Active optics is a technology used with reflecting telescopes developed in the 1980s, which actively shapes a telescope's mirrors to prevent deformation due to external influences such as wind, temperature, mechanical stress. Without active optics, the construction of 8 metre class telescopes is not possible, nor would telescopes with segmented mirrors be feasible.
This method is used by, among others, the Nordic Optical Telescope, the New Technology Telescope, the Telescopio Nazionale Galileo and the Keck telescopes, as well as all of the largest telescopes built since the mid-1990s.
Active optics is not to be confused with adaptive optics, which operates at a shorter timescale and corrects atmospheric distortions.
Most modern telescopes are reflectors, with the primary element being a very large mirror. Historically, primary mirrors were quite thick in order to maintain the correct surface figure in spite of forces tending to deform it, like wind and the mirror's own weight. This limited their maximum diameter to 5 or 6 metres (200 or 230 inches), such as Palomar Observatory's Hale telescope.
A new generation of telescopes built since the 1980s use thin, lighter weight mirrors instead. They are too thin to maintain themselves rigidly in the correct shape, so an array of actuators is attached to the rear side of the mirror. The actuators apply variable forces to the mirror body to keep the reflecting surface in the correct shape over repositioning. The telescope may also be segmented into multiple smaller mirrors, which reduce the sagging due to weight that occurs for large, monolithic mirrors.
The combination of actuators, an image quality detector, and a computer to control the actuators to obtain the best possible image, is called active optics.
The name active optics means that the system keeps a mirror (usually the primary) in its optimal shape against environmental forces such as wind, sag, thermal expansion, and telescope axis deformation. Active optics compensate for distorting forces that change relatively slowly, roughly on timescales of seconds. The telescope is therefore actively still, in its optimal shape.
Comparison with adaptive optics
Active optics should not be confused with adaptive optics, which operates on a much shorter timescale to compensate for atmospheric effects, rather than for mirror deformation. The influences that active optics compensate (temperature, gravity) are intrinsically slower (1 Hz) and have a larger amplitude in aberration. Adaptive optics on the other hand corrects for atmospheric distortions that affect the image at 100–1000 Hz (the Greenwood frequency, depending on wavelength and weather conditions). These corrections need to be much faster, but also have smaller amplitude. Because of this, adaptive optics uses smaller corrective mirrors. This used to be a separate mirror not integrated in the telescope's light path, but nowadays this can be the second, third or fourth mirror in a telescope.
Complicated laser set-ups and interferometers can also be actively stabilized.
A small part of the beam leaks through beam steering mirrors and a four-quadrant-diode is used to measure the position of a laser beam and another in the focal plane behind a lens is used to measure the direction. The system can be sped up or made more noise-immune by using a PID controller. For pulsed lasers the controller should be locked to the repetition rate. A continuous (non-pulsed) pilot beam can be used to allow for up to 10 kHz bandwidth of stabilization (against vibrations, air turbulence, and acoustic noise) for low repetition rate lasers.
Sometimes Fabry–Pérot interferometers have to be adjusted in length to pass a given wavelength. Therefore, the reflected light is extracted by means of a Faraday rotator and a polarizer. Small changes of the incident wavelength generated by an acousto-optic modulator or interference with a fraction of the incoming radiation delivers the information whether the Fabry Perot is too long or to short.
Long optical cavities are very sensitive to the mirror alignment. A control circuit can be used to peak power. One possibility is to perform small rotations with one end mirror. If this rotation is about the optimum position, no power oscillation occurs. Any beam pointing oscillation can be removed using the beam steering mechanism mentioned above.
- Adaptive optics – faster technology for smaller aberrations.
- Active surface – similar technology for radio telescopes.
- List of telescope parts and construction
- Hardy, John W. (June 1977). "Active optics: A new technology for the control of light". Proceedings of the IEEE: 110. Bibcode:1978IEEEP..66..651H.
- Andersen, T.; Andersen, T.; Larsen, O. B.; Owner-Petersen, M.; Steenberg, K. (April 1992). Ulrich, Marie-Helene, ed. Active Optics on the Nordic Optical Telescope. ESO Conference and Workshop Proceedings. Progress in Telescope and Instrumentation Technologies. pp. 311–314. Bibcode:1992ESOC...42..311A.
- "ESO Awards Contract for E-ELT Adaptive Mirror Design Study". ESO Announcements. Retrieved 25 May 2012.
- Greenwood, Darryl P. (March 1977). "Bandwidth specification for adaptive optics systems" (PDF). Journal of the Optical Society of America. 67 (3): 390–303. Bibcode:1977JOSA...67..390G. doi:10.1364/JOSA.67.000390.
- Riccardi, Armando; Brusa, Guido; Salinari, Piero; Gallieni, Daniele; Biasi, Roberto; Andrighettoni, Mario; Martin, Hubert M (February 2003). "Adaptive secondary mirrors for the Large Binocular Telescope" (PDF). Proceedings of the SPIE. Adaptive Optical System Technologies II. SPIE. 4839: 721–732. Bibcode:2003SPIE.4839..721R. doi:10.1117/12.458961.
- Salinari, P.; Del Vecchio, C.; Biliotti, V. (August 1994). A Study of an Adaptive Secondary Mirror. ESO Conference and Workshop Proceedings. Active and adaptive optics. Garching, Germany: ESO. pp. 247–253. Bibcode:1994ESOC...48..247S.
- Crépy, B.; et al. (June 2009). The M4 adaptive unit for the E-ELT. 1st AO4ELT conference – Adaptative Optics for Extremely Large Telescopes Proceedings. Paris, France: EDP Sciences. Bibcode:2010aoel.confE6001C. doi:10.1051/ao4elt/201006001.
- "Research Partnership Advances X-ray Active Optics" (PDF). adaptiveoptics.org. March 2005. Archived from the original on March 2005. Retrieved 2 June 2011.