Matrix-assisted laser desorption/ionization
In mass spectrometry, matrix-assisted laser desorption/ionization (MALDI) is an ionization technique that uses a laser energy absorbing matrix to create ions from large molecules with minimal fragmentation. It has been applied to the analysis of biomolecules (biopolymers such as DNA, proteins, peptides and carbohydrates) and various organic molecules (such as polymers, dendrimers and other macromolecules), which tend to be fragile and fragment when ionized by more conventional ionization methods. It is similar in character to electrospray ionization (ESI) in that both techniques are relatively soft (low fragmentation) ways of obtaining ions of large molecules in the gas phase, though MALDI typically produces far fewer multi-charged ions.
MALDI methodology is a three-step process. First, the sample is mixed with a suitable matrix material and applied to a metal plate. Second, a pulsed laser irradiates the sample, triggering ablation and desorption of the sample and matrix material. Finally, the analyte molecules are ionized by being protonated or deprotonated in the hot plume of ablated gases, and then they can be accelerated into whichever mass spectrometer is used to analyse them.
The term matrix-assisted laser desorption ionization (MALDI) was coined in 1985 by Franz Hillenkamp, Michael Karas and their colleagues. These researchers found that the amino acid alanine could be ionized more easily if it was mixed with the amino acid tryptophan and irradiated with a pulsed 266 nm laser. The tryptophan was absorbing the laser energy and helping to ionize the non-absorbing alanine. Peptides up to the 2843 Da peptide melittin could be ionized when mixed with this kind of "matrix". The breakthrough for large molecule laser desorption ionization came in 1987 when Koichi Tanaka of Shimadzu Corporation and his co-workers used what they called the "ultra fine metal plus liquid matrix method" that combined 30 nm cobalt particles in glycerol with a 337 nm nitrogen laser for ionization. Using this laser and matrix combination, Tanaka was able to ionize biomolecules as large as the 34,472 Da protein carboxypeptidase-A. Tanaka received one-quarter of the 2002 Nobel Prize in Chemistry for demonstrating that, with the proper combination of laser wavelength and matrix, a protein can be ionized. Karas and Hillenkamp were subsequently able to ionize the 67 kDa protein albumin using a nicotinic acid matrix and a 266 nm laser. Further improvements were realized through the use of a 355 nm laser and the cinnamic acid derivatives ferulic acid, caffeic acid and sinapinic acid as the matrix. The availability of small and relatively inexpensive nitrogen lasers operating at 337 nm wavelength and the first commercial instruments introduced in the early 1990s brought MALDI to an increasing number of researchers. Today, mostly organic matrices are used for MALDI mass spectrometry.
|Compound||Other names||Solvent||Wavelength (nm)||Applications|
|2,5-dihydroxy benzoic acid (gentisic acid)||DHB, gentisic acid||acetonitrile, water, methanol, acetone, chloroform||337, 355, 266||peptides, nucleotides, oligonucleotides, oligosaccharides|
|3,5-dimethoxy-4-hydroxycinnamic acid||sinapic acid; sinapinic acid; SA||acetonitrile, water, acetone, chloroform||337, 355, 266||peptides, proteins, lipids|
|4-hydroxy-3-methoxycinnamic acid||ferulic acid||acetonitrile, water, propanol||337, 355, 266||proteins|
|α-cyano-4-hydroxycinnamic acid||CHCA||acetonitrile, water, ethanol, acetone||337, 355||peptides, lipids, nucleotides|
|3-hydroxy picolinic acid||HPA||Ethanol||337, 355||oligonucleotides|
The matrix consists of crystallized molecules, of which the three most commonly used are sinapinic acid, α-cyano-4-hydroxycinnamic acid (α-CHCA, alpha-cyano or alpha-matrix) and 2,5-dihydroxybenzoic acid (DHB). A solution of one of these molecules is made, often in a mixture of highly purified water and an organic solvent such as acetonitrile (ACN) or ethanol. A counter ion source such as trifluoroacetic acid (TFA) is usually added to generate the [M+H] ions. A good example of a matrix-solution would be 20 mg/mL sinapinic acid in ACN:water:TFA (50:50:0.1).
The identification of suitable matrix compounds is determined to some extent by trial and error, but they are based on some specific molecular design considerations. They are of a fairly low molecular weight (to allow easy vaporization), but are large enough (with a low enough vapor pressure) not to evaporate during sample preparation or while standing in the mass spectrometer. They are often acidic, therefore act as a proton source to encourage ionization of the analyte. Basic matrices have also been reported. They have a strong optical absorption in either the UV or IR range, so that they rapidly and efficiently absorb the laser irradiation. This efficiency is commonly associated with chemical structures incorporating several conjugated double bonds, as seen in the structure of cinnamic acid. They are functionalized with polar groups, allowing their use in aqueous solutions. They typically contain a chromophore.
The matrix solution is mixed with the analyte (e.g. protein-sample). A mixture of water and organic solvent allows both hydrophobic and water-soluble (hydrophilic) molecules to dissolve into the solution. This solution is spotted onto a MALDI plate (usually a metal plate designed for this purpose). The solvents vaporize, leaving only the recrystallized matrix, but now with analyte molecules embedded into MALDI crystals. The matrix and the analyte are said to be co-crystallized. Co-crystallization is a key issue in selecting a proper matrix to obtain a good quality mass spectrum of the analyte of interest.
In analysis of biological systems, Inorganic salts, which are also part of protein extracts, interfere with the ionization process. The salts can be removed by solid phase extraction or by washing the dried-droplet MALDI spots with cold water. Both methods can also remove other substances from the sample. The matrix-protein mixture is not homogenous because the polarity difference leads to a separation of the two substances during co-crystallization. The spot diameter of the target is much larger than that of the laser, which makes it necessary to make many laser shots at different places of the target, to get the statistical average of the substance concentration within the target spot.
The matrix can be used to tune the instrument to ionize the sample in different ways. As mentioned above, acid-base like reactions are often utilized to ionize the sample, however, molecules with conjugated pi systems, such as naphthalene like compounds, can also serve as an electron acceptor and thus a matrix for MALDI/TOF. This is particularly useful in studying molecules that also possess conjugated pi systems. The most widely used application for these matrices is studying porphyrin-like compounds such as chlorophyll. These matrices have been shown to have better ionization patterns that do not result in odd fragmentation patterns or complete loss of side chains. It has also been suggested that conjugated porphryin like molecules can serve as a matrix and cleave themselves eliminating the need for a separate matrix compound.
There are several variations of the MALDI technology and comparable instruments are today produced for very different purposes. From more academic and analytical, to more industrial and high throughput. The MS field has expanded into requiring ultrahigh resolution mass spectrometry such as the FT-ICR instruments as well as more high-throughput instruments. As many MALDI MS instruments can be bought with an interchangeable ionization source (Electrospray ionization, MALDI, Atmospheric pressure ionization, etc.) the technologies often overlap and many times any soft ionization method could potentially be used. For more variations of soft ionization methods go to Soft laser desorption or Ion source.
