An integrating sphere (also known as an Ulbricht sphere) is an optical component consisting of a hollow spherical cavity with its interior covered with a diffuse white reflective coating, with small holes for entrance and exit ports. Its relevant property is a uniform scattering or diffusing effect. Light rays incident on any point on the inner surface are, by multiple scattering reflections, distributed equally to all other points. The effects of the original direction of light are minimized. An integrating sphere may be thought of as a diffuser which preserves power but destroys spatial information. It is typically used with some light source and a detector for optical power measurement. A similar device is the focusing or Coblentz sphere, which differs in that it has a mirror-like (specular) inner surface rather than a diffuse inner surface.
The practical implementation of the integrating sphere was due to work by R. Ulbricht (1849–1923), published in 1900. It has become a standard instrument in photometry and radiometry. It has the advantage over a goniophotometer for measuring the light produced by a source that total power can be obtained in a single measurement.
The theory of a light-collecting cubical box was described by W. E. Sumpner in 1910.
The theory of integrating spheres is based on these assumptions:
- Light hitting the sides of the sphere is scattered in a diffuse way i.e. Lambertian reflectance
- Only light that has been diffused in the sphere hits the ports or detectors used for probing the light
Using these assumptions the sphere multiplier can be calculated. This number is the average number of times a photon is scattered in the sphere, before it is absorbed in the coating or escapes through a port. This number increases with the reflectivity of the sphere coating and decreases with the ratio between the total area of ports and other absorbing objects and the sphere inner area. To get a high homogeneity a recommended sphere multiplier is 10-15. The theory further states that if the above criteria are fulfilled then the irradiance on any area element on the sphere will be proportional to the total radiant flux input to the sphere. Absolute measurements of for instance luminous flux can then be by done by measuring a known light source and determining the transfer function or calibration curve.
Light scattered by the interior of the integrating sphere is evenly distributed over all angles. The integrating sphere is used in optical measurements. The total power (flux) of a light source can be measured without inaccuracy caused by the directional characteristics of the source, or the measurement device. Reflection and absorption of samples can be studied. The sphere creates a reference radiation source that can be used to provide a photometric standard.
Integrating spheres are used for a variety of optical, photometric or radiometric measurements. They are used to measure the total light radiated in all directions from a lamp. An integrating sphere can be used to measure the diffuse reflectance of surfaces, providing an average over all angles of illumination and observation. An integrating sphere can be used to create a light source with apparent intensity uniform over all positions within its circular aperture, and independent of direction except for the cosine function inherent to ideally diffuse radiating surfaces (Lambertian surfaces).
Since all the light incident on the input port is collected, a detector connected to an integrating sphere can accurately measure the sum of all the ambient light incident on a small circular aperture. The total power of a laser beam can be measured, free from the effects of beam shape, incident direction, and incident position, as well as polarization.
The optical properties of the lining of the sphere greatly affect its accuracy. Different coatings must be used at visible, infrared and ultraviolet wavelengths. High-powered illumination sources may heat or damage the coating, so an integrating sphere will be rated for a maximum level of incident power. Various coating materials are used. Early experimenters used a deposit of magnesium oxide. Barium sulfate has a usefully flat reflectance over the visible spectrum. Finely-deposited gold is used for infrared measurements. Various proprietary PTFE compounds are also used for visible light measurements.
An important requirement for the coating material is the absence of fluorescence. Fluorescent materials absorb short-wavelength light and re-emit light at longer wavelengths. Due to the many scatterings this effect is much more pronounced in an integrating sphere than for materials irradiated normally.
The theory of the integrating sphere assumes a uniform inside surface with diffuse reflectivity approaching 100%. Openings where light can exit or enter, used for detectors and sources, are normally called ports. The total area of all ports must be small, less than about 5% of the surface area of the sphere, for the theoretical assumptions to be valid. Unused ports should therefore have matching plugs, with the interior surface of the plug coated with the same highly white material as the rest of the sphere.
Integrating spheres vary in size from a few centimeters in diameter up to a few meters in diameter. Smaller spheres are typically used to diffuse incoming radiation, while larger spheres are used to measure integrating properties like the luminous flux of a lamp or luminaries which is then placed inside the sphere.
If the entering light is incoherent (rather than a laser beam), then it typically fills the source-port, and the ratio of source-port area to detector-port area is relevant.
Baffles are normally inserted in the sphere to block the direct path of light from a source-port to a detector-port, since this light will have non-uniform distribution.
- James M. Palmer, Barbara G. Grant The Art of Radiometry, SPIE Press, 2010,ISBN 978-0-8194-7245-8, page 5
- H. Buckley, "The Whitened Cube as a Precision Integrating Photometer" (1920) Proceedings of the Institution of Electrical Engineers 59 (London)
Hopoo optoelectronics technology Co.Ltd, Hopoo brand Integrating sphere
- RP Photonics, Encyclopedia of Laser Physics and Technology, Integrating spheres
- Brian Lai, Labsphere, Integrating Sphere and Applications Training[dead link]
- Labsphere, Inc., A Guide to Integrating Sphere Theory and Applications
- Pike Technologies, Integrating Spheres – Introduction and Theory, Pike Technologies Application Note
- Newport, Flange Mount Integrating Spheres
- Whitehead, Lorne A.; Mossman, Michele A. (2006). "Jack O'Lanterns and integrating spheres: Halloween Physics". American Journal of Physics 74 (6): 537–541. Bibcode:2006AmJPh..74..537W. doi:10.1119/1.2190687.
- Ducharme, Alfred; Daniels, Arnold; Grann, Eric; Boreman, Glenn (1997). "Design of an Integrating Sphere as a Uniform Illumination Source". IEEE Transactions on Education 40 (2): 131–134. Bibcode:1997ITEdu..40..131D. doi:10.1109/13.572326.
- Peter Hiscocks, Integrating Sphere for Luminance Calibration, March 2012