Mean corpuscular hemoglobin concentration
It is calculated by dividing the haemoglobin by the haematocrit. Reference ranges for blood tests are 32 to 36 g/dL (320 to 360g/L), or between 4.81 and 5.58 mmol/L. It is thus a mass or molar concentration. Still, many instances measure MCHC in percentage (%), as if it were a mass fraction (mHb / mRBC). Numerically, however, the MCHC in g/dL and the mass fraction of haemoglobin in red blood cells in % are identical, assuming an RBC density of 1g/mL and negligible haemoglobin in plasma.
A low MCHC can be interpreted as identifying decreased production of hemoglobin. MCHC can be normal even when hemoglobin production is decreased (such as in iron deficiency) due to a calculation artifact. MCHC can be elevated ("polychromatic") in hereditary spherocytosis, sickle cell disease and homozygous haemoglobin C disease, depending upon the hemocytometer. MCHC can be elevated in some megaloblastic anemias. MCHC can be falsely elevated when there is agglutination of red cells (falsely lowering the measured RBC) or when there is opacifaction of the plasma (falsely increasing the measured hemoglobin). Causes of plasma opacification that can falsely increase the MCHC include hyperbilirubinemia, hypertryglyceridemia, and free hemoglobin in the plasma (due to hemolysis).
Complications with cold agglutinin
Because of the way automated analysers count blood cells, a very high MCHC (greater than about 370 g/L) may indicate the blood is from someone with a cold agglutinin. This means that when their blood gets colder than 37 °C it starts to clump together. As a result, the analyzer may incorrectly report a low number of very dense red blood cells for blood samples in which agglutination has occurred.
This problem is usually picked up by the laboratory before the result is reported. The blood is warmed until the cells separate from each other, and quickly put through the machine while still warm. This is the most sensitive test for iron deficiency anaemia.
There are four steps to perform when an increase MCHC(>370 g/L or 37.0 g/dL) is received from the analyzer:
- Remix the EDTA tube—if the MCHC corrects, report corrected results
- Incubation at 37 °C—if the MCHC corrects, report corrected results and comment on possible cold agglutinin
- Saline replacement: Replace plasma with same amount of saline to exclude interference e.g. Lipemia and Auto-immune antibodies—if the MCHC corrects, report corrected results and comment on Lipemia
- Check the slide for spherocytosis (e.g. in hereditary spherocytosis, among other causes)
Auto-agglutination: Falsely ↓RBC and ↑MCV
Lipaemia: Falsely ↑haemoglobin.
Haemolysis: Will falsely increase the MCHC (measured haemoglobin is proportionally higher than HCT or PCV) and falsely decrease the calculated haemoglobin (fewer intact RBC)
Other: Heinz bodies (many, particularly if large) may falsely increase the MCHC but not the MCH. Agglutination: Falsely increases the MCHC (measured haemoglobin is proportionally higher than HCT). The MCH is more accurate in this setting. Excess EDTA: Dehydrates RBC, falsely increasing MCHC and MCH.
|Hb||100 grams/liter||10 grams/deciliter||(deci- is 10−1)|
|RBC||5E+12 cells/liter||5E+6 cells/μL||(micro is 10−6)|
|MCV = (Hct/100) / RBC||8E-14 liters/cell||80 femtoliters/cell||(femto- is 10−15)|
|MCH = Hb / RBC||2E-11 grams/cell||20 picograms/cell||(pico- is 10−12)|
|MCHC = Hb / (Hct/100)||250 grams/liter||25 grams/deciliter||(deci is 10−1)|
- MedlinePlus Encyclopedia RBC indices
- Blood Test Results - Normal Ranges Bloodbook.Com. Retrieved on Jan 7, 2009
- MedicineNet > Definition of MCHC Last Editorial Review: 7/21/1999
- Hill, Valerie L.; Simpson, Virginia Z.; Higgins, Jeanette M.; Hu, Zonghui; Stevens, Randy A.; Metcalf, Julie A.; Baseler, Michael (2017-01-28). "Evaluation of the Performance of the Sysmex XT-2000i Hematology Analyzer With Whole Bloods Stored at Room Temperature". Laboratory medicine. 40 (12): 709–718. doi:10.1309/T0FJYP2RBXEHX4. ISSN 0007-5027. PMC . PMID 20431699.
- Rifkind, David; Cohen, Alan S. (2002). The Pediatric Abacus. Informa Healthcare. p. 54. ISBN 1-84214-147-3.