Infrared laser wavelengths used for infrared MALDI include the 2.94 μm Er:YAG laser, mid-IR optical parametric oscillator, and 10.6 μm carbon dioxide laser. Although not as common, infrared lasers are used due to their softer mode of ionization. IR-MALDI also has the advantage of greater material removal (useful for biological samples), less low-mass interferences, and compatibility with other matrix-free laser desorption mass spectrometry methods.
Time of flight
The type of a mass spectrometer most widely used with MALDI is the TOF (time-of-flight mass spectrometer), mainly due to its large mass range. The TOF measurement procedure is also ideally suited to the MALDI ionization process since the pulsed laser takes individual 'shots' rather than working in continuous operation. MALDI-TOF instruments are often equipped with a reflectron (an "ion mirror") that reflects ions using an electric field. This increases the ion flight path, thereby increasing time of flight between ions of different m/z and increasing resolution. Modern commercial reflectron TOF instruments reach a resolving power m/Δm of 50,000 FWHM (full-width half-maximum, Δm defined as the peak width at 50% of peak height) or more.
Atmospheric pressure (AP) matrix-assisted laser desorption/ionization (MALDI) is an ionization technique (ion source) that in contrast to vacuum MALDI operates at normal atmospheric environment. The main difference between vacuum MALDI and AP-MALDI is the pressure in which the ions are created. In vacuum MALDI, ions are typically produced at 10 mTorr or less while in AP-MALDI ions are formed in atmospheric pressure. In the past the main disadvantage of the AP-MALDI technique compared to the conventional vacuum MALDI has been its limited sensitivity; however, ions can be transferred into the mass spectrometer with high efficiency and attomole detection limits have been reported. AP-MALDI is used in mass spectrometry (MS) in a variety of applications ranging from proteomics to drug discovery. Popular topics that are addressed by AP-MALDI mass spectrometry include: proteomics; mass analysis of DNA, RNA, PNA, lipids, oligosaccharides, phosphopeptides, bacteria, small molecules and synthetic polymers, similar applications as available also for vacuum MALDI instruments. The AP-MALDI ion source is easily coupled to an ion trap mass spectrometer or any other MS system equipped with ESI (electrospray ionization) or nanoESI source.
In aerosol mass spectrometry, one of the ionization techniques consists in firing a laser to individual droplets. These systems are called single particle mass spectrometers (SPMS). The sample may optionally be mixed with a MALDI matrix prior to aerosolization.
The laser is fired at the matrix crystals in the dried-droplet spot. The matrix absorbs the laser energy and it is thought that primarily the matrix is desorbed and ionized (by addition of a proton) by this event. The hot plume produced during ablation contains many species: neutral and ionized matrix molecules, protonated and deprotonated matrix molecules, matrix clusters and nanodroplets. Ablated species may participate in the ionization of analyte, though the mechanism of MALDI is still debated. The matrix is then thought to transfer protons to the analyte molecules (e.g., protein molecules), thus charging the analyte. An ion observed after this process will consist of the initial neutral molecule [M] with ions added or removed. This is called a quasimolecular ion, for example [M+H]+ in the case of an added proton, [M+Na]+ in the case of an added sodium ion, or [M-H]− in the case of a removed proton. MALDI is capable of creating singly charged ions or multiply charged ions ([M+nH]n+) depending on the nature of the matrix, the laser intensity, and/or the voltage used. Note that these are all even-electron species. Ion signals of radical cations (photoionized molecules) can be observed, e.g., in the case of matrix molecules and other organic molecules.
The gas phase proton transfer model, implemented as the coupled physical and chemical dynamics (CPCD) model, of UV laser MALDI postulates primary and secondary processes leading to ionization. Primary processes involve initial charge separation through absorption of photons by the matrix and pooling of the energy to form matrix ion pairs. Primary ion formation occurs through absorption of a UV photon to create excited state molecules by
- S0 + hν → S1
- S1 + S1 → S0 + Sn
- S1 + Sn → M+ + M−
where S0 is the ground electronic state, S1 the first electronic excited state, and Sn is a higher electronic excited state. The product ions can be proton transfer or electron transfer ion pairs, indicated by M+ and M− above. Secondary processes involve ion-molecule reactions to form analyte ions.
The lucky survivor model (cluster ionization mechanism) postulates that analyte molecules are incorporated in the matrix maintaining the charge state from solution. Ion formation occurs through charge separation upon fragmentation of laser ablated clusters. Ions that are not neutralized by recombination with photoelectrons or counter ions are the so-called lucky survivors.
The thermal model postulates that the high temperature facilitates the proton transfer between matrix and analyte in melted matrix liquid. Ion-to-neutral ratio is an important parameter to justify the theoretical model, and the mistaken citation of ion-to-neutral ratio could result in an erroneous determination of the ionization mechanism. The model quantitatively predicts the increase in total ion intensity as a function of the concentration and proton affinity of the analytes, and the ion-to-neutral ratio as a function of the laser fluences. This model also suggests that metal ion adducts (e.g., [M+Na]+ or [M+K]+) are mainly generated from the thermally induced dissolution of salt.
The matrix-assisted ionization (MAI) method uses matrix preparation similar to MALDI but does not require laser ablation to produce analyte ions of volatile or nonvolatile compounds. Simply exposing the matrix with analyte to the vacuum of the mass spectrometer creates ions with nearly identical charge states to electrospray ionization. It is suggested that there are likely mechanistic commonality between this process and MALDI.
Ion yield is typically estimated to range from 10−4 to 10−7, with some experiments hinting to even lower yields of 10−9. The issue of low ion yields had been addressed, already shortly after introduction of MALDI by various attempts, including post-ionization utilizing a second laser. Most of these attempts showed only limited success, with low signal increases. This might be attributed to the fact, that axial time-of-flight instruments were used, which operate at pressures in the source region of 10−5 to 10−6, which results in rapid plume expansion with particle velocities of up to 1000 m/s. In 2015 successful laser post-ionization was reported, using a modified MALDI source operated at an elevated pressure of ~3 mbar coupled to an orthogonal time-of-flight mass analyzer, and employing a wavelength-tunable post-ionization laser, operated at wavelength from 260 nm to 280 nm, below the two-photon ionization threshold of the matrices used, which elevated ion yields of several lipids and small molecules by up to three orders of magnitude. This approach, called MALDI-2, due to the second laser, and the second MALDI-like ionization process, was afterwards adopted for other mass spectrometers, all equipped with sources operating in the low mbar range.
In proteomics, MALDI is used for the rapid identification of proteins isolated by using gel electrophoresis: SDS-PAGE, size exclusion chromatography, affinity chromatography, strong/weak ion exchange, isotope coded protein labelling (ICPL), and two-dimensional gel electrophoresis. Peptide mass fingerprinting is the most popular analytical application of MALDI-TOF mass spectrometers. MALDI TOF/TOF mass spectrometers are used to reveal amino acid sequence of peptides using post-source decay or high energy collision-induced dissociation (further use see mass spectrometry).
