A volumetric flask (measuring flask or graduated flask) is a piece of laboratory glassware, a type of laboratory flask, calibrated to contain a precise volume at a particular temperature. Volumetric flasks are used for precise dilutions and preparation of standard solutions. These flasks are usually pear-shaped, with a flat bottom, and made of glass or plastic. The flask's mouth is either furnished with a plastic snap/screw cap or fitted with a joint to accommodate a PTFE or glass stopper. The neck of the volumetric flasks is elongated and narrow with an etched ring graduation marking. The marking indicates the volume of liquid contained when filled up to that point. The marking is typically calibrated "to contain" (marked "TC" or "IN") at 20 °C and indicated correspondingly on a label. The flask's label also indicates the nominal volume, tolerance, precision class, relevant manufacturing standard and the manufacturer’s logo. The volumetric flasks are of various sizes, containing from 1 to 10 000 mL of liquid.
Calibration and toleration standards for volumetric flasks are defined in the following standard specifications and practices: ASTM E288, E542, E694, ISO 1042, and GOST 1770-74. According to these specifications, volumetric flasks come in two different classes. The higher standard flasks (Class A, Class 1, USP or equivalent depending on the country) are made with a more accurately placed graduation mark, and have a unique serial number for traceability. Where this is not required, a lower standard (Class B or equivalent) is used for qualitative or educational work.
A modification of the volumetric flask exists for dealing with large quantities of solids that are to be transferred into and dissolved in a volumetric vessel. Such flask has an wide mouth and known as Kohlrausch volumetric flask. This kind of volumetric flask found particular application in the procedure for the sugar content analysis in sugar beet.
While conventional volumetric flasks have a single mark, volumetric tests in the analytical and food chemistry may employ specialized two-mark volumetric flasks used to combine two accurately measured volumes.
- ASTM E288 - 10 Standard Specification for Laboratory Glass Volumetric Flasks.
- ASTM E542 - 01(2007) Standard Practice for Calibration of Laboratory Volumetric Apparatus.
- ASTM E694 - 99(2010) Standard Specification for Laboratory Glass Volumetric Apparatus.
- ISO 1042:1998 Laboratory glassware -- One-mark volumetric flasks.
- GOST 1770-74 Laboratory volumetric glassware. Cylinders, beakers, enmma roberts s ewird measuring flasks, test tubes. General specifications.
- ASTM C188 - 09 Standard Test Method for Density of Hydraulic Cement.
Hughes, J. C. (Apr. 1959), Testing of Glass Volumetric Apparatus, NBS Circular 602., Natl. Bur. Stand. (U.S.)
Houser, J. F. (Aug. 1973), Procedures for the Calibration of Volumetric Test Measures, NBSIR 73-287., Natl. Bur. Stand. (U.S.)
Lembeck, J. (Oct. 1974), Calibration of Small Volumetric Laboratory Glassware, NBSIR 74-461., Natl. Bur. Stand. (U.S.)