial release surface micromachining
- complicated manufacturing process for constructing and sealing etch/drainage channels of the sacrificial material;
- the need for sacrificial-release channels reduces the available space for transducers, thereby reducing the achievable sound generation capability;
- limited control of layers' thickness during the manufacturing process;
- limited cavity thickness due to residues of fluid inside the cell cavity, which can cause stiction between the upper and lower parts of the cell, if the cell is not thick enough.
Wafer bonding is the most popular method. In this method, a CMUT is built from two separate wafers, which are later bonded to achieve cells with cavities.
- Fusion-bonding of wafers.
- Multi-user MUMPS (polyMUMPS) process. CMUTs manufactured in the multi-user MUMPS were reported to have reduced performance, such as relatively low resonating frequency.
In this method the manufacturing is performed in reverse order, compared to the traditional way. The structural membrane is in silicon-nitride LPCVD, but the entire process is low-temperature, so it is CMOS-compatible. There are no etch-hole on the radiating surface of the device. The connection pads are on the back of the device, without using of through VIAs in the silicon, and the silicon substrate is completely removed. A custom acoustic backing is used to improve acoustic performances of the device. The process uses few masks (7-8).
Integration with electrical circuits
As mentioned earlier, one of the significant advantages of CMUTs over piezoelectric transducers is the ability to integrate CMUTs with electrical circuits, using existing manufacturing methods.
CMUT on CMOS
CMUT performance is benchmarked using pitch-catch and pulse-echo experiments, and operation uniformity is tested in air and in immersion. In a pitch-catch experiment, the transducer is benchmarked using a hydrophone, and in a pulse-echo experiment, the transducer is used both for transmitting and receiving, while comparing the measured signal to the hydrophone response.
Challenges in CMUT operation
Cross-talk between array elements
- "General Description and Advantages of CMUTs". Stanford University. Archived from the original on 20 July 2011. Retrieved 7 February 2011.
- "Capacitive Micromachined Ultrasonic Transducers: Next-Generation Arrays for Acoustic Imaging" (PDF). IEEE Transactions on UFFC,Vol.49,published November 2002.
- A. S. Ergun, Y. Huang, X. Zhuang, O. Oralkan, G. G. Yaralioglu, and B. T. Khuri-Yakub, "Capacitive Micromachined Ultrasonic Transducers: Fabrication Technology," IEEE Trans. Ultrason., Ferroelect., Freq. Contr., vol. 52, pp. 2242-2258, 2005.
- Ergun, AS; Huang, Y; Zhuang, X (2005). "Capacitive micromachined ultrasonic transducers: fabrication technology". IEEE Trans Ultrason Ferroelectr Freq Control. 52 (12): 2242–58.
- Y. Huang, A. S. Ergun, E. Haeggstrom, M. H. Badin and B. T. Khuri-Yakub, "Fabricating Capacitive Micromachined Ultrasonic Transducers with Wafer-Bonding Technology", J. MEMS, vol. 12, pp. 128-137, 2003.
- A. Logan, J. T. W. Yeow, "Fabricating Capacitive Micromachined Ultrasonic Transducers with a Novel Silicon-Nitride-Based Wafer Bonding Process," IEEE Trans. Ultrason., Ferroelect., Freq. Contr., vol. 56, pp. 1074-1084, 2009.
- K. Midtbø, A. Rønnekleiv and D. T. Wang, "Fabrication and characterization of CMUTs realized by wafer bonding", Proc. IEEE Ultrason. Symp., vol. 1, 2006, pp. 938-941.
- K. K. Park, H. J. Lee, M. Kupnik, O.Oralkan, and B. T. Khuri-Yakub, "Fabricating Capacitive Micromachined Ultrasonic Transducers with Direct Wafer-Bonding and LOCOS Technology," 2008 IEEE MEMS Conf., pp. 339-342, 2008.
- Liu, Jessica; Oakley, Clyde; Shandas, Robin (2009). "Capacitive micromachined ultrasonic transducers using commercial multi-user MUMPs process: Capability and limitations". Ultrasonics. 49 (8): 765–773. doi:10.1016/j.ultras.2009.06.003. ISSN 0041-624X. PMC 2783530.
- S. Olcum, K. Oguz, M. N. Senlik, F. Y. Yamaner, A. Bozkurt, A. Atalar and H. Koymen, "Wafer Bonded Capacitive Micromachined Underwater Transducers", 2009 IEEE Ultrasonics Symposium, pp. 976-979, 2009.
- A. Coppa, E. Cianci, V. Foglietti, G. Caliano, and M. Pappalardo, "Building CMUTs for Imaging Applications From Top to Bottom," Microelect. Eng., vol. 84, pp. 1312-1314, 2007.
- A. Caronti, A. Coppa, A. Savoia, C. Longo, P. Gatta, B. Mauti, A. Corbo, B. Calabrese, G. Bollino, A. Paz, G. Caliano, and M. Pappalardo, "Curvilinear Capacitive Micromachined Ultrasonic Transducer (CMUT) Array Fabricated Using a Reverse Process," 2008 IEEE Ultrasonics Symposium, pp. 2092-2095, 2008.
- Patent US7790490