MALDI-TOF have been used to characterise post-translational modifications. For example, it has been widely applied to study protein methylation and demethylation. However, careful must be taken when studying post-translational modifications by MALDI-TOF. For example, it has been reported that loss of sialic acid has been identified in papers when DHB has been used as a matrix for MALDI MS analysis of glycosylated peptides. Using sinapinic acid, 4-HCCA and DHB as matrices, S. Martin studied loss of sialic acid in glycosylated peptides by metastable decay in MALDI/TOF in linear mode and reflector mode. A group at Shimadzu Corporation derivatized the sialic acid by an amidation reaction as a way to improve detection sensitivity and also demonstrated that ionic liquid matrix reduces a loss of sialic acid during MALDI/TOF MS analysis of sialylated oligosaccharides. THAP, DHAP, and a mixture of 2-aza-2-thiothymine and phenylhydrazine have been identified as matrices that could be used to minimize loss of sialic acid during MALDI MS analysis of glycosylated peptides. It has been reported that a reduction in loss of some post-translational modifications can be accomplished if IR MALDI is used instead of UV MALDI
Besides proteins, MALDI-TOF has also been applied to study lipids. For example, it has been applied to study the catalytic reactions of phospholipases. In addition to lipids, oligonucleotides have also been characterised by MALDI-TOF. For example, in molecular biology, a mixture of 5-methoxysalicylic acid and spermine can be used as a matrix for oligonucleotides analysis in MALDI mass spectrometry, for instance after oligonucleotide synthesis.
Some synthetic macromolecules, such as catenanes and rotaxanes, dendrimers and hyperbranched polymers, and other assemblies, have molecular weights extending into the thousands or tens of thousands, where most ionization techniques have difficulty producing molecular ions. MALDI is a simple and fast analytical method that can allow chemists to rapidly analyze the results of such syntheses and verify their results.
In polymer chemistry MALDI can be used to determine the molar mass distribution. Polymers with polydispersity greater than 1.2 are difficult to characterize with MALDI due to the signal intensity discrimination against higher mass oligomers.
MALDI-TOF spectra are often used for the identification of microorganisms such as bacteria or fungi. A portion of a colony of the microbe in question is placed onto the sample target and overlaid with matrix. The mass spectra of expressed proteins generated are analyzed by dedicated software and compared with stored profiles for species determination in what is known as biotyping. It offers benefits to other immunological or biochemical procedures and has become a common method for species identification in clinical microbiological laboratories. Benefits of high resolution MALDI-MS performed on a Fourier transform ion cyclotron resonance mass spectrometry (also known as FT-MS) have been demonstrated for typing and subtyping viruses though single ion detection known as proteotyping, with a particular focus on influenza viruses.
One main advantage over other microbiological identification methods is its ability to rapidly and reliably identify, at low cost, a wide variety of microorganisms directly from the selective medium used to isolate them. The absence of the need to purify the suspect or "presumptive" colony allows for a much faster turn-around times. For example, it has been demonstrated that MALDI-TOF can be used to detect bacteria directly from blood cultures.
Another advantage is the potential to predict antibiotic susceptibility of bacteria. A single mass spectral peak can predict methicillin resistance of Staphylococcus aureus MALDI can also detect carbapenemase of carbapenem-resistant enterobacteriaceae, including Acinetobacter baumannii and Klebsiella pneumoniae. However, most proteins that mediate antibiotic resistance are larger than MALDI-TOF's 2000–20,000 Da range for protein peak interpretation and only occasionally, as in the 2011 KPC outbreak at the NIH, a correlation between a peak and resistance conferring protein can be made.
MALDI-TOF spectra have been used for the detection and identification of various parasites such as trypanosomatids, Leishmania  and Plasmodium. In addition to these unicellular parasites, MALDI/TOF can be used for the identification of parasitic insects such as lice  or cercariae, the free-swimming stage of trematodes.
MALDI-TOF spectra are often utilized in tandem with other analysis and spectroscopy techniques in the diagnosis of diseases. MALDI/TOF is a diagnostic tool with much potential because it allows for the rapid identification of proteins and changes to proteins without the cost or computing power of sequencing nor the skill or time needed to solve a crystal structure in X-ray crystallography.
One example of this is necrotizing enterocolitis (NEC), which is a devastating disease that affects the bowels of premature infants. The symptoms of NEC are very similar to those of sepsis, and many infants die awaiting diagnosis and treatment. MALDI/TOF was used to identify bacteria present in the fecal matter of NEC positive infants. This study focused on characterization of the fecal microbiota associated with NEC and did not address the mechanism of disease. There is hope that a similar technique could be used as a quick, diagnostic tool that would not require sequencing.
Another example of the diagnostic power of MALDI/TOF is in the area of cancer. Pancreatic cancer remains one of the most deadly and difficult to diagnose cancers. Impaired cellular signaling due to mutations in membrane proteins has been long suspected to contribute to pancreatic cancer. MALDI/TOF has been used to identify a membrane protein associated with pancreatic cancer and at one point may even serve as an early detection technique.[non-primary source needed]
MALDI/TOF can also potentially be used to dictate treatment as well as diagnosis. MALDI/TOF serves as a method for determining the drug resistance of bacteria, especially to β-lactams (Penicillin family). The MALDI/TOF detects the presence of carbapenemases, which indicates drug resistance to standard antibiotics. It is predicted that this could serve as a method for identifying a bacterium as drug resistant in as little as three hours. This technique could help physicians decide whether to prescribe more aggressive antibiotics initially.
Detection of protein complexes
- Hillenkamp, Franz; Karas, Michael; Beavis, Ronald C.; Chait, Brian T. (1991). "Matrix-assisted laser desorption/ionization mass spectrometry of biopolymers". Analytical Chemistry. 63 (24): 1193A–1203A. doi:10.1021/ac00024a002. ISSN 0003-2700. PMID 1789447.
- Karas, Michael; Krüger, Ralf (2003). "Ion Formation in MALDI: The Cluster Ionization Mechanism". Chemical Reviews. 103 (2): 427–440. doi:10.1021/cr010376a. ISSN 0009-2665. PMID 12580637.
- Karas, M.; Bachmann, D.; Hillenkamp, F. (1985). "Influence of the Wavelength in High-Irradiance Ultraviolet Laser Desorption Mass Spectrometry of Organic Molecules". Analytical Chemistry. 57 (14): 2935–9. doi:10.1021/ac00291a042.
- Karas, M.; Bachmann, D.; Bahr, U.; Hillenkamp, F. (1987). "Matrix-Assisted Ultraviolet Laser Desorption of Non-Volatile Compounds". International Journal of Mass Spectrometry and Ion Processes. 78: 53–68. Bibcode:1987IJMSI..78...53K. doi:10.1016/0168-1176(87)87041-6.
- Tanaka, K.; Waki, H.; Ido, Y.; Akita, S.; Yoshida, Y.; Yoshida, T.; Matsuo, T. (1988). "Protein and Polymer Analyses up to m/z 100 000 by Laser Ionization Time-of flight Mass Spectrometry". Rapid Communications in Mass Spectrometry. 2 (20): 151–3. Bibcode:1988RCMS....2..151T. doi:10.1002/rcm.1290020802.
