This article has multiple issues. Please help improve it or discuss these issues on the talk page. (Learn how and when to remove these template messages)
|Other names||Magnetic mixer|
|Related items||Vortex mixer, static mixer|
A magnetic stirrer or magnetic mixer is a laboratory device that employs a rotating magnetic field to cause a stir bar (or flea) immersed in a liquid to spin very quickly, thus stirring it. The rotating field may be created either by a rotating magnet or a set of stationary electromagnets, placed beneath the vessel with the liquid. It is used in chemistry and biology where other forms of stirring, such as motorized stirrers and stirring rods, may not be viable for use.
A magnetic stirrer consists of a magnetic bar placed within the liquid which provides the stirring action. The stir bar's motion is driven by another rotating magnet or assembly of electromagnets in the stirrer device, beneath the vessel containing the liquid. Stir bars are typically coated in PTFE, or, less often, in glass; the coatings are intended to be chemically inert, not contaminating or reacting with the reaction mixture they are in. Glass may be viable as an alternative if PTFE is unsuitable. In dissolving metal reductions that use an alkali metal dissolved in a primary amine, PTFE may be attacked to some extent. Birch reductions (a common dissolving metal reduction) are often conducted in a glass vessel, thus indicating that a glass stir bar would likewise be compatible. Glass can be attacked by strong alkali (such as lye) depending on heat, exposure time, and concentration.
Magnetic stirrers are bar-shaped and usually octagonal or circular in cross-section, although a variety of special shapes exist for more efficient stirring. Many stir bars have a pivot ring around the center on which they rotate. The smallest are only a few millimeters long and the largest a few centimeters.
Laboratory hot plates often serve a dual purpose by incorporating both the stirring assembly and a heating element. Such heating elements may range in power from a few hundred to a few thousand watts, and allow the reaction flask to be heated and stirred at the same time. The maximum reachable fluid temperature depends on the size of the flask, the quantity of solution to be heated, the power of the heating element, and amount of insulation provided to the system.
Because of its small size, a stirring bar is more easily cleaned and sterilized than other stirring devices. They do not require lubricants which could contaminate the reaction vessel and the product.
A stir bar retriever is a separate magnet on the end of a long stick (also coated with chemically inert PTFE) which can be used to remove stir bars from a vessel.
The first patent for a magnetic mixer is US 1,242,493, issued 9 October 1917 to Richard H. Stringham of Bountiful, Utah, U.S. Mr. Stringham's mixer used stationary electromagnets in the base, rather than a rotating permanent magnet, to rotate the stirrer.
Arthur Rosinger of Newark, New Jersey, U.S. obtained US Patent 2,350,534, titled Magnetic Stirrer on 6 June 1944, having filed an application therefore on 5 October 1942. Mr. Rosinger's patent includes a description of a coated bar magnet placed in a vessel, which is driven by a rotating magnet in a base below the vessel. Mr. Rosinger also explains in his patent that coating the magnet in plastic or covering it with glass or porcelain makes it chemically inert.
The plastic-coated bar magnet was independently invented in the late 1940s by Edward McLaughlin, of the Torpedo Experimental Establishment (TEE), Greenock, Scotland, who named it the 'flea' because of the way it jumps about if the rotating magnet is driven too fast.
The first multi-point magnetic stirrer was developed and patented by Salvador Bonet of SBS Company in 1977. He also introduced the practice of noting the denomination of stirring power in "litres of water", which is a market standard today.
Uses and limitations
Magnetic stirrers are often used in chemistry and biology, where they can be used to stir hermetically closed vessels or systems without the need for complicated rotary seals. They are preferred over gear-driven motorized stirrers because they are quieter, more efficient, and have no moving external parts to break or wear out (other than the simple bar magnet itself). Magnetic stir bars work well in glass vessels commonly used for chemical reactions, as glass does not appreciably affect a magnetic field. The limited size of the bar means that magnetic stirrers can only be used for relatively small experiments, of 4 litres or less. Stir bars also have difficulty in dealing with viscous liquids or thick suspensions. For larger volumes or more viscous liquids, some sort of mechanical stirring (e.g., an overhead stirrer) is typically needed. In synthetic chemistry, a combined magnetic stirrer/heater, equipped with a built-in temperature control mechanism and temperature probe, is commonly used with a heating bath (commonly oil, sand, or low-melting metal) or cooling bath (commonly water, ice, or an organic liquid mixed with liquid nitrogen or dry ice as coolant), allowing reactions vessels placed in the bath to be maintained at temperatures between approximately –120 and 250 °C.
- "Stir Bars". University of Colorado at Boulder. Archived from the original on 7 June 2014. Retrieved 16 February 2013.
- Del Campo FJ, Neudeck A, Compton RG, Marken F, Bull SD, Davies SG (July 2001). "Low-temperature sonoelectrochemical processes: Part 2: Generation of solvated electrons and Birch reduction processes under high mass transport conditions in liquid ammonia". Journal of Electroanalytical Chemistry. 507 (1): 144–151. doi:10.1016/S0022-0728(01)00368-0.
- de Los Reyes CA, Smith McWilliams AD, Hernández K, Walz-Mitra KL, Ergülen S, Pasquali M, Martí AA (March 2019). "Adverse Effect of PTFE Stir Bars on the Covalent Functionalization of Carbon and Boron Nitride Nanotubes Using Billups-Birch Reduction Conditions". ACS Omega. 4 (3): 5098–5106. doi:10.1021/acsomega.8b03677. PMC 6648908. PMID 31459687.
- Girolami GS, Thomas B, Rauchfuss TB, Angelici RJ (1 August 1999). Synthesis and Technique in Inorganic Chemistry: A Laboratory Manual (3 ed.). University Science Books. p. 87. ISBN 978-0-935702-48-4. Retrieved 23 April 2013.
- US 2350534, Rosinger A, "Magnetic Stirrer", issued 6 June 1944
|Wikimedia Commons has media related to Magnetic stirrers.|
- Short video of a homemade stir plate. Creative Commons Attribution license (reuse allowed).