- Markides, K.; Gräslund, A. (9 October 2002). "Advanced information on the Nobel Prize in Chemistry 2002" (PDF). The Royal Swedish Academy of Sciences. pp. 1–13. Retrieved 2013-08-28.
- Karas, M.; Hillenkamp, F. (1988). "Laser desorption ionization of proteins with molecular masses exceeding 10,000 daltons". Analytical Chemistry. 60 (20): 2299–301. doi:10.1021/ac00171a028. PMID 3239801.
- Beavis, R. C.; Chait, B. T.; Standing, K. G. (1989). "Matrix-assisted laser-desorption mass spectrometry using 355 nm radiation". Rapid Communications in Mass Spectrometry. 3 (12): 436–9. Bibcode:1989RCMS....3..436B. doi:10.1002/rcm.1290031208. PMID 2520224.
- Karas, M.; Bahr, U. (1990). "Laser Desorption Ionization Mass Spectrometry of Large Biomolecules". Trends in Analytical Chemistry. 9 (10): 321–5. doi:10.1016/0165-9936(90)85065-F.
- Strupat, K.; Karas, M.; Hillenkamp, F. (1991). "2,5-Dihidroxybenzoic acid: A new matrix for laser desorption—ionization mass spectrometry". International Journal of Mass Spectrometry and Ion Processes. 72 (111): 89–102. Bibcode:1991IJMSI.111...89S. doi:10.1016/0168-1176(91)85050-V.
- Beavis, R. C.; Chait, B. T.; Fales, H. M. (1989). "Cinnamic acid derivatives as matrices for ultraviolet laser desorption mass spectrometry of proteins". Rapid Communications in Mass Spectrometry. 3 (12): 432–5. Bibcode:1989RCMS....3..432B. doi:10.1002/rcm.1290031207. PMID 2520223.
- Beavis, R. C.; Chaudhary, T.; Chait, B. T. (1992). "α-Cyano-4-hydroxycinnamic acid as a matrix for matrix-assisted laser desorption mass spectrometry". Organic Mass Spectrometry. 27 (2): 156–8. doi:10.1002/oms.1210270217.
- Tang, K.; Taranenko, N. I.; Allman, S. L.; Cháng, L. Y.; Chen, C. H.; Lubman, D. M. (1994). "Detection of 500-nucleotide DNA by laser desorption mass spectrometry". Rapid Communications in Mass Spectrometry. 8 (9): 727–30. Bibcode:1994RCMS....8..727T. doi:10.1002/rcm.1290080913. PMID 7949335.
- Wu, K. J.; Steding, A.; Becker, C. H. (1993). "Matrix-assisted laser desorption time-of-flight mass spectrometry of oligonucleotides using 3-hydroxypicolinic acid as an ultraviolet-sensitive matrix". Rapid Communications in Mass Spectrometry. 7 (2): 142–6. Bibcode:1993RCMS....7..142W. doi:10.1002/rcm.1290070206. PMID 8457722.
- Korfmacher, Walter A. (2009). Using Mass Spectrometry for Drug Metabolism Studies. CRC Press. p. 342. ISBN 9781420092219.
- Fitzgerald, M. C.; Parr, G. R.; Smith, L. M. (1993). "Basic matrixes for the matrix-assisted laser desorption/ionization mass spectrometry of proteins and oligonucleotides". Analytical Chemistry. 65 (22): 3204–11. doi:10.1021/ac00070a007. PMID 8291672.
- Zenobi, R.; Knochenmuss, R. (1998). "Ion formation in MALDI mass spectrometry". Mass Spectrometry Reviews. 17 (5): 337–366. Bibcode:1998MSRv...17..337Z. doi:10.1002/(SICI)1098-2787(1998)17:5<337::AID-MAS2>3.0.CO;2-S.
- Xu, Yingda; Bruening, Merlin L.; Watson, J. Throck (2003). "Non-specific, on-probe cleanup methods for MALDI-MS samples". Mass Spectrometry Reviews. 22 (6): 429–440. Bibcode:2003MSRv...22..429X. doi:10.1002/mas.10064. ISSN 0277-7037. PMID 14528495.
- Nazim Boutaghou, M.; Cole, R. B. (2012). "9,10-Diphenylanthracene as a matrix for MALDI-MS electron transfer secondary reactions". J. Mass Spectrom. 47 (8): 995–1003. Bibcode:2012JMSp...47..995N. doi:10.1002/jms.3027. PMID 22899508.
- Suzuki, T.; Midonoya, H.; Shioi, Y. (2009). "Analysis of chlorophylls and their derivatives by matrix-assisted laser desorption/ionization–time-of-flight mass spectrometry". Anal. Biochem. 390 (1): 57–62. doi:10.1016/j.ab.2009.04.005. hdl:10297/4147. PMID 19364490.
- Wei, J.; Li, H.; Barrow, M. P.; O'Connor, P. B. (2013). "Structural characterization of chlorophyll-a by high resolution tandem mass spectrometry". J. Am. Soc. Mass Spectrom. 24 (5): 753–760. Bibcode:2013JASMS..24..753W. doi:10.1007/s13361-013-0577-1. PMID 23504642. S2CID 43158125.
- Srinivasan, N.; Haney, C. A.; Lindsey, J. S.; Zhang, W.; Chait, B. T. (1999). "Investigation of MALDI-TOF mass spectrometry of diverse synthetic metalloporphyrins, phthalocyanines and multiporphyrin arrays". J. Porphyrins Phthalocyanines. 3 (4): 283–291. doi:10.1002/(SICI)1099-1409(199904)3:4<283::AID-JPP132>3.0.CO;2-F.
- "Talking About a Revolution: FT-ICR Mass Spectrometry Offers High Resolution and Mass Accuracy for Pr".
- Schmitt-Kopplin, P; Hertkorn, N (2007). "Ultrahigh resolution mass spectrometry". Anal Bioanal Chem. 389 (5): 1309–1310. doi:10.1007/s00216-007-1589-0. PMC 2129108.
- Ghyselinck, Jonas; Van Hoorde, Koenraad; Hoste, Bart; Heylen, Kim; De Vos, Paul (September 2011). "Evaluation of MALDI-TOF MS as a tool for high-throughput dereplication". Journal of Microbiological Methods. 86 (3): 327–336. doi:10.1016/j.mimet.2011.06.004. PMID 21699925.
- Dreisewerd, Klaus (2014). "Recent methodological advances in MALDI mass spectrometry". Analytical and Bioanalytical Chemistry. 406 (9–10): 2261–2278. doi:10.1007/s00216-014-7646-6. ISSN 1618-2642. PMID 24652146. S2CID 5467069.
- Murray, K (2006). "Chapter 9 Desorption by Photons: Laser Desorption and Matridx-Assisted Laser Desorption Ionization (MALDI) – Infrared Matrix-Assisted Laser Desorption Ionization". In Gross, Michael L.; Caprioli, Richard M. (eds.). The Encyclopedia of Mass Spectrometry. Volume 6: Ionization Methods. Elsevier Science. ISBN 9780080438016. Retrieved 2017-04-05.
- Xian, Feng; Hendrickson, Christopher L.; Marshall, Alan G. (2012-01-17). "High Resolution Mass Spectrometry". Analytical Chemistry. 84 (2): 708–719. doi:10.1021/ac203191t. ISSN 0003-2700. PMID 22263633.
- Ruotolo, B. T.; Gillig, K. J.; Woods, A. S.; Egan, T. F.; Ugarov, M. V.; Schultz, J. A.; Russell, D. H. (2004). "Analysis of Phosphorylated Peptides by Ion Mobility-Mass Spectrometry". Analytical Chemistry. 76 (22): 6727–6733. doi:10.1021/ac0498009. PMID 15538797.(subscription required)
- Ruotolo, B. T.; Verbeck, G. F.; Thomson, L. M.; Woods, A. S.; Gillig, K. J.; Russell, D. H. (2002). "Distinguishing between Phosphorylated and Nonphosphorylated Peptides with Ion Mobility−Mass Spectrometry". Journal of Proteome Research. 1 (4): 303–306. doi:10.1021/pr025516r. PMID 12645885.(subscription required)
- Pasa-Tolic, L.; Huang, Y.; Guan, S.; Kim, H. S.; Marshall, A. G. (1995). "Ultrahigh-resolution matrix-assisted laser desorption/ionization Fourier transform ion cyclotron resonance mass spectra of peptides". Journal of Mass Spectrometry. 30 (6): 825–833. Bibcode:1995JMSp...30..825P. doi:10.1002/jms.1190300607.(subscription required)
- Laiko, V. V.; Baldwin, M. A.; Burlingame, A. L. (2000). "Atmospheric pressure matrix-assisted laser desorption/ionization mass spectrometry". Analytical Chemistry. 72 (4): 652–7. doi:10.1021/ac990998k. PMID 10701247.
- Strupat, K.; Scheibner, O.; Arrey, T.; Bromirski, M. (2011). "Biological Applications of AP MALDI with Thermo Scientific Exactive Orbitrap MS" (PDF). 443-539-1710. Thermo Scientific. Retrieved 17 June 2011.
- Laiko, V. V.; Moyer, S. C.; Cotter, R. J. (2000). "Atmospheric pressure MALDI/ion trap mass spectrometry". Analytical Chemistry. 72 (21): 5239–43. doi:10.1021/ac000530d. PMID 11080870.
- Li, Yong Jie; Sun, Yele; Zhang, Qi; Li, Xue; Li, Mei; Zhou, Zhen; Chan, Chak K. (June 2017). "Real-time chemical characterization of atmospheric particulate matter in China: A review". Atmospheric Environment. 158: 270–304. Bibcode:2017AtmEn.158..270L. doi:10.1016/j.atmosenv.2017.02.027.
- Knochenmuss, R (2006). "Ion formation mechanisms in UV-MALDI". Analyst. 131 (9): 966–986. Bibcode:2006Ana...131..966K. doi:10.1039/b605646f. PMID 17047796.
- Knochenmuss, Richard (2016). "The Coupled Chemical and Physical Dynamics Model of MALDI". Annual Review of Analytical Chemistry. 9 (1): 365–385. Bibcode:2016ARAC....9..365K. doi:10.1146/annurev-anchem-071015-041750. ISSN 1936-1327. PMID 27070182.
- Knochenmuss, Richard (2006). "Ion formation mechanisms in UV-MALDI". The Analyst. 131 (9): 966–86. Bibcode:2006Ana...131..966K. doi:10.1039/b605646f. ISSN 0003-2654. PMID 17047796.
- Karas, Michael; Glückmann, Matthias; Schäfer, Jürgen (2000). "Ionization in matrix-assisted laser desorption/ionization: singly charged molecular ions are the lucky survivors". Journal of Mass Spectrometry. 35 (1): 1–12. Bibcode:2000JMSp...35....1K. doi:10.1002/(SICI)1096-9888(200001)35:1<1::AID-JMS904>3.0.CO;2-0. ISSN 1076-5174. PMID 10633229.
- McEwen, Charles N.; Larsen, Barbara S. (2015). "Fifty years of desorption ionization of nonvolatile compounds". International Journal of Mass Spectrometry. 377: 515–531. Bibcode:2015IJMSp.377..515M. doi:10.1016/j.ijms.2014.07.018. ISSN 1387-3806.
- Lu, I.-Chung; Lee, Chuping; Lee, Yuan-Tseh; Ni, Chi-Kung (2015). "Ionization Mechanism of Matrix-Assisted Laser Desorption/Ionization". Annual Review of Analytical Chemistry. 8: 21–39. Bibcode:2015ARAC....8...21L. doi:10.1146/annurev-anchem-071114-040315. PMID 26132345.
- Tsai, Ming-Tsang; Lee, Sheng; Lu, I-Chung; Chu, Kuan Yu; Liang, Chi-Wei; Lee, Chih Hao; Lee, Yuan T.; Ni, Chi-Kung (2013-05-15). "Ion-to-neutral ratio of 2,5-dihydroxybenzoic acid in matrix-assisted laser desorption/ionization". Rapid Communications in Mass Spectrometry. 27 (9): 955–963. Bibcode:2013RCMS...27..955T. doi:10.1002/rcm.6534. ISSN 1097-0231. PMID 23592197.
- Lu, I-Chung; Lee, Chuping; Chen, Hui-Yuan; Lin, Hou-Yu; Hung, Sheng-Wei; Dyakov, Yuri A.; Hsu, Kuo-Tung; Liao, Chih-Yu; Lee, Yin-Yu (2014). "Ion Intensity and Thermal Proton Transfer in Ultraviolet Matrix-Assisted Laser Desorption/Ionization". The Journal of Physical Chemistry B. 118 (15): 4132–4139. doi:10.1021/jp5008076. ISSN 1520-6106. PMID 24707818.
- Lu, I.-Chung; Chu, Kuan Yu; Lin, Chih-Yuan; Wu, Shang-Yun; Dyakov, Yuri A.; Chen, Jien-Lian; Gray-Weale, Angus; Lee, Yuan-Tseh; Ni, Chi-Kung (2015). "Ion-to-Neutral Ratios and Thermal Proton Transfer in Matrix-Assisted Laser Desorption/Ionization". Journal of the American Society for Mass Spectrometry. 26 (7): 1242–1251. Bibcode:2015JASMS..26.1242L. doi:10.1007/s13361-015-1112-3. ISSN 1044-0305. PMID 25851654. S2CID 24085468.
- Lee, Chuping; Lu, I.-Chung; Hsu, Hsu Chen; Lin, Hou-Yu; Liang, Sheng-Ping; Lee, Yuan-Tseh; Ni, Chi-Kung (2016). "Formation of Metal-Related Ions in Matrix-Assisted Laser Desorption Ionization". Journal of the American Society for Mass Spectrometry. 27 (9): 1491–1498. Bibcode:2016JASMS..27.1491L. doi:10.1007/s13361-016-1424-y. ISSN 1044-0305. PMID 27306427. S2CID 2139197.
- Trimpin, Sarah (2015). ""Magic" Ionization Mass Spectrometry". Journal of the American Society for Mass Spectrometry. 27 (1): 4–21. Bibcode:2016JASMS..27....4T. doi:10.1007/s13361-015-1253-4. ISSN 1044-0305. PMC 4686549. PMID 26486514.
- Trimpin, S, Inutan, ED (2013). "Matrix Assisted Ionization in Vacuum, a Sensitive and Widely Applicable Ionization Method for Mass Spectrometry". J. Am. Soc. Mass Spectrom. 24 (5): 722–732. Bibcode:2013JASMS..24..722T. doi:10.1007/s13361-012-0571-z. PMID 23526166. S2CID 29978978.CS1 maint: multiple names: authors list (link)
- Lu, I-Chung; Chu, Kuan Yu; Lin, Chih-Yuan; Wu, Shang-Yun; Dyakov, Yuri A.; Chen, Jien-Lian; Gray-Weale, Angus; Lee, Yuan-Tseh; Ni, Chi-Kung (July 2015). "Ion-to-Neutral Ratios and Thermal Proton Transfer in Matrix-Assisted Laser Desorption/Ionization". Journal of the American Society for Mass Spectrometry. 26 (7): 1242–1251. Bibcode:2015JASMS..26.1242L. doi:10.1007/s13361-015-1112-3. ISSN 1044-0305. PMID 25851654. S2CID 24085468.
- Robinson, Kenneth N.; Steven, Rory T.; Race, Alan M.; Bunch, Josephine (July 2019). "The Influence of MS Imaging Parameters on UV-MALDI Desorption and Ion Yield". Journal of the American Society for Mass Spectrometry. 30 (7): 1284–1293. Bibcode:2019JASMS..30.1284R. doi:10.1007/s13361-019-02193-8. ISSN 1044-0305. PMID 30949969. S2CID 96435099.
- Spengler, B.; Bahr, U.; Karas, M.; Hillenkamp, F. (January 1988). "Postionization of Laser-Desorbed Organic and Inorganic Compounds in a Time of Flight Mass Spectrometer". Instrumentation Science & Technology. 17 (1–2): 173–193. Bibcode:1988IS&T...17..173S. doi:10.1080/10739148808543672. ISSN 1073-9149.
- Dreisewerd, Klaus (February 2003). "The Desorption Process in MALDI". Chemical Reviews. 103 (2): 395–426. doi:10.1021/cr010375i. ISSN 0009-2665. PMID 12580636.
- Soltwisch, J.; Kettling, H.; Vens-Cappell, S.; Wiegelmann, M.; Muthing, J.; Dreisewerd, K. (2015-04-10). "Mass spectrometry imaging with laser-induced postionization". Science. 348 (6231): 211–215. Bibcode:2015Sci...348..211S. doi:10.1126/science.aaa1051. ISSN 0036-8075. PMID 25745064. S2CID 206632790.
- Soltwisch, Jens; Heijs, Bram; Koch, Annika; Vens-Cappell, Simeon; Höhndorf, Jens; Dreisewerd, Klaus (2020-07-07). "MALDI-2 on a Trapped Ion Mobility Quadrupole Time-of-Flight Instrument for Rapid Mass Spectrometry Imaging and Ion Mobility Separation of Complex Lipid Profiles". Analytical Chemistry. 92 (13): 8697–8703. doi:10.1021/acs.analchem.0c01747. ISSN 0003-2700. PMID 32449347.
- Ellis, S. R.; Soltwisch, J.; Paine, M. R. L.; Dreisewerd, K.; Heeren, R. M. A. (2017). "Laser post-ionisation combined with a high resolving power orbitrap mass spectrometer for enhanced MALDI-MS imaging of lipids". Chemical Communications. 53 (53): 7246–7249. doi:10.1039/C7CC02325A. ISSN 1359-7345. PMID 28573274.
- Stacie L. Richardson, Pahul Hanjra, Gang Zhang, Brianna D. Mackie, Darrell L. Peterson, Rong Huang (2015). "A direct, ratiometric, and quantitative MALDI–MS assay for protein methyltransferases and acetyltransferases". Anal. Biochem. 478: 59–64. doi:10.1016/j.ab.2015.03.007. PMC 4855292. PMID 25778392.CS1 maint: multiple names: authors list (link)
- Karine Guitot, Thierry Drujon, Fabienne Burlina, Sandrine Sagan, Sandra Beaupierre, Olivier Pamlard, Robert H Dodd, Catherine Guillou, Gérard Bolbach, Emmanuelle Sachon, Dominique Guianvarc'h (2017). "A direct label-free MALDI-TOF mass spectrometry based assay for the characterization of inhibitors of protein lysine methyltransferases". Anal. Bioanal. Chem. 409 (15): 3767–3777. doi:10.1007/s00216-017-0319-5. PMID 28389916.CS1 maint: multiple names: authors list (link)
- Huberty, M. C.; Vath, J. E.; Yu, W.; Martin, S. A. (1993). "Site-specific carbohydrate identification in recombinant proteins using MALD-TOF MS". Analytical Chemistry. 65 (20): 2791–2800. doi:10.1021/ac00068a015. PMID 8250262.
- Sekiya, S.; Wada, Y.; Tanaka, K. (2005). "Derivatization for Stabilizing Sialic Acids in MALDI-MS". Analytical Chemistry. 77 (15): 4962–4968. doi:10.1021/ac050287o. PMID 16053310.
- Fukuyama, Y.; Nakaya, S.; Yamazaki, Y.; Tanaka, K. (2008). "Ionic Liquid Matrixes Optimized for MALDI-MS of Sulfated/Sialylated/Neutral Oligosaccharides and Glycopeptides". Analytical Chemistry. 80 (6): 2171–2179. doi:10.1021/ac7021986. PMID 18275166.
- Papac, D. I.; Wong, A.; Jones, A. J. S. (1996). "Analysis of Acidic Oligosaccharides and Glycopeptides by Matrix-Assisted Laser Desorption/Ionization Time-of-Flight Mass Spectrometry". Analytical Chemistry. 68 (18): 3215–3223. doi:10.1021/ac960324z. PMID 8797382.
- Harvey, D. J. (1999). "Matrix-assisted laser desorption/ionization mass spectrometry of carbohydrates". Mass Spectrometry Reviews. 18 (6): 349–451. Bibcode:1999MSRv...18..349H. CiteSeerX 10.1.1.1012.1931. doi:10.1002/(SICI)1098-2787(1999)18:6<349::AID-MAS1>3.0.CO;2-H. PMID 10639030.
- Lattova, E.; Chen, V. C.; Varma, S.; Bezabeh, T.; Perreault, H. (2007). "Matrix-assisted laser desorption/ionization on-target method for the investigation of oligosaccharides and glycosylation sites in glycopeptides and glycoproteins". Rapid Communications in Mass Spectrometry. 21 (10): 1644–1650. Bibcode:2007RCMS...21.1644L. doi:10.1002/rcm.3007. PMID 17465012.
- Tajiri, M.; Takeuchi, T.; Wada, Y. (2009). "Distinct Features of Matrix-Assisted 6 μm Infrared Laser Desorption/Ionization Mass Spectrometry in Biomolecular Analysis". Analytical Chemistry. 81 (16): 6750–6755. doi:10.1021/ac900695q. PMID 19627133.
- T. W. Jaskolla, K. Onischke, J. Schiller (2014). "2,5‐Dihydroxybenzoic acid salts for matrix‐assisted laser desorption/ionization time‐of‐flight mass spectrometric lipid analysis: Simplified spectra interpretation and insights into gas‐phase fragmentation". Rapid Commun. Mass Spectrom. 28: 1353–1363. doi:10.1002/rcm.6910.CS1 maint: multiple names: authors list (link)
- J. Lee, Y.-K. Kim, D.-H. Min (2010). "Laser Desorption/Ionization Mass Spectrometric Assay for Phospholipase Activity Based on Graphene Oxide/Carbon Nanotube Double-Layer Films". J. Am. Chem. Soc. 132: 14714–14717. doi:10.1021/ja106276j.CS1 maint: multiple names: authors list (link)
- Nabangshu Sharma, Ries J. Langley, Chatchakorn Eurtivong, Euphemia Leung, Ryan Joseph Dixon, Emily K. Paulin, Shaun W. P. Rees, Lisa I Pilkington, David Barker, Johannes Reynisson, Ivanhoe K. H. Leung (2021). "An optimised MALDI-TOF assay for phosphatidylcholine-specific phospholipase C". Analytical Methods. doi:10.1039/D0AY02208J.CS1 maint: multiple names: authors list (link)
- Distler, A. M.; Allison, J. (2001). "5-Methoxysalicylic acid and spermine: A new matrix for the matrix-assisted laser desorption/ionization mass spectrometry analysis of oligonucleotides". Journal of the American Society for Mass Spectrometry. 12 (4): 456–62. doi:10.1016/S1044-0305(01)00212-4. PMID 11322192. S2CID 18280663.
- Schrepp, W.; Pasch, H. (2003). MALDI-TOF Mass Spectrometry of Synthetic Polymers. Springer-Verlag. ISBN 978-3-540-44259-2.
- Nielen, M. W. F.; Malucha, S. (1997). "Characterization of polydisperse synthetic polymers by size-exclusion chromatography/matrix-assisted laser desorption/ionization time-of-flight mass spectrometry". Rapid Communications in Mass Spectrometry. 11 (11): 1194–1204. Bibcode:1997RCMS...11.1194N. doi:10.1002/(SICI)1097-0231(199707)11:11<1194::AID-RCM935>3.0.CO;2-L.
- Wu, K. J.; Odom, R. W. (1998). "Characterizing synthetic polymers by MALDI MS". Analytical Chemistry. 70 (13): 456A–461A. doi:10.1021/ac981910q. PMID 9666717.
- Schriemer, D. C.; Li, L. (1997). "Mass Discrimination in the Analysis of Polydisperse Polymers by MALDI Time-of-Flight Mass Spectrometry. 2. Instrumental Issues". Analytical Chemistry. 69 (20): 4176–4183. doi:10.1021/ac9707794.
- Schaiberger, Audrey M.; Moss, Jason A. (2008). "Optimized sample preparation for MALDI mass spectrometry analysis of protected synthetic peptides". Journal of the American Society for Mass Spectrometry. 19 (4): 614–619. doi:10.1016/j.jasms.2008.01.010. ISSN 1044-0305. PMID 18295503. S2CID 5305444.
- Bahr, Ute; Deppe, Andreas; Karas, Michael; Hillenkamp, Franz; Giessmann, Ulrich (1992). "Mass spectrometry of synthetic polymers by UV-matrix-assisted laser desorption/ionization". Analytical Chemistry. 64 (22): 2866–2869. doi:10.1021/ac00046a036. ISSN 0003-2700.
- Seng, P.; Drancourt, M.; Gouriet, F.; La Scola, B.; Fournier, P. E.; Rolain, J. M.; Raoult, D. (2009). "Ongoing revolution in bacteriology: routine identification of bacteria by matrix-assisted laser desorption ionization time-of-flight mass spectrometry". Clinical Infectious Diseases. 49 (4): 552–3. doi:10.1086/600885. PMID 19583519.
- Sandrin, Todd R.; Goldstein, Jason E.; Schumaker, Stephanie (2013). "MALDI TOF MS profiling of bacteria at the strain level: A review". Mass Spectrometry Reviews. 32 (3): 188–217. Bibcode:2013MSRv...32..188S. doi:10.1002/mas.21359. ISSN 0277-7037. PMID 22996584.
- Downard, Kevin M. (2013). "Proteotyping for the rapid identification of influenza virus and other biopathogens". Chemical Society Reviews. 42 (22): 8584–8595. doi:10.1039/c3cs60081e. ISSN 1460-4744. PMID 23632861.
- Capocefalo M., Ridley E. V., Tranfield E. Y. & Thompson K. C. (2015). "Ch. 9 - MALDI-TOF: A rapid microbiological confirmation technique for food and water analysis". In Cook N.; D'Agostino M.; Thompson K. C. (eds.). Molecular Microbial Diagnostic Methods - Pathways to Implementation for the Food and Water Industries. Elsevier. ISBN 978-0-12-416999-9.CS1 maint: multiple names: authors list (link)
- Seon Young Kim, Jeong Su Park, Yun Ji Hong, Taek Soo Kim, Kiho Hong, Kyoung-Ho Song, Hyunju Lee, Eu Suk Kim, Hong Bin Kim, Kyoung Un Park, Junghan Song, Sun Hoe Koo, Eui-Chong Kim (2019). "Microarray-Based Nucleic Acid Assay and MALDI-TOF MS Analysis for the Detection of Gram-Negative Bacteria in Direct Blood Cultures". Am. J. Clin. Pathol. 151: 143–153. doi:10.1093/AJCP/AQY118.CS1 maint: multiple names: authors list (link)
- Rhoads, DD (2016). "The presence of a single MALDI-TOF mass spectral peak predicts methicillin resistance in staphylococci. et al.". Diagn Microbiol Infect Dis. 86 (3): 257–261. doi:10.1016/j.diagmicrobio.2016.08.001. PMID 27568365.
- Vogne, C.; et al. (2014). ""A simple, robust and rapid approach to detect carbapenemases in Gram-negative isolates by MALDI-TOF mass spectrometry " validation with triple quadripole tandem mass spectrometry, microarray and PCR". Clin Microbiol Infect. 20 (12): O1106-12. doi:10.1111/1469-0691.12715. PMID 24930405.
- Abouseada, N; Raouf, M; El-Attar, E; Moez, P (2017). "Matrix-assisted laser desorption ionisation time-of-flight mass spectrometry rapid detection of carbapenamase activity in Acinetobacter baumannii isolates". Indian J Med Microbiol. 35 (1): 85–89. doi:10.4103/0255-0857.202335. PMID 28303824. S2CID 20480725.
- Sakarikou, C; Ciotti, M; Dolfa, C; Angeletti, S; Favalli, C (2017). "Rapid detection of carbapenemase-producing Klebsiella pneumoniae strains derived from blood cultures by Matrix-Assisted Laser Desorption Ionization-Time of Flight Mass Spectrometry (MALDI-TOF MS)". BMC Microbiol. 17 (1): 54. doi:10.1186/s12866-017-0952-3. PMC 5343375. PMID 28274205.
- Lau, A.; et al. (August 2014). "A Rapid Matrix-Assisted Laser Desorption Ionization–Time of Flight Mass Spectrometry-Based Method for Single-Plasmid Tracking in an Outbreak of Carbapenem-Resistant Enterobacteriaceae". Journal of Clinical Microbiology. 52 (8): 2804–2812. doi:10.1128/JCM.00694-14. PMID 4136129. S2CID 43633405.
- Avila, C. C.; Almeida, F. G.; Palmisano, G. (2016). "Direct identification of trypanosomatids by matrix-assisted laser desorption ionization-time of flight mass spectrometry (DIT MALDI-TOF MS)". Journal of Mass Spectrometry. 51 (8): 549–557. Bibcode:2016JMSp...51..549A. doi:10.1002/jms.3763. ISSN 1076-5174. PMID 27659938.
- Lachaud, Laurence; Fernández-Arévalo, Anna; Normand, Anne-Cécile; Lami, Patrick; Nabet, Cécile; Donnadieu, Jean Luc; Piarroux, Martine; Djenad, Farid; Cassagne, Carole; Ravel, Christophe; Tebar, Silvia; Llovet, Teresa; Blanchet, Denis; Demar, Magalie; Harrat, Zoubir; Aoun, Karim; Bastien, Patrick; Muñoz, Carmen; Gállego, Montserrat; Piarroux, Renaud; Loeffelholz, Michael J. (2017). "Identification of Leishmania by Matrix-Assisted Laser Desorption Ionization–Time of Flight (MALDI-TOF) Mass Spectrometry Using a Free Web-Based Application and a Dedicated Mass-Spectral Library". Journal of Clinical Microbiology. 55 (10): 2924–2933. doi:10.1128/JCM.00845-17. ISSN 0095-1137. PMC 5625378. PMID 28724559.
- Laroche, Maureen; Almeras, Lionel; Pecchi, Emilie; Bechah, Yassina; Raoult, Didier; Viola, Angèle; Parola, Philippe (2017). "MALDI-TOF MS as an innovative tool for detection of Plasmodium parasites in Anopheles mosquitoes". Malaria Journal. 16 (1): 5. doi:10.1186/s12936-016-1657-z. ISSN 1475-2875. PMC 5209920. PMID 28049524.
- Ouarti, Basma; Laroche, Maureen; Righi, Souad; Meguini, Mohamed Nadir; Benakhla, Ahmed; Raoult, Didier; Parola, Philippe (2020). "Development of MALDI-TOF mass spectrometry for the identification of lice isolated from farm animals". Parasite. 27: 28. doi:10.1051/parasite/2020026. ISSN 1776-1042. PMC 7191974. PMID 32351208.
- Huguenin, Antoine; Depaquit, Jérôme; Villena, Isabelle; Ferté, Hubert (2019). "MALDI-TOF mass spectrometry: a new tool for rapid identification of cercariae (Trematoda, Digenea)". Parasite. 26: 11. doi:10.1051/parasite/2019011. ISSN 1776-1042. PMC 6402365. PMID 30838972.
- Sim, K.; Shaw, A. G.; Randell, P.; Cox, M. J.; McClure, Z. E.; Li, M. S.; Haddad, M.; Langford, P. R.; Cookson, W. O.; Moffatt, M. F.; Kroll, J. S. Dysbiosis anticipating necrotizing enterocolitis in very premature infants. Clin. Infect. Dis. 2014.
- Grantzdorffer, I; Carl-McGrath, S; Ebert, MP; Rocken, C (2008). "Proteomics of pancreatic cancer". Pancreas. 36 (4): 329–36. doi:10.1097/MPA.0b013e31815cc452. PMID 18437077. S2CID 29118712.
- Dhillon, AS; Hagan, S; Rath, O; Kolch, W (2007). "MAP kinase signalling pathways in cancer". Oncogene. 26 (22): 3279–90. doi:10.1038/sj.onc.1210421. PMID 17496922. S2CID 17464318.
- Zhong, N.; Cui, Y.; Zhou, X.; Li, T.; Han, J. Identification of prohibitin 1 as a potential prognostic biomarker in human pancreatic carcinoma using modified aqueous two-phase partition system combined with 2D-MALDI-TOF-TOF-MS/MS. Tumour Biol. 2014.
- Hrabák, Jaroslav (2015). "Detection of Carbapenemases Using Matrix-Assisted Laser Desorption/Ionization Time-of-Flight Mass Spectrometry (MALDI-TOF MS) Meropenem Hydrolysis Assay". Methods in Molecular Biology (1064-3745), 1237, p. 91.
- Woods, AS; Buchsbaum, JS; Worrall, TA; Berg, JM; Cotter, RJ (1995). "Matrix-Assisted Laser Desorption/Ionization of Non-covalently Bound Compounds". Anal. Chem. 67 (24): 4462–4465. doi:10.1021/ac00120a005.
- Kiselar, JJG; Downard, KM (2000). "Preservation and Detection of Specific Antibody-Peptide Complexes by Matrix-Assisted Laser Desorption/Ionization Mass Spectrometry". J. Am. Soc. Mass Spectrom. 11 (8): 746–750. doi:10.1016/S1044-0305(00)00144-6. PMID 10937798. S2CID 33248551.
- Downard, KM (2006). "2". Softly, Softly—Detection of Protein Complexes by Matrix‐Assisted Laser Desorption Ionization Mass Spectrometry. Wiley. pp. 25–43. doi:10.1002/9780470146330.ch2. ISBN 9780470146330.
- Ragoussis, J.; Elvidge, G. P.; Kaur, K.; Colella, S. (2006). "Matrix-Assisted Laser Desorption/Ionisation, Time-of-Flight Mass Spectrometry in Genomics Research". PLOS Genetics. 2 (7): e100. doi:10.1371/journal.pgen.0020100. PMC 1523240. PMID 16895448.
- Hardouin, J. (2007). "Protein sequence information by matrix-assisted laser desorption/ionization in-source decay mass spectrometry". Mass Spectrometry Reviews. 26 (5): 672–82. Bibcode:2007MSRv...26..672H. doi:10.1002/mas.20142. PMID 17492750.
- Peter-Katalinic, J.; Hillenkamp, F. (2007). MALDI MS: A Practical Guide to Instrumentation, Methods and Applications. Wiley-VCH. ISBN 978-3-527-31440-9.
- Schrepp, W.; Pasch, H. (2003). MALDI-TOF Mass Spectrometry of Synthetic Polymers. Springer-Verlag. ISBN 978-3-540-44259-2.
- Primer on Matrix-Assisted Laser Desorption Ionization (MALDI) National High Magnetic Field Laboratory